US 20060075807 A1
In accordance with one aspect of the present invention, a cantilever of a probe-based instrument is deflected by directing a beam of ultrasonic acoustic energy at the cantilever to apply acoustic radiation pressure to the cantilever. The energy is generated by an acoustic actuator. The transmitted beam preferably is focused using a cylindrical lens, providing a beam tightly focused in one dimension and unfocused in a second dimension. In accordance with another aspect of the present invention, a power source such as an RF signal generator is operated so as to spread the spectrum of acoustic radiation on a time scale that is short or comparable to the acoustic roundtrip time. Such a design diminishing the resonance effects and sensitivity to spacing between the cantilever and the acoustic source.
10. A probe-based instrument comprising:
an acoustic actuator configured to direct a beam of acoustic energy at the cantilever so as to drive the cantilever; and
an RF excitation signal passed to the acoustic actuator to generate the beam of acoustic energy, the RF excitation signal being spread over a frequency spectrum to reduce resonance effects of the signal.
11. The probe-based instrument according to
12. The probe-based instrument according to
13. The probe-based instrument according to
14. The probe-based instrument according to
15. The probe-based instrument according to
16. The probe-based instrument according to
17. The probe-based instrument of
18. The probe-based instrument of
19. The probe-based instrument according to
20. The probe-based instrument according to
21. The probe-based instrument according to
22. The probe-based instrument according to
23. The probe-based instrument according to
24. The probe-based instrument according to
25. The probe-based instrument according to
26. A method of moving a cantilever of a probe-based instrument comprising:
generating an RF excitation signal;
passing the RF excitation signal to an acoustic actuator; and
generating a beam of acoustic energy, from the RF signal, at the cantilever so as to drive the cantilever, wherein the RF excitation signal is spread over a frequency spectrum to reduce resonance effects of the signal.
27. A probe-based instrument comprising:
an acoustic actuator configured to direct a beam of acoustic energy at the cantilever so as to drive the cantilever; and
an RF excitation signal passed to the acoustic actuator to generate the beam of acoustic energy, the RF excitation signal being spread over a frequency spectrum to drive the actuation so that acoustic power perturbations caused by resonances and anti-resonances of different frequencies from the acoustic actuator act to neutralize each other.
This application claims the benefit of U.S. patent application Ser. No. 10/323,384, filed Dec. 18, 2002, the disclosure of which is incorporated herein in its entirety, by reference.
The present application is directed to driving systems, and more particularly to a method and apparatus for driving a cantilever through the use of acoustic wave pressure generated by an ultrasonic actuator.
Scanning probe microscopes (SPMs) represent a category of probe-based instruments designed to characterize a surface of a sample at an atomic level, through the monitoring of an interaction between a sample and a tip on a cantilever probe. This interaction is primarily a scanning operation between the tip and sample, whereby data regarding characteristics of the surface is acquired and used to generate an image of the sample region. The image data is commonly acquired via a raster scan of the sample.
A particular type of SPM is known as an atomic force microscope (AFM), which functions by measuring local properties of a sample, such as height, optical absorption, magnetism or other measurable characteristic. The resulting image will resemble an image on a television screen, as it consists of both many rows or lines of information placed one above the other.
AFMs are designed to operate in a variety of modes, including non-contact mode, contact mode and oscillating mode, also known as a tapping mode. In the non-contact mode, the AFM generates a topographic image from measurements based on attractive forces. In this design, the tip does not touch the sample. The non-contact mode does not function effectively in liquids.
In the contact mode of operation, the AFM scans the tip across the surface of the sample, while the force of the tip on the surface of the sample is maintained at a generally constant value. This contact operation is accomplished by moving either the sample or the probe assembly vertically to the surface of the sample upon sensing a deflection of the cantilever as the probe is scanned horizontally across the surface.
In the oscillating mode, the tip is made to oscillate at or near a resonant frequency of the cantilever. The amplitude or phase of this oscillation is kept constant during scanning using feedback signals, which are generated in response to tip-sample interaction. Similar to the contact mode, the feedback signals are then collected, stored and used as data to generate an image of the sample area.
Atomic force microscopes are known to have a resolution down to the atomic level for a wide variety of surfaces. While the general concept of an AFM is similar to that of a record player, as well as the stylus profilometer, to obtain and enable the atomic-scale resolution, AFMs have refinements, including an optical sensor which operates by reflecting a laser beam off of the cantilever. Angular deflection of the cantilever causes an angular deflection of the laser beam. The reflected laser beam strikes a position detector to indicate the position of the laser spot on the detector, and thus the angular deflection of the cantilever.
An area of particular interest in designing an AFM is the mechanism employed to provide an external force to deflect or oscillate the cantilever. In existing AFMs, the cantilever will typically be oscillated by a piezoelectric actuation.
Traditional piezoelectric drives act on the base of the cantilever, not on a free-end portion. Therefore, these systems must apply substantially greater forces to the cantilever to obtain a given deflection magnitude at the free end than would be required if force were applied directly to the free end or the body of the cantilever. Such a design results in certain limitations.
Since a typical AFM cantilever is easily excited to resonance in air, the piezoelectric drive is useful in this environment. However, piezoelectric drives are not very useful in liquid (e.g., water) environments. The reason for this has to do with the quality factor or Q of a resonance of the cantilever. The quality factor, Q, denotes the sharpness of a cantilever's resonance curve. A resonance with a large Q can be excited to relatively large cantilever oscillation amplitudes with relatively small excitation forces. For operation in air or other gaseous environments, the typical piezoelectric drive provides ample excitation force to drive the cantilever to produce a resonance peak that is easily identified and distinguished from parasitic resonance peaks, such as those of the mounts for the cantilever and the piezoelectric drive itself.
Conversely, a cantilever operated in liquids such as water, has a dramatically lower Q, as the liquid dampens the oscillating cantilever. A typical piezoelectric drive does not have enough gain to excite the cantilever sufficiently to produce a resonance peak that is easily located and differentiated from parasitic resonances.
In view of this, specialized cantilever drives have been developed to act along the length of the cantilever rather than only on the base. One such drive is known as a magnetic drive. The typical magnetic drive has a magnetic cantilever that is driven by an electromagnetic force. The cantilever has a fixed base rigidly attached to a support and bears a tip on its free end that interacts with a sample. The cantilever is rendered magnetic by coating one or more of its surfaces with a magnetic layer. By controlling the amplitude of the applied magnetic field, the cantilever can be deflected while the tip interacts with the sample.
However, a magnetic drive has inherent limitations that considerably restrict its range of applications. For instance, it requires a special magnetically-coated cantilever and cannot therefore be used in applications where the cantilever should not be coated with magnetic material. It is also not applicable to situations where the magnetic properties of the sample and/or the environment results in undesirable deflection of the cantilever, producing errors in the measurements. The operating ranges of the magnetic drive system are also limited.
An acoustic drive has also been considered to drive the cantilever. In this design, a cantilever and piezoelectric drive are mounted on a common head in a spaced-apart relationship. The head is mounted above a fluid cell, and the cantilever extends into the fluid cell to interact with the sample in the cell. The piezoelectric drive can be excited by a signal generator to generate acoustic waves that propagate through the glass walls of the fluid cell, through the fluid in the cell, then on to the cantilever, causing the cantilever to oscillate.
Acoustic drives of this type have various disadvantages. For instance, the unfocused acoustic energy will impinge on many other components of the system, such as mounts for the cantilever and the piezoelectric drive, the fluid cell, and even the fluid, exciting resonances in those components. These resonances can be difficult to distinguish from the cantilever resonance. The acoustic drive also has sufficient actuation force at a limited selection of operation frequencies and can be a challenge to match the cantilever resonance with the operation frequency of the acoustic actuator. If a user selects a resonance that does not overlap with the cantilever resonance, the measurements may be unstable.
Pending application U.S. Ser. No. 10/456,136 (Publication No. US 2004-0020279 A1, entitled “Method and Apparatus for the Ultrasonic Actuation of the Cantilever of a Probe-Based Instrument”; U.S. Pat. No. 6,779,387 (U.S. Ser. No. 10/095,850 (Publication No. US 2003-0041657 A1)), entitled “Method and Apparatus for the Ultrasonic Actuation of the Cantilever of a Probe-Based Instrument”; and U.S. Pat. No. 6,694,817 (U.S. Ser. No. 10/096,367 (Publication No. US2003-0041669A1)), entitled “Method and Apparatus for the Ultrasonic Actuation of the Cantilever of a Probe-Based Instrument”, (all claiming priority to provisional patent application Ser. No. 60/313,911, filed Aug. 21, 2001) (commonly assigned) (all hereby incorporated by reference), describes an ultrasonic force microscope (UFM) intended to have an actuator that drives a cantilever to produce a “clean” frequency response, preferably by driving the cantilever body, rather than the base. It is stated that by driving the body of the cantilever with an ultrasonic actuator, a much higher localized force can be achieved through the use of a traditional piezoelectric actuator. The beam used for actuation is preferably shaped, i.e., manipulated to limit unwanted propagation and directions other than toward the cantilever, so that ultrasonic energy impinges at least primarily on the cantilever.
Two suitable techniques for shaping the beam are listed as focusing and collimation. The ultrasonic small diameter beams can be focused on the cantilever using a Fresnel lens or other focusing device located between the ultrasonic actuator and the cantilever. It is noted that the Fresnel lens may focus the ultrasonic beam to a diameter of approximately 5 μm to 10 μm at a focal distance of 360 μm, where the 5 μm diameter is even smaller than the 8 μm to 12 μm diameter of most laser beams. As a result, it is stated the lens can be used to apply a pinpoint force to the free end of the cantilever or any other point of interest along the length of the cantilever.
It is also proposed that in an alternative embodiment, the beam may be intentionally sized larger than the cantilever to account for tolerances in alignment of the cantilever and the ultrasonic actuator. If, for example, the cantilever is 50 microns wide, and can be reproducibly aligned within ±100 microns, an ultrasonic actuator with a beam width of 250-300 μm in the region of the cantilever could guarantee that a portion of the ultrasonic beam would always strike the cantilever.
Thus, in the incorporated applications, the preferred embodiment purports to disclose an apparatus and procedure for providing pinpoint actuation energy to a cantilever. It is also acknowledged that alignment errors may exist between the cantilever and actuator, whereby the pinpoint accuracy may result in the acoustic beam not impacting the cantilever.
Misalignment of a probe may occur during manufacture or when the cantilever probe is replaced. Particularly, it is known that in operation, the tip carried on the cantilever becomes damaged or worn and will require replacement. Normally, the cantilever and tip come as a single unit, and the entire unit is replaced with a new cantilever/tip arrangement. This replacement operation is a mechanical operation, and a degree of imprecision in the alignment procedure exists. Therefore, when the cantilever/tip arrangement is inserted, and a focused pinpoint (i.e., small diameter) beam is used, misalignment may result in improper interaction between the actuator and the cantilever. To address this issue it is proposed that the beam (i.e., the diameter) is enlarged.
From the foregoing, it can be seen there are concerns related to use of a pinpoint acoustic beam due to misalignment issues. A further issue, however, is that widening the beam to address misalignment causes an increase in undesirable reflections and resonance between the acoustic source and the cantilever. These resonances can result in strong variations in the acoustic force delivered to the cantilever as the resonance conditions vary with variations in spacing between the cantilever and acoustic source.
In accordance with one aspect of the present invention, a cantilever of a probe-based instrument is deflected by directing a beam of ultrasonic acoustic energy at the cantilever to apply acoustic radiation pressure to the cantilever. The energy is generated by an acoustic actuator. The transmitted beam preferably is focused using a cylindrical lens, providing a beam tightly focused in one dimension and unfocused in a second dimension. In accordance with another aspect of the present invention, a power source such as an RF signal generator is operated so as to spread the spectrum of acoustic radiation on a time scale that is short or comparable to the acoustic roundtrip time. Such a design diminishes the resonance effects sensitivity to spacing between the cantilever and the acoustic source.
The base of the cantilever 22 is fixed to an optically transparent probe support 26, which may fit into a commercial AFM scanhead 28. The fluid cell 14 is positioned beneath the scanhead 28 with the probe 12 closely located. An ultrasonically transparent substrate 30, preferably made of a hard substrate like glass or silicon, is placed below the probe 12 and supports fluid cell 14. The ultrasonic actuator 16, which is preferably formed from a zinc oxide (ZnO) transducer, is mounted on the bottom of the substrate 30, although other materials which will generate the appropriate output may also be used. Alternatively, the ultrasonic actuator can be placed on the side of the substrate directly facing the cantilever. The ultrasonic transducer can then be shaped as a focusing device and electrically isolated from the fluid environment, removing the need for an ultrasonically transparent substrate.
The ultrasonic actuator 16 is driven by a RF signal generator 32 to generate a beam 34, focussed by a lens 36 that deflects the cantilever 22 away from the substrate 30. The RF signal generator 32 has an optional modulation input that allows the amplitude of the RF signal to be varied with time. The modulation signal may be a square wave, a sinusoidal wave, a triangle wave or an arbitrary time-varying modulation. The RF signal generator also has an input or an internal adjustment that allows control over the baseline (unmodulated) power of the RF signal.
The scanhead 28 may include an XY actuator and a Z actuator to permit the probe 12 to scan a sample (not shown) in the fluid cell 14. Alternatively, the scanhead 28 could be stationary, and the substrate 30 could be driven to scan relative to the scanhead 28.
Detector system 18, 20 detects cantilever deflection during scanning. The detector system includes a laser 18 positioned above cantilever 22 and a photodetector system 20 configured to receive laser light reflected from the upper surface of the cantilever 22. As is conventional, signals from the photodetector 20 can be used as feedback to control operation of the RF signal generator 32 to maintain a desired characteristic of cantilever deflection, such as magnitude, and/or phase during scanning.
It is known that a plane target placed in the path of an acoustic wave beam in an unconfined medium experiences a time averaged force per unit area. This pressure is known as the “Langevin acoustic radiation pressure” (ARP). Forces imposed by the application of this pressure are used to deflect the cantilever of the probe-based-instrument, such as probe 12. The ARP is related to the average energy density, U, at the target surface. As a simple model based on plane waves, it can be assumed that, at the focal plane, a time harmonic acoustic pressure wave of amplitude, Pi, is normally incident on a cantilever immersed in a liquid and that the wave is reflected with a complex pressure reflection coefficient, Γ. This reflection coefficient can be considered as a weighted average over the incident spectrum of plane waves that would be included in a focused beam. In this case, the time-averaged energy density at the cantilever surface will be given by
where ρ is the bulk density of fluid, c is the speed of sound in the liquid and Γ denotes the absolute value of the reflection coefficient. Using the relation that the average intensity of the incident beam is given by II=Pi 2/(2ρc), the Langevin radiation pressure on the cantilever 22, Ω, can be expressed in terms of the intensity as
As previously mentioned, and now more particularly illustrated in
An alternative operation to avoid this issue is to widen the beam, such as for example shown in
To address these issues, in one embodiment, lens 36 of
By implementing a cylindrical lens with AFM 10, it is ensured, as illustrated in
While some of wave energy from the linear beam towards the cylindrical lens, providing a degree of undesirable buildup of standing waves due to multiple reflections, most of incident energy falls outside of the cantilever, and is dispersed so as to minimize reflective buildup. Also, diffraction losses are greater in such a focused system, and round-trip acoustic reflections will result in smaller resonances than for an unfocused plane wave illumination.
It is appreciated that while the implementation of a cylindrical lens greatly reduces undesirable reflection, some resonances due to reflected waves will still exist.
Particularly, it is understood that typically the roundtrip propagation time for the return of the reflected radiation to cantilever is shorter than the duration of the very narrow band (i.e., single frequency) RF tone bursts that are used for driving the actuator, so the reflected and the non-reflected radiation incident on the cantilever coherently interfere. This interference may be constructive or destructive, and the resonance conditions change as the spacing between the cantilever and acoustic source varies by as little as one quarter acoustic wavelength. These resonances will result in strongly varying acoustic pressure on the cantilever for small changes in the spacing between the cantilever and acoustic source.
In accordance with the present embodiment, provision is made for significantly reducing the effect of resonances on the acoustic power density of the acoustic beam or beams that are incident on the cantilever, which thereby reduce the accuracy of the detected readings. Approaches to accomplish this employ multi-frequency and/or employ frequency RF voltage pulses for driving the actuation so that the acoustic power perturbations caused by the resonances and anti-resonances of the different frequencies tend to neutralize each other.
A first approach to diminish the effects of undesirable reflectance is to drive the actuator with multifrequency RF tone bursts, such that the power perturbations caused by the resonances of one frequency component substantially offset or neutralize the perturbations caused by the anti-resonances of another frequency component, and vice-versa. More particularly, referring to the dual tone driver 70 illustrated in
If applied together to the acoustic source, the power perturbations caused by the resonances and anti-resonances of f1 and f2 will tend to neutralize each other, thereby reducing the inaccuracies of the detected deflection values.
Alternatively, or in addition, the frequency content of the RF drive pulses may be increased. For example, as shown in
Alternatively, a linear chirp signal could be employed to modulate the RF carrier frequency, this embodiment is desirably implemented with the carrier frequency modulated at a high rate. Still another alternative is to employ data modulated, essentially “white” RF noise, for driving the transducer 16.
In the dual tone embodiment of
By implementation of these reduction techniques, undesirable reflections of
As is apparent from the above discussion, the force imposed by the transducer 16 is unidirectional. As a result, ultrasonic beam 34 cannot pull the cantilever 22 in the propagation direction. If it is desired to configure the AFM 10 to selectively pull the cantilever 22 toward the substrate 30 rather than push it away from the substrate, the cantilever 22 could be manufactured with a bias that maintains it in contact with the substrate 30 in the absence of a drive signal to the RF signal generator 32. The RF signal generator 32 could then be driven to overcome the bias and push the cantilever 22 from the substrate 30. The drive voltage could then be reduced to permit the cantilever 22 to move towards the substrate 30, hence, in effect, pulling the cantilever 22 towards the substrate 30. Alternatively, in some configurations, an ultrasonic actuator could be placed both above and/or below the cantilever to push from one or both sides.
The previous discussion demonstrates a particular capability of the current concepts. The ultrasonic cantilever actuator with a cylindrical lens can independently control both the DC and AC forces applied to the cantilever over an extremely wide bandwidth. This opens up a range of applications for this actuator. One example is an imaging method called Force Modulation. In this method, an AFM tip is brought into contact with a sample surface and then an AC modulation force is applied to the cantilever. The detector then measures the amount of AC deflection of the cantilever. On hard samples, the cantilever cannot indent into the surface, and no deflection is detected. On softer samples, the ultrasonic modulation force causes the tip to indent into the sample, resulting in a measurable AC deflection of the cantilever. Separate control over the DC force allows control over the tracking force that the AFM system uses to maintain contact between the tip and the surface.
The RF voltage can also be varied slowly to permit a quasistatic measurement to be performed. A quasistatic force imposition process is considered to be one in which, if the forces were to be removed at any stage during the process, the system would be in equilibrium from that time on. Hence, the RF voltage can be changed slowly enough to maintain equilibrium while the voltage is being altered. This procedure is in contrast to a dynamic (AC) measurement in which the RF signal is modulated at a high frequency and the system requires time to stabilize when force adjustment terminates. The cutoff between quasistatic and dynamic measurements is usually considered to be a frequency value below the cantilever's fundamental resonant frequency.
The relationship between cantilever deflection and drive voltage magnitude is proportional and can be used to obtain useful information concerning a sample. For instance, the RF signal generator 32 of
AFM cantilevers can also respond dynamically to radiation pressure, and those dynamic responses can be measured. Specifically, time harmonic forces can be generated by applying a sinusoidal amplitude modulation on the RF input signal. By choosing the modulation factor to be less than one, an appropriate biasing force can be applied to actuate the cantilever at the modulation frequency and its second harmonic. The deflection of the cantilever can then be recorded, e.g., by using a lock-in amplifier, which uses the modulation signal as its reference input and locks to the modulation frequency.
It has been discovered that an ultrasonic-actuator based system has a very wide bandwidth for exciting the cantilever. In practice, it is possible to modulate the RF signal up to about 1/10 of the RF signal frequency or even higher. For a 300 MHz RF frequency, a cantilever actuation bandwidth of even 30 MHz is realizable. In fact, it is believed that the RF frequency from a low of 10 MHz or possibly even lower for air applications to 1 GHz or even higher for water applications with surface micromachined cantilevers for which attenuation may not be a problem. This cantilever actuation bandwidth is much greater than is provided by other AFM cantilever actuators, particularly acoustic and magnetic drives. An AFM having an ultrasonically driven cantilever can therefore be used to scan at rates that would have theretofore been considered unobtainable.
An ultrasonic actuator of the type described above can be used to drive an AFM cantilever to oscillate at virtually any desired frequency significantly below the RF carrier frequency. An ultrasonic actuator therefore can be directly used as a tapping mode actuator in an AFM. A tapping mode AFM 110 using an ultrasonic actuator is shown schematically in
The traditional tapping mode piezoelectric drive may be taken out of the loop and replaced by the ultrasonic actuator 116. In this case, the RF drive signal described in the preceding paragraph would always be used to drive the ultrasonic actuator 116. In the preferred embodiment however, a piezoelectric drive 140 can be retained, and a suitable switch 142 can be provided to permit a drive signal to be selectively transmitted to either modulate the output of RF signal generator 132 and hence activate the ultrasonic actuator, or the drive signal can be sent to the piezoelectric drive 140 directly from the AFM controller 138. For example, the resulting instrument could be operated in either air or liquid, with the piezoelectric drive 140 being used to effect operation in air and the ultrasonic actuator 116 being used to effect operation in liquid.
An issue of the instruments previously illustrated is that the sample is ultrasonically transmissive to permit unfettered transmission of the ultrasonic beam from the ultrasonic actuator, through the sample, and to the cantilever. A more versatile ultrasonic actuator assembly is schematically illustrated in
An instrument 210 configured for elasticity characterization is illustrated in
While the discussion has described the invention primarily incorporated into AFMs, it is to be understood the invention is not limited to the described embodiments or even to AFMs in general. Rather, it is applicable to virtually any probe-based instrument in which a cantilever is deflected by directing a focused beam of acoustic energy at the cantilever to apply ultrasonically generated acoustic radiation pressure to the cantilever. A variety of different ultrasonic acoustic actuators and associated drives may be applied to achieve these affects, and cantilever deflection may be measured using a variety of techniques. Also, while there is particular usefulness for use of cantilevers in fluid, such as water embodiments, other fluids may also take advantage of the present applications, including those in gases, the air or other non-water liquids as well as use in vacuums.
The invention has been described with reference to the preferred embodiments. Obviously, modifications and alterations will occur to others upon reading and understanding the preceding detailed description. It is intended that the invention be construed as including all such modifications and alterations insofar as they come within the scope of the appended claims or the equivalents thereof.