|Publication number||US4278089 A|
|Application number||US 05/959,161|
|Publication date||Jul 14, 1981|
|Filing date||Nov 9, 1978|
|Priority date||Nov 9, 1978|
|Also published as||DE2945379A1|
|Publication number||05959161, 959161, US 4278089 A, US 4278089A, US-A-4278089, US4278089 A, US4278089A|
|Inventors||Charles M. Huck, John E. Studer, Philip H. Sauer|
|Original Assignee||Howmedica, Inc.|
|Export Citation||BiBTeX, EndNote, RefMan|
|Patent Citations (17), Referenced by (105), Classifications (10), Legal Events (4)|
|External Links: USPTO, USPTO Assignment, Espacenet|
This invention relates to suction devices and more particularly to an improved device for automatic removal of fluids from a wound.
The use of suction devices to remove fluids from a wound is known to the art. Such devices are employed to reduce or remove the body fluids which normally collect at a wound site after surgery.
Body fluids which collect at a wound, if left to remain especially in a closed wound, may cause complications in the healing process and thereby reduce the chance of a successful convalescence. Moreover, not only will a dry wound area accelerate the healing stage but will also promote cicatrization.
Known typical suction devices are both disclosed and illustrated in Canadian Pat. No. 879,849; British Pat. No. 1,304,324 to Astra-Meditec; British Pat. No. 1,400,139 to Mathys; U.S. Pat. No. 3,115,138 to McElvenny et al.; U.S. Pat. No. 3,376,868 to Mondiadis; U.S. Pat. No. 3,742,952 to Magers et al.; U.S. Pat. Nos. 3,774,611 and 3,779,249 to Tussey et al.; U.S. Pat. No. 3,809,086 to Schachet et al.; U.S. Pat. No. 3,809,087 to Lewis, Jr.; and U.S. Pat. No. 3,875,941 to Adair.
With reference to the Canadian patent, the device disclosed therein includes a container with a concertina-pleated wall that can be expanded to increase the volume of the container. A detachable external spring bow serves to extend the pleated wall thereby causing suction within the container. The concertina-pleated walls are consecutively numbered to permit measurement of the fluid collected when the container is in an open, extended position.
The Astra-Meditec patent discloses a bellows operated disposable aspiration drainage device. A non-return inlet valve permits the collection of body fluids into the bellows from an inlet catheter connected to the wound to be drained. The collected body fluids are discharged through a non-return outlet valve into a receiving bag. The latter is graduated to allow measurement of removed body fluids.
The Mathys patent discloses a suction drainage device which includes a resiliently compressible, airtight container having a watertight plastic bag detachably secured therein. The container upon expansion produces a negative pressure which results in the flow of body fluids into the bag within. A helical spring can be used inside the container to provide expansion of the container.
The McElvenny patent discloses an evacuator that includes a fluid tight container. After manual compression, the container is expansible to provide suction either by means of internally positioned springs between additional members placed within the container or by virtue of the container being constructed of resilient material.
The evacuator disclosed in the Mondiadis patent includes a compressible container made from resilient elastomeric material whose memory causes the walls to return to an uncompressed state. A valve turret on the top of the container provides both an inlet and a vent opening which is sealable by means such as a plug or a resilient diaphragm connected to the top.
The Magers et al. patent discloses a surgical suction assembly including a suction container which is resiliently compressible and expansible. A one way plug is provided to admit fluid into the container while drawing fluid and to block flow outwardly therefrom.
The Tussey '611 patent discloses a contamination free evacuator including a compressible-expandable container having an inlet and an outlet region, and a fluid stabilizer inside the container.
A contamination free evacuator is also disclosed in the Tussey '243patent which includes a compressible and expandable evacuator bag having both a drain inlet and an exhaust outlet formed in the bag. The latter further includes a magnetic one-way exhaust valve which prevents the contamination of the bag by closing upon completion of exhausting fluids from the bag.
The Schachet et al. patent is directed to a wound suction device including a vessel having therein an elastic sheet diaphragm which is sealed to the walls of the vessel. Movement of the diaphragm by a plate with an actuating means decreases the volume of the vessel which results in the creation of a suction force upon release of the actuating means. Provision is made to lock the actuator means in a position such that the resilient diaphragm remains displaced.
In the Lewis patent, a suction apparatus is disclosed having a collapsible bag attached at opposed surfaces to plate members one of which is biased to move away from the other plate member. Disclosed biasing means include springs and rubber bands.
The evacuating system disclosed in the Adair patent includes a bellows-type container having resilient accordian-like sidewalls made of an elastomeric material. Initial suction of the container is provided by the resilient accordian-like sidewalls. Further suction results from the added weight of collected fluid within the container. Operation of this system requires that the bellows-type container be mounted in a vertical position with evacuating tubing attached to the top portion of the container.
Although each of the above-mentioned patented devices is useful in varying degrees, they all suffer from some of several disadvantages which include, e.g., the necessity of activation just prior to use, the lack of ease in operation, the need of resilient containers or component members, complex structure, and the inability to maintain a substantially constant level of negative pressure.
All of the above patents except Schachet provide no disclosure of a device which can be received by the handler in a preactivated or compressed state. These devices require that the handler collapse the bellows or container to activate the device before use. Difficulty in operation is presented particularly in the Astra-Meditec apparatus, the Canadian drain, and the Adair device. The Astra-Meditec apparatus discloses a directional multicomponent valve system which requires that the handler properly connect the device to the wound and the collecting bag. Multi-component complex valve arrangements are also disclosed in the Magers valve means and the Tussey '243 magnetic exhaust valve. The Canadian drain not only requires that the user compress the concertina-pleated walls to drain air from the device but also that an external spring bow be attached further complicating the operation thereof. In the Adair device, difficulty arises from the requirements that the device be maintained upright for proper operation. Increased cost of operation or construction results from the multi-component or complex valve systems disclosed in the Astra-Meditec, Magers, and Tussey '243 patents; and from the need of additional collecting bags in the Astra-Meditec and Mathys patents. Resilient retaining containers or component members are disclosed in one form or another in most of the above patents. Complex structure is formed in the multiple spring arrangement disclosures of the McElvenny, Tussey '243, and Lewis patents, and in the need of a fluid stabilizer as disclosed in the Tussey '611 patents. Such complex structures complicate construction as well as increase the cost thereof. Finally, none of the above patents disclose a device providing substantially constant suction over the range of operation. The spring means disclosed in the Canadian, Mathys, McElvenny, Tussey '243, and the Lewis device do not allow for a substantially constant suction operation.
The device of the present invention is intended to improve over the apparatus discussed above and to overcome the limitations found therein. The device of the present invention is designed to provide a substantially constant flow rate throughout the range of drawing and collecting body fluids from a wound into the device. Also the device is available to the user in a preactivated state. The device thus avoids the need for the handler to discharge the air within the device as is found to be the case with almost all of the apparatus of the above discussed prior art. Additional ease of operation is provided herein in that this device permits one hand operation freeing the handler to attend to additional matters while simultaneously activating the device. The device is also designed to permit alternatively either total discard or evacuation of the collected fluids and reactivation for subsequent reuse.
The device of the present invention comprises a substantially rigid housing with an end wall and an opening therein at one end. Inside the housing is an airtight and watertight passive container which collects the body fluids. A means is provided to allow communication between the inside of the container and outside of the housing. Means are also provided to maintain the passive container adjacent the housing cover thus allowing a substantially fixed point from which the passive container expands. Other means provide for the collapsing and retaining of the passive container in a collapsed state. A means external to the passive container provides for automatic expansion of the passive container to provide a constant rate of suction. In one preferred embodiment of the present invention, the container is a passive bellows and the expansion means is a constant force spring assembly.
Accordingly, it is an object of the present invention to provide a wound suction device having an expansion means that provides a substantially constant suction over the range of evacuation.
Another object of the present invention is to provide a wound suction device that may be used easily with one hand.
Still another object of the present invention is to provide a wound suction device that is inexpensive and totally discardable.
Yet still another object is to provide a wound suction device that may be evacuated and reactivated for subsequent reuse.
A further object of the present invention is to provide a device that is available to a user in a preactivated or compressed state.
FIG. 1 is a perspective view of a device according to the present invention.
FIG. 2 is an exploded view of the device according to FIG. 1.
FIG. 3 is a cutaway perspective view of a base of the device of FIG. 2 housing a constant force spring assembly.
FIG. 4a is a bottom view of the base of FIG. 3.
FIG. 4b is a cross-sectional view of the base of assembly FIG. 3 along line 4--4 in FIG. 2.
FIG. 5 is a cross-sectional view of the device of FIG. 1 along line 4--4 in FIG. 2.
FIG. 6a is a perspective view of a slide collar of the device of FIG. 5.
FIG. 6b is a top view of the slide collar of FIG. 6a.
FIG. 1 illustrates the device 10 according to the present invention.
FIG. 2 provides an exploded view of the device 10 thereby illustrating the component parts and structure and assembly thereof. Generally the device 10 comprises a substantially rigid cylindrical housing 20 closed at one end and open at the other end, and enclosing in sequence from the closed end a passive bellows 80, a slide collar 60, and a base 40 which is fixedly mated to the open end of the housing 20. A constant force spring assembly 50 is housed within the base 40. Both the spring assembly 50 and the bellows 80 are attached to the slide collar 60. Thereby the bellows 80 is expanded or kept constantly expanding by the force provided by the spring assembly 50.
The cylindrical housing 20 of the device 10 has an integral end wall 22 at the closed end thereof. The end wall 22 has an opening 24 which permits communication with the bellows 80 within the housing 20. Two slits 26 are oppositely positioned in the side walls of the housing 20 and are substantially aligned with the longitudinal axis of the housing 20. The slits 26 run from a first position to a second position coincident with the other end of the housing 20. The slits 26 at the first position include cutouts 28 as shown in FIGS. 1 and 2. The purpose of these slits 26 and cutouts 28 will be made evident in the description to follow. The other end of the housing 20 terminates integrally in a ring 29 which has an outside diameter less than the outside diameter of the housing 20.
The bellows 80 has an upper end 85 and a lower end 86. The bellows 80 is disposed within the housing with its upper end 85 adjacent the end wall 22. The bellows 80 has ribbed portions 84 which permit the bellows 80 to expand to increase the volumetric capacity. At the lower end 86 of the bellows 80 is an appendage 88 with opening 89 therein which, as will be described herein, permits coupling of the bellows 80 with the remaining structure within the housing 20. Integral with the upper end 85 and communicating with the interior of the bellows 80 is a tube 81. The tube 81 has circumferentially thereon a flange 82 having a frusto-conical shape which permits passage of the tube 81 through the end wall opening 24 but which does not allow the tube 81 to be withdrawn back through the opening 24. In this manner the upper end 85 of the bellows 80 is kept adjacent the end wall 22. The bellows 80 can then be expanded downward toward the other end of the housing 20.
Also enclosed within the housing 20 is a slide collar 60 which comprises an annular ring 61 that slides within the housing 20. The longitudinal axis of annular ring 61 is aligned with that of the housing 20. The slide collar 60 includes a cross bar 64 which is disposed along one diameter of the annular ring 61 in slots (not shown) and extends beyond that one diameter to opposite tabs 62. These tabs 62 when engaged within the slits 26 protrude beyond the side walls of the housing 20. The tabs 62 are shaved at an angle toward the base 40 and are ribbed so that they may be easily engaged by the fingers. The cross bar 64 also has a first opening 65 and a second opening (not shown) positioned beneath the first opening 65. By means of opening 65, the bellows 80 can be attached to the slide collar 60. After aligning opening 89 of the bellows 80 with opening 65, a bolt 67 passes through bellows opening 85, cross bar opening 65, and is engaged by nut 66 thereby attaching bellows 80 to the slide collar 60.
The base 40 comprises a generally cylindrical side wall structure 41 with an integral top end wall 42. The top end wall 42 has a generally rectangular opening 44 and on its underside are two generally parallelly positioned ribs 46 as shown in FIGS. 3, 4a and 4b, and a curved wedge 48 particularly shown in FIG. 4b disposed between the ends of the ribs 46 and adjacent one end of generally rectangular opening 44. The spring assembly 50 comprises a coiled spring 51 a portion of which is confined in the space defined by the ribs 46 as shown in FIG. 3. The coiled portion 51 of the spring assembly 50 is housed between the ribs 46 and against wedge 48. The uncoiled or free portion 52 of the spring assembly 50 extends through the top end wall 42 opening 44 and terminates at a free end 54 with an opening (not shown) thereat which permits the attachment of the spring assembly 50 to the slide collar 60 after alignment with the second opening (not shown) of the cross bar 64 via a bolt and nut (both not shown) in the same manner that the bellows 80 was attached to the slide collar 60. The top end wall 42 has circumferentially within its upper edge a groove 43 adapted to matably receive the housing ring 29. A bottom end wall 49 seals the spring assembly 50 within the base 40.
After the bellows 80 and base 40 are each attached as described above to the slide collar 60, assembly of the device 10 is accomplished by inserting the above-mentioned components bellows 80 first within the housing 20 so that the tabs 62 are slidably engaged within slits 26. The bellows 80 is inserted tube 81 first. Once the flange 82 of the tube 81 passes through the end wall opening 24, the tube 81 cannot pass back into the housing 20. In addition, the upper end of the bellows 80 is thus kept adjacent the cover 22 and thus serves as a fixed point from which the bellows 80 will expand by means of the force of spring assembly 50. Assembly is completed when the base 40 is fixed to the housing 20 by means of fitting housing ring 29 into base groove 43. When the tabs 62 are advanced to the first position or slide into the housing 20 as far as possible, a slight twisting of the tabs 62 into the cutouts 28, dimensioned to at least partially receive the tabs 62 as shown in FIG. 2, will lock the tabs 62 in place and expand spring assembly 50. The first position of the slits 26 is preferably chosen so that the bellows 80 is fully collapsed when the tabs 62 are so engaged within the cutouts 28. Upon release of the tabs 62 from their locked positions within cutouts 28, the force of the spring assembly 50 will automatically draw the slide collar 60 without further action on the part of the user, toward the other end of the housing 20 and will simultaneously automatically expand the bellows 80 which is attached to the slide collar 60 thereby producing a negative pressure within the bellows 80.
FIG. 5, a vertical cross-sectional view along line 4--4 in FIG. 2, illustrates the positions of the component structure of the device 10 when the bellows 80 is in a collapsed or preactivated state within the housing 20.
FIGS. 6a and 6b illustrate another embodiment of the slide collar 60. The cross bar 74 has a recessed portion 75 which is of suitable dimension to receive the free end 54 of the spring assembly 50. Within the recessed portion 75 is a knob 78 having a shank 76 and enlarged head 77. Both the bellows 80 and spring free end 54 are attached to the slide collar 60 by engaging the bellows opening 89 and the spring free end opening (not shown) over the enlarged head 77 and onto the shank 76 as shown in FIG. 5. The force supplied by the spring assembly 50 maintains the bellows 80 and the spring free end 54 fixed to the slide collar 60.
The assembly of the device 10 may be finalized by the gluing of separate structural components together. Preferably the ring 29 of the other end of housing 20 is glued into the groove 43 on the upperside of the integral base top end wall 42. The cross bar 64 can also be glued into its cutouts (not shown) in the annular ring 61 of slide collar 60. Alternatively, the end walls 22 and 42, and tube 81 can be constructed separately from their respective structures and glued thereto. In a preferred assembly, ultrasonic welding is used to attach the ring 29 to the groove 43. Assembly of the device 10 is not limited to gluing and ultrasonic welding but may be accomplished by other conventional means known and appreciated by those skilled in such arts including but not limited to mechanical means such as screws.
Preferably the housing 20 will be made of a transparent plastic and will have etched graduations 110 thereon as shown in FIG. 1. The user will therefore be able to visibly inspect the proper operation of the device 10. Additionally he can measure the amount of fluid collected upon holding the housing 20 bellows upper end 85 down and comparing the fluid level against the graduations 110.
The device 10 can be supplied with a protective wrapping (not shown) to prevent inadvertent release of tabs 62 from their locked position while the bellows 80 is in a preactivated state. The wrapping can also serve to provide user or other desired information printed thereon. Additionally information may be imprinted on the base bottom end wall 49 by the user to identify the patient and other pertinent data relevant to the particular usage of the device 10.
Preferably the bellows 80, base 40, slide collar 60, and tube 81 are made by injection molding of suitable plastic material.
Advantageously the housing 20 is substantially rigid thereby avoiding the possibility of inadvertent compression by a patient rolling over onto the device 10.
The bellows tube 81 can be supplied, if desired, with a cap 130 as shown in FIG. 1. The cap 130 preferably is connected via a strand 132 attached to the circumference of ring 134 which is adapted to be press fitted over flange 82. In this fashion, the tube 81 can be covered at its outer end by the cap to maintain a sterile environment within the bellows 80 prior to use or to seal the fluids within the bellows 80 after use.
The bellows tube 81 can alternatively be supplied, if desired, with a conventional Y connector (not shown). One duct of the conventional Y connector would be connected to the tube 81. Another duct would be connected via conventional urethane tubing to an additional Y connector (not shown) which would also be connected to wound tubing (not shown) having one end disposed in the wound. A conventional non-return valve (not shown) could be inserted at some position along the tubing configuration. The remaining duct of the Y connector connected to the tube 81 would have a cap (not shown) arranged similarly to cap 130, strand 132, and ring 134 for tube 81. Normally this cap would seal the remaining duct while fluid is drawn from the wound through the conventional tubing and into the bellows 80. However, removal of the cap on the remaining Y connector duct permits evacuation of fluid from the bellows 80 by advancing the tabs 62 toward the first position thereby compressing the bellows 80. Thus the fluid may be totally evacuated from the bellows 80 which then is ready for continued use or reuse, or partially into another container to permit immediate transfer of the fluid for study, e.g. in a pathology laboratory, without disrupting the hookup of the device 10 for further removal of fluid from a wound.
The device 10 is designed to permit either discarding of the entire unit, or reuse after evacuating the fluids therein. After complete evacuation, the bellows 80 may be cleansed and reactivated for further use, if desired, by means of the Y connector via the discharge procedure described above.
In the preferred embodiment, the spring assembly 50 is a constant force ribbon spring as shown in FIG. 3 and as disclosed and illustrated in U.S. Pat. Nos. 2,609,191; 2,609,192; and 2,609,193. Such springs are desirable since they have the feature of providing a nearly constant force throughout their range of expansion. In this manner, the device 10 provides a substantially constant suction throughout the range of evacuation, a feature found by most doctors to be quite desirable in such devices. Furthermore, such spring assemblies render the device 10 insensitive to orientation so that no particular orientation of the device 10 is required.
Other embodiments are possible with a constant force means different from the spring assembly 50 as disclosed. Moreover, different structure configurations are possible without departing from the scope of the invention, e.g., the bellows 80 may be fixed at the other end of the housing 20 and permitted to expand towards the housing end wall 22 with suitable tubing and force means.
|Cited Patent||Filing date||Publication date||Applicant||Title|
|US799297 *||Apr 3, 1905||Sep 12, 1905||Joseph F Betzler||Fountain-pen.|
|US2074223 *||Nov 5, 1935||Mar 16, 1937||Fred T Horiuchi||Blood transfusion apparatus|
|US2595493 *||Sep 9, 1949||May 6, 1952||Le Roy K Mills||Liquid extracting apparatus|
|US3115138 *||Jul 14, 1960||Dec 24, 1963||Mcelvenny||Evacuator|
|US3376868 *||Jun 4, 1964||Apr 9, 1968||Howe Sound Co||Surgical evacuator device|
|US3416431 *||Mar 30, 1967||Dec 17, 1968||Mc Graw Edison Co||Negator spring toaster|
|US3421662 *||Mar 7, 1967||Jan 14, 1969||Hanson Ralph W||Dispersal device|
|US3680560 *||Nov 26, 1968||Aug 1, 1972||Voys Inc Le||Vacuum drainage collecting apparatus with disposable liner|
|US3742952 *||Apr 28, 1971||Jul 3, 1973||Alpha Ind Inc||Surgical suction pump assembly|
|US3774611 *||Jun 8, 1972||Nov 27, 1973||Oakes G||Stabilized contamination free surgical evacuator|
|US3779243 *||Oct 15, 1971||Dec 18, 1973||Carr H||Contamination free surgical evacuator|
|US3809086 *||Dec 9, 1971||May 7, 1974||Sherwood Medical Ind Inc||Wound drainage device|
|US3809087 *||May 17, 1973||May 7, 1974||Lewis R||Closed wound suction apparatus having biased plate members|
|US3871554 *||Feb 4, 1974||Mar 18, 1975||Sybron Corp||Eye wash station|
|US3875941 *||Apr 3, 1974||Apr 8, 1975||Medical Dynamics Inc||System for evacuating fluids from the body|
|US4136802 *||Sep 21, 1977||Jan 30, 1979||The Continental Group, Inc.||Spray dispenser with spring biased flexible container|
|CA879849A *||Aug 31, 1971||Protek Ag.||Automatic suction drains|
|Citing Patent||Filing date||Publication date||Applicant||Title|
|US4311141 *||Mar 13, 1980||Jan 19, 1982||Diamond Arthur D||Breast pump|
|US4404924 *||Sep 5, 1980||Sep 20, 1983||Uresil Company||Body fluid suction device indicators|
|US4460354 *||Jul 8, 1980||Jul 17, 1984||Snyder Laboratories, Inc.||Closed wound suction evacuator|
|US4525167 *||Aug 4, 1983||Jun 25, 1985||Goldberg Edward M||Body fluid suction device|
|US4529402 *||Feb 21, 1984||Jul 16, 1985||Snyder Laboratories, Inc.||Closed wound suction evacuator with rotary valve|
|US4559035 *||Sep 26, 1983||Dec 17, 1985||Snyder Laboratories, Inc.||Collapsible wound suction evacuator|
|US4578060 *||Jul 20, 1983||Mar 25, 1986||Howmedica, Inc.||Wound drainage device|
|US4636197 *||Feb 15, 1985||Jan 13, 1987||Ping Chu||Intravenous fluid infusion device|
|US4648870 *||May 10, 1985||Mar 10, 1987||Goldberg Edward M||Device for evacuating body fluids|
|US4681571 *||Apr 23, 1981||Jul 21, 1987||C. R. Bard, Inc.||Suction canister with disposable liner and check valve|
|US4740202 *||Aug 8, 1986||Apr 26, 1988||Haemonetics Corporation||Suction collection device|
|US5019059 *||Aug 2, 1988||May 28, 1991||Uresil Corporation||Apparatus and method for collecting body fluids|
|US5149325 *||Feb 25, 1991||Sep 22, 1992||Baxter International Inc.||Vacuum system for auto transfusion device|
|US5223228 *||Feb 25, 1991||Jun 29, 1993||Baxter International Inc.||Tray for autotransfusion module|
|US5382244 *||Feb 25, 1991||Jan 17, 1995||Baxter International Inc.||Stand alone control module|
|US5695124 *||May 30, 1996||Dec 9, 1997||Fendall Company||Self-contained emergency eye wash station|
|US7855024||Dec 27, 2006||Dec 21, 2010||Proton Energy Systems, Inc.||Compartmentalized storage tank for electrochemical cell system|
|US8007257 *||Oct 15, 2007||Aug 30, 2011||Kci Licensing Inc.||Reduced pressure delivery system having a manually-activated pump for providing treatment to low-severity wounds|
|US8007491||Sep 19, 2008||Aug 30, 2011||The Board Of Trustees Of The Leland Stanford Junior University||Methods for application of reduced pressure therapy|
|US8025650||Jun 12, 2007||Sep 27, 2011||Wound Care Technologies, Inc.||Negative pressure wound treatment device, and methods|
|US8128607 *||Apr 14, 2010||Mar 6, 2012||Spiracur Inc.||Devices and methods for treatment of damaged tissue|
|US8162908 *||Sep 26, 2011||Apr 24, 2012||Spiracur Inc.||Devices for treatment of damaged tissue|
|US8162965||Sep 9, 2009||Apr 24, 2012||Tyco Healthcare Group Lp||Low profile cutting assembly with a return spring|
|US8177764 *||Feb 17, 2009||May 15, 2012||Spiracur Inc.||Devices and methods for treatment of damaged tissue|
|US8246590||Oct 10, 2008||Aug 21, 2012||Spiracur, Inc.||Closed incision negative pressure wound therapy device and methods of use|
|US8287507||May 1, 2009||Oct 16, 2012||Kci Licensing, Inc.||Reduced pressure indicator for a reduced pressure source|
|US8337474||Apr 14, 2010||Dec 25, 2012||Spiracur Inc.||Devices and methods for treatment of damaged tissue|
|US8361043||Jan 7, 2010||Jan 29, 2013||Spiracur Inc.||Reduced pressure therapy of the sacral region|
|US8366693||Jun 18, 2010||Feb 5, 2013||Spiracur, Inc.||Methods and devices for applying closed incision negative pressure wound therapy|
|US8398604||Sep 26, 2011||Mar 19, 2013||Spiracur, Inc.||Methods and devices for applying closed incision negative pressure wound therapy|
|US8409159||Apr 9, 2010||Apr 2, 2013||Spiracur, Inc.||Methods and devices for applying closed incision negative pressure wound therapy|
|US8435221||Jun 18, 2010||May 7, 2013||Spiracur, Inc.||Closed incision negative pressure wound therapy device and methods of use|
|US8444614||Sep 24, 2010||May 21, 2013||Spiracur, Inc.||Methods and devices for applying closed incision negative pressure wound therapy|
|US8491617||Apr 23, 2012||Jul 23, 2013||Covidien Lp||Low profile cutting assembly with a return spring|
|US8529532||Sep 26, 2011||Sep 10, 2013||The Board Of Trustees Of The Leland Stanford Junior University||Reduced pressure therapy devices|
|US8535283||Feb 8, 2008||Sep 17, 2013||Kci Licensing, Inc.||System and method for applying reduced pressure at a tissue site|
|US8562576||Jan 11, 2011||Oct 22, 2013||Spiracur, Inc.||Closed incision negative pressure wound therapy device and methods of use|
|US8641692||Jul 9, 2009||Feb 4, 2014||Kci Licensing, Inc.||Manually-actuated, reduced-pressure systems for treating wounds|
|US8679079 *||Aug 9, 2011||Mar 25, 2014||Kci Licensing, Inc.||Reduced pressure delivery system having a manually-activated pump for providing treatment to low-severity wounds|
|US8728045||Mar 4, 2010||May 20, 2014||Spiracur Inc.||Devices and methods to apply alternating level of reduced pressure to tissue|
|US8728046 *||Sep 26, 2011||May 20, 2014||Spiracur Inc.||Controlled negative pressure apparatus and alarm mechanism|
|US8753322||Aug 10, 2011||Jun 17, 2014||Spiracur Inc.||Controlled negative pressure apparatus and alarm mechanism|
|US8795246||Jul 1, 2011||Aug 5, 2014||Spiracur Inc.||Alarm system|
|US8834434||Oct 18, 2013||Sep 16, 2014||Spiracur Inc.||Closed incision negative pressure wound therapy device and methods of use|
|US8834451||Jan 31, 2012||Sep 16, 2014||Smith & Nephew Plc||In-situ wound cleansing apparatus|
|US8858516 *||Sep 26, 2011||Oct 14, 2014||Spiracur Inc.||Controlled negative pressure apparatus and absorbency mechanism|
|US8864748||May 1, 2009||Oct 21, 2014||Kci Licensing, Inc.||Manually-actuated reduced pressure treatment system having regulated pressure capabilities|
|US8926575||Sep 13, 2012||Jan 6, 2015||Spiracur Inc.||Devices and methods for treatment of damaged tissue|
|US8961481||Feb 3, 2011||Feb 24, 2015||Spiracur Inc.||Devices and methods for treatment of damaged tissue|
|US8992492||Sep 27, 2011||Mar 31, 2015||Wound Care Technologies, Inc.||Negative pressure wound treatment device, and methods|
|US8998866||Jul 1, 2011||Apr 7, 2015||Smith & Nephew Plc||Provision of wound filler|
|US9039678||Apr 3, 2009||May 26, 2015||The Brigham And Women's Hospital, Inc||Micromechanical force devices for wound healing acceleration|
|US9044234||Mar 1, 2013||Jun 2, 2015||Spiracur Inc.||Methods and devices for applying closed incision negative pressure wound therapy|
|US9198801||May 24, 2013||Dec 1, 2015||Bluesky Medical Group, Inc.||Flexible reduced pressure treatment appliance|
|US9205001||Sep 12, 2014||Dec 8, 2015||Smith & Nephew Plc||Apparatus for aspirating, irrigating and cleansing wounds|
|US9259358||Jan 28, 2013||Feb 16, 2016||Kci Licensing, Inc.||Reduced pressure therapy of the sacral region|
|US9283307||Mar 31, 2011||Mar 15, 2016||Kci Licensing, Inc.||Devices and methods for treatment of damaged tissue|
|US9345822||Feb 27, 2013||May 24, 2016||Kci Licensing, Inc.||Methods and devices for applying closed incision negative pressure wound therapy|
|US9387126||Sep 26, 2012||Jul 12, 2016||Smith & Nephew Plc||Apparatus for aspirating, irrigating and cleansing wounds|
|US9421133||Sep 12, 2014||Aug 23, 2016||Kci Licensing, Inc.||Closed incision negative pressure wound therapy device and methods of use|
|US9561312||Feb 4, 2014||Feb 7, 2017||Kci Licensing, Inc.||Reduced pressure delivery system having a manually-activated pump for providing treatment to low-severity wounds|
|US9579430 *||May 1, 2014||Feb 28, 2017||Kci Licensing, Inc.||Controlled negative pressure apparatus and alarm mechanism|
|US20070270764 *||Nov 1, 2004||Nov 22, 2007||Gordon Angus B||Closed Sterile Drainage Device|
|US20080004549 *||Jun 12, 2007||Jan 3, 2008||Anderson Paul J||Negative pressure wound treatment device, and methods|
|US20080108977 *||Oct 15, 2007||May 8, 2008||Heaton Keith P|
|US20080160382 *||Dec 27, 2006||Jul 3, 2008||Proton Energy Systems, Inc.||Compartmentalized storage tank for electrochemical cell system|
|US20080200905 *||Feb 8, 2008||Aug 21, 2008||Keith Patrick Heaton||System and method for applying reduced pressure at a tissue site|
|US20090012482 *||Mar 13, 2008||Jan 8, 2009||Pinto Moshe||Devices and methods for application of reduced pressure therapy|
|US20090076467 *||Sep 19, 2008||Mar 19, 2009||Pinto Moshe||Methods for application of reduced pressure therapy|
|US20090254066 *||May 1, 2009||Oct 8, 2009||Keith Patrick Heaton||Reduced pressure indicator for a reduced pressure source|
|US20090259203 *||Oct 10, 2008||Oct 15, 2009||Dean Hu||Closed incision negative pressure wound therapy device and methods of use|
|US20090270843 *||Apr 3, 2009||Oct 29, 2009||The Brigham And Women's Hospital, Inc.||Micromechanical force devices for wound healing acceleration|
|US20090275922 *||May 1, 2009||Nov 5, 2009||Richard Daniel John Coulthard||Manually-actuated reduced pressure treatment system having regulated pressure capabilities|
|US20100030166 *||Jul 9, 2009||Feb 4, 2010||Aidan Marcus Tout||Manually-actuated, reduced-pressure systems for treating wounds|
|US20100042021 *||Feb 17, 2009||Feb 18, 2010||Spiracur, Inc.||Devices and methods for treatment of damaged tissue|
|US20100137775 *||Nov 25, 2009||Jun 3, 2010||Spiracur Inc.||Device for delivery of reduced pressure to body surfaces|
|US20100160901 *||Dec 23, 2009||Jun 24, 2010||Dean Hu||Device for delivery of reduced pressure to body surfaces|
|US20100174250 *||Jan 7, 2010||Jul 8, 2010||Spiracur Inc.||Reduced pressure therapy of the sacral region|
|US20100198173 *||Apr 14, 2010||Aug 5, 2010||Spiracur, Inc.||Devices and methods for treatment of damaged tissue|
|US20100198174 *||Apr 14, 2010||Aug 5, 2010||Spiracur, Inc.||Devices and methods for treatment of damaged tissue|
|US20100228205 *||Mar 4, 2010||Sep 9, 2010||Spiracur Inc.||Devices and methods to apply alternating level of reduced pressure to tissue|
|US20110015594 *||Jun 18, 2010||Jan 20, 2011||Dean Hu||Methods and Devices for Applying Closed Incision Negative Pressure Wound Therapy|
|US20110060356 *||Sep 9, 2009||Mar 10, 2011||Tyco Healthcare Group Lp||Low Profile Cutting Assembly with a Return Spring|
|US20110105963 *||Jan 11, 2011||May 5, 2011||Spiracur, Inc.||Closed incision negative pressure wound therapy device and methods of use|
|US20110106026 *||Sep 24, 2010||May 5, 2011||Kenneth Wu||Methods and devices for applying closed incision negative pressure wound therapy|
|US20110130691 *||Feb 3, 2011||Jun 2, 2011||Dean Hu||Devices and methods for treatment of damaged tissue|
|US20110137270 *||Feb 17, 2011||Jun 9, 2011||Dean Hu||Pressure indicator|
|US20110295220 *||Aug 9, 2011||Dec 1, 2011||Keith Patrick Heaton|
|US20120022475 *||Sep 26, 2011||Jan 26, 2012||Spiracur Inc.||Devices for treatment of damaged tissue|
|US20120071845 *||Sep 26, 2011||Mar 22, 2012||Dean Hu||Controlled negative pressure apparatus and absorbency mechanism|
|US20120083754 *||Sep 26, 2011||Apr 5, 2012||Dean Hu||Controlled negative pressure apparatus and alarm mechanism|
|US20140243767 *||May 1, 2014||Aug 28, 2014||Spiracur Inc.||Controlled negative pressure apparatus and alarm mechanism|
|CN101547712B||Oct 15, 2007||May 30, 2012||凯希特许有限公司|
|CN102006895B||Feb 13, 2009||Jul 2, 2014||斯皮拉克尔公司||Devices and methods for treatment of damaged tissue|
|CN102626529B *||Oct 15, 2007||Oct 28, 2015||凯希特许有限公司||用于对低严重性伤口提供治疗的、具有手动启动泵的减压输送系统|
|CN102908681A *||Aug 2, 2011||Feb 6, 2013||常熟市赛爱斯医疗器材有限公司||Negative-pressure drainage device with metering function|
|CN103285435B *||Feb 8, 2008||Mar 30, 2016||凯希特许有限公司||用于在组织部位施加减压的系统和方法|
|CN103721304A *||Dec 26, 2013||Apr 16, 2014||关淑萍||Negative pressure drainage device|
|CN103721304B *||Dec 26, 2013||Jun 1, 2016||关淑萍||一种负压引流装置|
|EP2114483B1||Feb 8, 2008||Oct 7, 2015||KCI Licensing, Inc.||System for applying reduced pressure at a tissue site|
|EP2340063B1||Jul 9, 2009||Aug 19, 2015||KCI Licensing, Inc.||Manually-actuated, reduced-pressure systems for treating wounds|
|EP2977068A1 *||Jul 9, 2009||Jan 27, 2016||KCI Licensing Inc.||Manually-actuated, reduced-pressure systems for treating wounds|
|WO2008048527A3 *||Oct 15, 2007||Jul 24, 2008||Kci Licensing Inc|
|WO2010102146A1 *||Mar 4, 2010||Sep 10, 2010||Spiracur Inc.||Devices and methods to apply alternating level of reduced pressure to tissue|
|WO2017040021A1||Aug 16, 2016||Mar 9, 2017||Kci Licensing, Inc.||Reduced pressure tissue therapy device|
|U.S. Classification||604/134, 600/579, 600/573, 141/67, 141/26, 417/328, 417/472|
|Oct 18, 1985||AS||Assignment|
Owner name: PFIZER HOSPITAL PRODUCTS GROUP, INC. A CORP OF DE.
Free format text: CHANGE OF NAME;ASSIGNOR:HOWMEDICA, INC.;REEL/FRAME:004470/0386
Effective date: 19840626
|Jan 27, 1993||AS||Assignment|
Owner name: HOWMEDICA, INC., NEW YORK
Free format text: CHANGE OF NAME;ASSIGNOR:PFIZER HOSPITAL PRODUCTS GROUP, INC.;REEL/FRAME:006325/0764
Effective date: 19920709
|Mar 8, 1999||AS||Assignment|
Owner name: STRYKER TECHNOLOGIES CORPORATION, MICHIGAN
Free format text: ASSIGNMENT OF ASSIGNORS INTEREST;ASSIGNOR:HOWMEDICA INC.;REEL/FRAME:009781/0191
Effective date: 19981202
|Mar 22, 1999||AS||Assignment|
Owner name: BANK OF AMERICA NATIONAL TRUST AND SAVINGS ASSOCIA
Free format text: SECURITY INTEREST;ASSIGNORS:STRYKER CORPORATION;STRYKER FAR EAST, INC.;STRYKER INTERNATIONAL INC.;AND OTHERS;REEL/FRAME:009817/0001
Effective date: 19981204
Owner name: BANK OF AMERICA NATIONAL TRUST AND SAVINGS ASSOCIA
Free format text: SECURITY AGREEMENT;ASSIGNORS:STRYKER CORPORATION;STRYKER FAR EAST, INC.;REEL/FRAME:014137/0212
Effective date: 19981204