|Publication number||US4686695 A|
|Application number||US 06/207,351|
|Publication date||Aug 11, 1987|
|Filing date||Nov 17, 1980|
|Priority date||Feb 5, 1979|
|Publication number||06207351, 207351, US 4686695 A, US 4686695A, US-A-4686695, US4686695 A, US4686695A|
|Original Assignee||Board Of Trustees Of The Leland Stanford Junior University|
|Export Citation||BiBTeX, EndNote, RefMan|
|Patent Citations (10), Non-Patent Citations (4), Referenced by (208), Classifications (30), Legal Events (1)|
|External Links: USPTO, USPTO Assignment, Espacenet|
This is a continuation of application Ser. No. 009,484 filed Feb. 5, 1979, now abandoned.
1. Field of the Invention
This invention relates to x-ray projection imaging systems. In a primary application the invention relates to obtaining projection x-ray images of a specific material in a volume of the body. In another application the invention relates to obtaining isolated images of blood vessels without bone and soft tissue using non-invasive injections of iodinated contrast material.
2. Description of Prior Art
At present images of blood vessels are made using invasive procedures where a catheter is inserted into the vessel and large amounts of an iodinated contrast material is inserted. The large amount is necessary to sufficiently attenuate x-rays so that the vessels will be visible. There have been a number of efforts to provide greater sensitivity to the iodine so that the vessels can be imaged with a non-invasive injection of iodine into a peripheral vein. Thusfar these have been relatively unsuccessful. In general much of the efforts at obtaining improved sensitivity to iodinated contrast agents concentrated on utilizing the unique energy spectral characteristics of iodine compared to those of body tissues.
The earliest approaches used relatively slow techniques which required long exposure times. These long exposure times are inadequate for imaging vessels which exhibit significant motion, as many do. Some of these early approaches used low-power monoenergetic sources on either side of the K edge of iodine. A system of this type using mechanically scanned x-ray beams and mechanical analog computers is described in Vol. VI of the Advances in Biological and Medical Physics published by the Academic Press in the chapter by B. Jacobson and R. Stuart Mackay on "Radiological Contrast Enhancement Methods." The section labelled "Dichromography," from page 224 to 231 describes a system using a special x-ray source with two fluorescent secondary emitters which alternately generated two monochromatic x-ray beams having energies on either side of the iodine K absorption edge. At each point wedge shaped materials of known composition are translated across the beam until the output beam reaches its predetermined value. The thickness of the wedges is then a direct indication of the amounts of the particular material present. A similar approach is described by B. Jacobson in the American Journal of Roentgenology, Vol. 91, January 1964, entitled "X-ray Spectrophometry in Vivo." In this paper a similar secondary emissive source alternately produced three monoenergetic wavelengths. Wedges of a soft tissue equivalent, bone and iodine were used in a mechanical analog computer to determine the amount of these body materials at each point in the scan.
Although these systems gave interesting results they suffered from using low power sources which required a long time to create an image. Normal heart and respiratory motions during the scanning time resulted in a blurred, low-resolution image. The low power of the sources was due to the use of monoenergetic beams.
At present, sufficient x-ray power for rapid exposures can only be derived from x-ray tubes. There is an early brief reference to the use of x-ray tube sources in an energy selective system in Appendix II of "Television X-ray Movies: Dose and Contrast Factors" by R. S. Mackay in the IRE Transactions on Medical Electronics, Vol. ME-7, No. 2, Apr. 1960, pp. 80-86. In this paper a heterochromatic source is used to produce different x-ray wavelengths. The various transmitted energies are detected using a scintillation detector and pulse height analyzer. A scanning beam is used, thus analyzing the tissue one point at a time. Since pulse height analyzers must isolate single photon events, this system is very slow. The long scan time makes it impractical for radiographic studies.
A number of studies have been made of isolating the iodinated contrast material by filtering an x-ray tube source on either side of the K edge of iodine. One such system is described in U.S. Pat. No. 3,854,049 issued to Charles A. Mistretta. In this system the x-ray beam is alternately filtered by a rotating filter wheel. Each filter produces a relatively narrow spectrum on either side of the K absorption edge of iodine. The television fluoroscopic images resulting from each filter are subtracted to provide an isolated image of the iodine.
These systems have thusfar been unable to provide non-invasive images of small vessels such as the coronary arteries for a number of reasons. Imaging detectors such as television-fluoroscopic systems or screen-film systems contain significant amounts of additive noise over and above the fundamental quantum noise due to the captured x-ray photons. These additive noise components obscure the detection of small difference signals as would be encountered with small vessels. Also, heavily filtered quasi-monoenergetic beams are relatively low in power, thus reducing the desired signal intensity. In addition, x-ray energies in the vicinity of the iodine K edge of 33 kev have relatively low transmission through the body since the attenuation coefficient of tissue is relatively high in this region. This further reduces the intensity of the desired signals as compared to the noise.
In general, imaging detectors, which simultaneously acquire the entire x-ray image, have a number of serious problems in systems attempting to isolate small amounts of iodine. As previously indicated, they have additive noise problems. Also, these systems are subject to significant amounts of scattered radiation since the entire volume is being irradiated simultaneously. Each imaging point thus receives scattered radiation from the entire volume. This scatter is reduced by the scatter-reducing grids, but it remains a significant noise source. In addition, imaging detectors are faced with fundamental tradeoffs between resolution and capture efficiency. If thick screens are used the capture efficiency of photons will be high, but the increased spreading of the light results in poor resolution. Conversely, thin screens produce high resolution and low capture efficiency. It is for these reasons that the present invention utilizes non-imaging detectors where a subsection of the image is acquired at each time.
The use of filtered spectra on either side of the K edge results in some error because of the finite width of the energy spectrum. These errors are particularly large in the case of bone whose attenuation coefficient varies significantly in this region. These systems can be compensated for by the use of a third energy as described in a publication by F. Kelcz, C. A. Mistretta and S. J. Riederer, "Spectral Considerations for Absorption Edge Fluoroscopy" in Medical Physics, Vol. 4, pp. 26-35, 1977. The output from this third energy can compensate for the bone errors, but the other problems, such as those due to the use of imaging detectors, remain.
A generalized system providing isolated images of specific materials is described in U.S. Pat. No. 3,848,130 issued to Albert Macovski. In this system transmission measurements are made at various energies and processed to provide isolated material images. Methods are shown for processing signals obtained from broad energy spectra so that narrow band filtering is not required. However, this system uses imaging detectors, including television fluoroscopy and film, so that the previously described problems will prevent the non-invasive visualization of small vessels.
In general all of the systems in the prior art used relatively low energies for the isolation of iodine in an attempt to take advantage of the large changes in the iodine attenuation coefficient. However, these lower energies present significant problems as was previously mentioned. The relatively low transmission of the body reduces the detected signals and increases the quantum noise. Also, at these lower energies, the mass attenuation coefficient of bone and soft tissue are quite different. This difference makes it difficult to provide an isolated iodine image in the presence of both soft tissue and bone.
In general, in the diagnostic region of the energy spectrum, the attenuation coefficients of all materials can be decomposed into a photo-electric component and a Compton scattering component. In U.S. Pat. No. 4,029,963 issued to R. E. Alvarez and A. Macovski, measurements are made at two energies spectra with the resultant signals processed to produce images of the photoelectric and Compton-scattering components. This system is used for both single projection images and for cross-sectional images using computerized tomography. In the single projection images, however, imaging detectors are used with all of the attendant problems of limited performance previously described. This patent also describes the concept of energy-sensitive integrating detectors. A dual detector is shown which simultaneously provides separate measurements at high and low energy spectra.
To obtain the desired accuracy it is important to obtain efficient, low-noise x-ray measurements which are essentially free of scatter. These are presently obtained in all of the commercial systems of computerized tomography. The scatter is not a problem since a relatively small region is irradiated at any one time using either a pencil beam or a sheet beam. The attendant collimation at the detector all but eliminates the scatter. The detectors used are individual high-efficiency detectors arranged in arrays. The thickness of the detectors determines the efficiency and the lateral dimensions determines the resolution. These are relatively independent since the detectors are isolated and provide separate measurements. Most important, the detectors are essentially free of additive noise. The relatively large electrical signals produced by either photo-multipliers or Xenon detectors are limited solely by quantum noise so that the accuracy is determined solely by the amount of radiation.
These accurate detectors are beginning to be used for single projection radiography. In the Microdose system produced by American Science and Engineering and in the Scout View System produced by General Electric the computerized tomography instrument is modified to provide a projection image. The body is indexed through the fan beam with the detector array providing a sequence of line information to produce a transmission image. Initially these transmission images were designed to identify the anatomy so as to facilitate the choice of the desired level for the cross-sectional image. Because of the detection accuracy, however, they have provided high-quality transmission images which have diagnostic value in their own right. These images, however, are not capable of non-invasive visualization of small vessels because of the intervening tissue. These are single energy spectra systems which are not capable of obtaining isolated images of specific materials.
An object of the invention is to provide a method for producing projection images of a specific material in a volume of the body. A further object of the invention is to provide images of a specific material with improved accuracy to enable non-invasive visualization of blood vessels. A further object of the invention is to provide a method of producing selective x-ray images which are essentially free of scattered radiation. A further object of the invention is to provide a method of producing selective x-ray images of iodinated contrast agents which are immune to the presence of intervening bone.
Briefly, in accordance with the invention, an x-ray beam is formed from an x-ray tube source which irradiates a subsection of the volume being studied. The beam is successively scanned through each subsection of the volume. At each subsection the x-ray transmission at two energy spectra is measured using integrating detectors. The resultant measurements are processed to produce a signal representing the projected amount of a specific material in the volume. The measurements are made at energy spectra where the mass attenuation coefficients of bone and soft tissue are substantially equal. A measurement at a third energy can be made to minimize the effect of intervening bone.
For a more complete disclosure of the invention, reference may be made to the following detailed description of several illustrative embodiments thereof which is given in conjunction with the accompanying drawings, of which:
FIG. 1 is a block diagram of an embodiment of the invention using a scanning x-ray pencil beam and a changing anode voltage supply;
FIG. 1(a) is a schematic representation of a Nipkow disc for forming a scanning x-ray pencil beam;
FIG. 2 is a graph indicating the mass attenuation coefficients of iodine, bone and soft tissue versus energy;
FIG. 3 is a block diagram of an embodiment of the invention using an array of scanning pencil beams and a changing anode supply providing three energy spectra;
FIG. 4 is a block diagram of an embodiment of the invention using a scanning sheet beam;
FIG. 4(a) is a schematic representation of an embodiment using a sheet beam and a filter wheel for changing energy;
FIG. 5 is a schematic representation of an embodiment of the invention using oscillated multiple sheet beams;
FIG. 6 is a schematic representation of an embodiment of the invention using an energy-sensitive detector;
FIG. 7 is a block diagram of an arrangement for controlled contrast enhancement; and
FIG. 8 is a schematic representation of an embodiment of the invention employing two views.
An understanding of the broad aspects of the invention may best be had by reference to FIG. 1 of the drawings. A projection image 25 is made of the iodine in a volume of the subject 10. An x-ray tube source 11 is used having an anode 13 and a cathode 12. The x-ray beam emitted from anode 13 is collimated into a pencil beam 15 by aperture 36. This aperture is one of the apertures of a Nipkow disk 14, a portion of which is shown in FIG. 1(a). These disks were used in the early days of mechanical television. They represent a simple method of providing a scanned light or x-ray beam. In the case of an x-ray beam the disk must be opaque to x-rays and thus use a high absorption material such as lead. As the disk rotates, the line scan is successively provided by different apertures at different radii. Thus apertures 35, 36 and 37 each supply a single line of a total raster scan which scans the volume of interest in subject 10.
The scanned x-ray beam is applied to subject 10 having an iodinated region 27. The emerging x-ray beam, after passing through aperture 29 of a second Nipkow disc collimator 16 is applied to an integrating x-ray detector 17. This detector 17 consists of scintillating crystal 18 and photomultiplier 19. This is an integrating detector which provides an output which represents the integrated number of photons transmitted through the body. Many energy selective systems employ pulse height analyzers which measure the energy of each photon. These systems are limited to relatively slow photon rates and thus result in very long imaging times. These long imaging times are usually unsuitable for most radiographic studies. It is for the same reasons that computerized tomography systems use integrating detectors and not pulse height analyzers.
In the system of FIG. 1 the energy change is affected by changing the anode voltage 26 on anode 13 of x-ray tube 11. The anode voltage, at each position of scanning beam 15 is changed between V1 and V2. These voltages are chosen so as to change the effective energy of beam 15 in a manner such that the iodine content can be calculated. FIG. 2 is a graph of the mass attenuation coefficients of iodine, bone, and soft tissue as a function of energy. As is shown, above approximately 100 kev the mass attenuation coefficients of bone and soft tissue are approximately equal and slow varying. The iodine mass attenuation coefficient, on the other hand, is varying relatively rapidly. By making two measurements in the energy range where bone and soft tissue have comparable mass attenuation coefficients the iodine content can be measured. This is shown by the equations
ln I01 I/1 =μt1 Zt +μI1 ZI
ln I02 /I2 =μt2 Zt +μI2 ZI
where I01 and I02 are the incident intensities at the two energies, I1 and I2 are the transmitted intensities, μt1 and μt2 are the mass attenuation coefficients of tissue at the two energies, μI1 and μI2 are the mass attenuation coefficients of iodine, and Zt and ZI are the equivalent path lengths of the tissue and iodine materials in gms/cm2, the length multiplied by the density.
To obtain an expression proportional to the iodine content ZI a weighted difference is taken of the logs of the normalized intensities with the first equation weighted by μt2 and the second by μt1 giving
ZI (μI1 μt2 -μI2 μt1)=μt2 ln I01 /I1 -μt1 ln I02 /I2
Thus a weighted difference of the logs of the two measurements provides a signal proportional to ZI. The logs of the incident intensities I01 and I02 merely provide offsetting constant values.
To perform this operation a sample and hold circuit 20 is used to store the two measured outputs 21 and 22 of integrating detector 17 when the anode voltage generator 26 is changed from V1 to V2. Thus output 21 repesents the transmitted intensity at the average energy corresponding to anode voltage V2. The lower average energy can be between 80 and 120 kev representing the energy region where bone and soft tissue have substantially equal mass attenuation coefficients as shown in FIG. 2. The higher average energy can be any value above about 120 kev where the mass attenuation coefficient of iodine has been reduced substantially below that of the lower average energy.
Logarithmic processor 23 is used to take the log of the two outputs 21 and 22. A weighted difference is taken of the logs of the measured transmitted intensities in 24 to form the processed projection signal 28 representing the projected amount of iodine in the volume of the subject 10. This signal is applied to display 25 which is raster scanned synchronously with the Nipkow disc 14. An isolated image of the iodinated region 27 is obtained.
In the prior art measurements of iodine content were made exclusively at lower energies, usually on either side of the k absorption edge. This was done to insure an adequate signal due to the large changes of μI in the lower energy regions. However, in this same region the attenuation coefficient of tissue μt is relatively high resulting in greater attenuation of the incident beam with its associated quantum noise. Thus at higher energies, the reduced iodine signal is more than compensated for by reduced quantum noise. In addition, the higher energies have an immunity to the presence of bone because of the substantial equality of the soft tissue and bone mass attenuation coefficients. Also, x-ray sources are considerably more efficient at the higher energies. These higher energies cannot profitably be used with imaging detectors because their inherent additive noise overcomes the advantage of reduced quantum noise. Thus the use of higher energies was not considered by previous investigators and they were therefore faced with difficult problems of removing the bone image from the separated iodine image.
In order to take full advantage of the fact that only a portion of the volume is being irradiated at one time the detector 17 must receive photons only from the scanning beam 15 and attenuate scattered photons. In FIG. 1 this is accomplished by a second Nipkow disk 16 which rotates synchronously with the beam forming disk 14 so that the apertures in disk 16 will always encompass the transmitted beam 15. Thus scatter from the subject 10 along the beam will be stopped by the opaque disk 16. An alternate approach not shown is the use of a scanning detector where detector 17 would have an aperture equal to the beam size and be mechanically scanned to follow the moving beam. Thus scattered photons would not be received by the detector. The wide area scintillator 18 is stationary and encompasses the imaging area. Thus it requires a moving collimator 16 for scatter elimination.
The system of FIG. 1 makes relatively inefficient use of the emitted photons from anode 13 since it uses a single pencil beam with the remainder of the emitted photons absorbed in the collimator 14. This results in relatively longer scan times and makes it difficult to study moving vessels such as the coronary arteries. This problem is reduced by the embodiment of FIG. 3 where the x-ray beam consists of an array of scanned pencil beams forming an array of contiguous rasters. In this way many more of the emitted source photons are used providing more rapid scans.
The photons emitted from anode 13 are collimated into an array of pencil beams using multiple aperture 40. One of the apertures 41 forms one of the pencil beams 15. The array of beams are projected through subject 10 and then through the second collimator 43. As in FIG. 1 this second collimator served to pass the desired array of pencil beams and attenuate the scattered radiation. For example, pencil beam 15 goes through aperture 44. Both multiple apertures 40 and 43 are synchronously scanned through small rasters using mechanical drivers 42 and 45 which provide the desired motions of the aperture plates 40 and 43. The transmitted array of pencil beams impinge on a detector system consisting of an array of scintillating crystals 46 and an array of photomultipliers 47. Each crystal photomultiplier detector element detects a small raster from one of the scanned beams.
As in FIG. 1 the anode voltage is changed with the detector outputs stored at each energy level. This could be done at two energy levels, as in FIG. 1, providing output signals 21 and 22 representing each energy level. For illustrative purposes, a three energy level system is shown where the anode voltage supply 26 is sequenced between three values; V1, V2 and a lower voltage V3. This lower energy is used to provide a more exact correction for bone for cases where it is required. For most systems the two energies resulting from V1 and V2 are sufficient for providing isolated iodine images in the presence of bone and soft tissue. However, as is seen in FIG. 2, in the vincinity of 80-100 kev there remains a small difference in the attenuation coefficients of bone and soft tissue. This is particularly true if the lower energy due to anode voltage V2 is reduced to about 60 kev. This might be done in some systems to provide a greater iodine signal by increasing the change in the iodine attenuation coefficient. At 60 kev the bone and soft tissue mass attenuation coefficients are different enough to cause some bone errors. This amounts to the presence of a residual bone image in the processed isolated iodine image.
If these bone errors are found to impair the diagnostic value of the image, they can be corrected by a third energy measurement due to anode voltage V3. In FIG. 3 a multiple scan converter storage system 48 is used to store the array of detector outputs at the three energy values. These three stored projection images are then sequentially read out forming projection signals 21, 22, and 49 representing the high, medium, and low energies. The low energy due to anode voltage V3 is chosen so as to have the greatest impact on the bone errors and minimize the effect on the iodine signal. A preferable energy is the region below the K edge of iodine where the iodine attenuation coefficient is reduced and the bone attenuation coefficient is relatively high. This energy region can be reached by using an appropriate anode voltage V3 in the vicinity of 40-50 kv. Alternatively a small amount of iodine filtering can be used as the beam exits from the x-ray tube. This filtering will tend to concentrate the energy below the iodine K edge and it will have a negligible affect at the higher anode voltages V1 and V2.
To process the three signals, linear equations are again formed using the normalized logarithms of the transmitted intensities as given by
ln I01 /I1 =μt1 Zt +μI1 ZI +μb1 Zb
ln I02 /I2 =μt2 Zt +μI2 ZI +μb2 Zb
ln I03 /I3 =μt3 Zt +μI3 ZI +μb3 Zb
where μb is the mass attenuation coefficient of bone and Zb is the effective path length. The three equations in three unknowns can be solved for Zt, ZI and Zb using standard linear algebra techniques or matrix inversion. The equation for the amount of iodine ZI will again be a weighted combination of the logs of the three measured intensities appropriately normalized. Thus the processing involves taking the logs of the intensities using logarithmic device 23 and taking weighted combinations of the logs using 50 to provide the processed projection signal 28 which is displayed in 25. In both FIG. 1 and FIG. 3 the logarithm and weighted combining would normally be done using a digital computer or using special purpose digital hardware. Analog devices can be used where less accuracy can be tolerated.
The mechanical motions of FIG. 3 can be relatively complex. A more straightforward multiple detector system is shown in FIGS. 4 and 4(a). Here the x-ray output from anode 13 is collimated into a fan shaped sheet beam 15. This configuration is similar to that used in the General Electric Scout View system of projection radiography using a computed tomography instrument. FIGS. 4 and 4(a) represent two views of the configuration. The x-ray output from anode 13 of x-ray tube 11 is collimated into a fan-shaped sheet beam 15 using collimator 55. This collimator consists of an opaque region with a slit opening forming the sheet beam 15. The beam is projected through subject 10 and again collimated using a second collimator 56. This second collimator 56 allows sheet beam 15 to pass and it attenuates scattered radiation from the subject 10. The beam is then detected by detector array 57 which provides an array of measurement signals 58 along the fan-shaped sheet beam.
In the configuration shown the entire system, x-ray tube 11, collimators 55 and 56 and detector 57 are mechanically connected together and scanned in a director normal to the plane of the sheet beam 15. This provides a projection image of a volume of the subject 10. The structure can be scanned using motor drive 61 and lead screw 62 to index sheet beam 15 along subject 10. Subject 10 remains stationary. In an alternative configurations, not shown, detector array 57 can consist of an array of elongated detector elements which span the entire image. In this alternate configuration detector 57 is not scanned but remains stationary as the collimated sheet beam 15 is scanned along the elongated elements. In this alternate configuration the second collimator 56 is particularly important in eliminating scatter since the elongated detector can receive scatter from the entire volume. Collimator 56 is less important in the configuration shown where the detector consists of a single scanned line and is relatively immune to scatter. Another, less desirable, mode of operation is to leave the beam 15 and detector 57 fixed and scan the patient 10. This is adequate for studies of relatively static organs but not for rapidly moving organs.
The energy modulation can take place exactly as indicated in FIG. 1. Anode supply 26 is indexed between two voltages. Or, the anode to cathode voltage can vary sinusoidally. One method of X-ray tube control is to utilize a tube with a control grid, and pulse the grid at times corresponding to different anode to cathode voltages. Storage system 59 stores the array of measurement signals 58 at each of the two energies to provide measurement signals 21 and 22 representing the high and low energy transmission. These can be processed exactly as in FIG. 1 to provide the projection signal 28 representing the specific material in the volume of interest.
In the system of FIG. 1, however, the processing system operates under the assumption that the high and low energy spectra can be approximated by monoenergetic spectra at the average energies. This has been found to be a good approximation, especially in the higher energy regions where μ is a slow function of energy. However, to be more accurate, the beam hardening affects can be taken into account. In that case the line intergrals, μz, are nonlinear functions of the log of the transmitted intensities. FIG. 4 shows a method of more accurate processing using nonlinear combiner 60 where weighted sums and differences are taken of nonlinear combinations of the log of the measurement signals. For example, a weighted difference can be taken of both the logs of measurement signals 21 and 22, as was done in FIG. 1, and an additional weighted difference can be included of the squares of the differences of logs of these signals. These squared components will be relatively small compared to the original components since the beam-hardening effect is small. In more complex nonlinear processing, higher order terms can be used including terms combining both measurement signals. These corrections, in general, are similar to the nonlinear beam-hardening corrections used in all commercial CT scanners.
FIG. 4(a) shows an alternate configuration for changing the energy of the beam 15. Instead of changing the anode voltage, the energy is changed by inserting a sequence of filters in the path of the x-ray beam. Rotating filter wheel 63 has an array of x-ray filters in the form of an annulus. These filters are sequenced in front of anode 13 to successively change the beam energy spectrum. For example, in a system using two energies, the filter wheel 63 would consist of alternating filters 53 and 54. Filter 53 can be the lower energy filter, such as tungsten, which has K edge at about 70 kev. Thus the energy spectrum would peak somewhat below 70 kev. Filter 54 can be the higher energy filter, such as uranium, which has a K edge at 115 kev. The K edge phenomenon need not be used for the filtering. Relatively lower atomic number materials can be used which eliminate the lower energies and thus increase the mean energy of the beam. Combinations of anode voltage changes and filter changes can be used to provide greater flexibility of the chosen energy spectra. For a system using three energies, as in FIG. 3, the filter wheel 63 would consist of three repetitive filter segments. The third segment could be iodine to create the desired mean energy below the iodine K edge for bone correction. The filter changes can be used with any of the previously described scanning systems such as those of FIGS. 1 and 3. As with the high voltage changes, storage system 58 stores the outputs at each energy spectrum forming measurement signals 21 and 22.
Often rapid repetitive scanning is desired for resolving motion. The configuration of FIG. 4 might be somewhat limited in its ability to provide a sequence of repeated rapid scans in the order of ten per second. The oscillating configuration of FIG. 5 is more suited to provide rapid repetitive scans. The x-ray tube source, not shown, remains stationary in the vicinity of the pivotal bearing 76. Ignoring detector 77 and its associated collimators, structure 75 contains source collimator 55, second collimator 56 and detector 57. This entire structure is rotated back and forth around pivot 76 to provide a sheet beam 15 which successively sweeps through the desired volume of subject 10. Signals 58 are processed exactly as previously described to provide the processed projection image signal. The energy of beam 15 can be changed in any of the previously described manners.
To provide more rapid studies it is desirable to both increase the collection efficiency of the emitted x-ray photons and to minimize the amount of mechanical movement required. This is shown in FIG. 5 where additional sheet beams and detectors are used to simultaneously collect measurement signals from a number of slices. For simplicity a single additional sheet beam 79 is shown in FIG. 5. This is formed using an additional aperture in source collimator 55 and in the second collimator 56. The transmitted photons for the second sheet beam 79 are collected in detector 77 to provide an array of detector output signals 78. Both the energy change mechanisms and the signal processing mechanisms for the system of FIG. 5 can be any of those previously described. Using multiple sheet beams, the structure 75 is oscillated over a smaller angle since each sheet beam scans a portion of the volume. Thus the scan time is decreased due to the smaller scanning volume for each sheet beam.
In the embodiments previously described measurements at different energy spectra were obtained by sequentially changing the energy of the x-ray beam. An alternate approach is shown in FIG. 6 where the two measurements are obtained simultaneously, thus reducing the scanning time. In FIG. 6 the beam 15, after passing through the subject 10, is detected using an energy sensitive integrating detector. This is in sharp distinction to energy sensitive detectors using pulse-height analysis. These latter detectors require the isolation of single photon events and are thus limited to relatively low values of x-ray intensities. These reduced intensities result in very long exposure intervals which are unsuitable for dynamic organs.
Integrating energy-sensitive detectors, as previously indicated, are described in U.S. Pat. No. 4,029,963. As shown in FIG. 6, beam 15 goes through two scintillating materials, 65 and 66. These can both be the same material, such as sodium iodide, or they can have different materials. In general the lower energy photons will give up their energy in the first portion 65 since these photons have a smaller mean free path. The higher energy photons with a longer mean free path will tend to give up their energy in the second scintillator 66.
Photomultiplier 67 receives the light from the first scintillator 65 to provide the lower energy measurement 22. Photomultiplier 68 receives the light from the second scintillator 66 to form the higher energy measurement 21. These measurement signals at the two energies, 21 and 22, are processed exactly as shown in FIG. 2 to provide the desired processed signal 28. Alternatively they can be processed as in FIG. 4 to correct for the nonlinearities. A thin separator which is opaque to light and passes x-rays can be placed between scintillator 65 and 66 to minimize cross talk.
Although FIG. 6 shows a single channel, an array of dual detectors can be used in the configuration of FIG. 4. In that case anode voltage changes are not required since arrays of measurements at each energy will be simultaneously available. In addition to scintillation detectors, ionization detectors using gases such as Xenon can be used. A split pickup electrode can be used with the front half receiving primarily lower-energy photons and the back half receiving primarily higher energy photons as described in the previously referenced U.S. Pat. No. 4,029,963.
The energy sensitive detector of FIG. 6 can also be used with the three-energy spectra system introduced in FIG. 3 where a lower energy measurement is used to aid in the removal of bone artifacts. An additional scintillator not shown, can be added on the x-ray beam side to receive the lower energy photons. It would be preferable if the materials used produced scintillations primarily below the iodine K edge so as to provide a measurement spectrum in that region. This third signal is used exactly as previously described in connection with FIG. 3. Also, the desired spectral measurements can be achieved through a combination of energy changes of the beam and energy sensitive detection.
Systems of computerized tomography achieve high contrast displays through the use of controlled windowing. The operator can control both the center value and the range of the displayed values so that subtle changes produce high contrast in the display. As shown in FIG. 7 a controlled windowing system 73 can also be used with this selective projection imaging system to provide a high contrast display. Processed signal 28 is appropriately amplified and clipped at the desired level before being displayed on display 25. This can aid, for example, in the visualization of coronary arteries.
In some cases, when an isolated iodine image 28 is formed, there exists an undesired iodinated background such as the heart chambers in the path of the coronary arteries. This background can be minimized by having the operator subtract portions of it before the windowing operation. The operator can manually adjust waveform generator 70 to eliminate the background from various selected regions of the coronary arteries while observing the display 25. Generator 70 produces signal 71 which is subtracted from the processed signal 28 in subtractor 74 to produce corrected signal 72. The subtractor can be a simple difference amplifier. Alternatively an automatic system can be used with a line following algorithm which follows the coronary artery and generates the required subtraction signal 71 at each point. Line-following algorithms have been widely published in the image processing literature.
In the systems described a single projection image of a selective material in a volume of the body was produced. In many diagnostic procedures, however, it is desirable to provide two views to help to define the anatomy in three dimensions. This can be accomplished using the system of FIG. 8 where two fan beam systems, of the type shown in FIG. 4, are simultaneously used to collect line projections. X-ray sources 13 and 80 produce sheet beams 15 and 81 with collimators not shown. These beams are projected through subject 10 and detected using line arrays 57 and 82 to provide measurement arrays 58 and 83. Any of the previously described energy sensitive systems can be used including sequential changing of the source of energy-sensitive detection. The measurements 58 and 83 are processed as previously described to produce two displays which represent the two views and help to identify the anatomy in space. Stereo viewers can be used by the operator.
The two fan beams and their detectors are simultaneously translated with respect to subject 10 to provide the projection images. For convenience it may be desirable if each beam intercepts a different plane in object 10 to avoid difficulties with the hardware. In the two displays, however, each line would be at the same level. For economy reasons a single beam can be used to acquire the two views in sequence. This can, however, produce severe motion artifacts.
Although iodinated contrast agents were used as examples, many other administered and natural body materials can be isolated using this system For example in chest radiography it is often desirable to visualize lesions, such as tumors, which underlie or overlie bone structures. If the system is used to separate bone and soft tissue, and thus provide separate images of each, this problem can be solved. Using lower energy measurements where bone and soft tissue have different attenuation coefficients a soft tissue only image can be formed to eliminate the bone and allow the visualization of the lesions.
Systems of this type require considerable accuracy in the measurements. It is therefore desirable to carefully monitor the source intensity for considerations of drift. An additional reference detector, as is used in CT systems, can be used to monitor the x-ray output is a region outside of the transmitted beam. A length of tissue-equivalent material can be used between the source and the reference detector to provide a standardized reference to aid in the calculations. Since it is known that the material contains no iodine, it can facilitate compensating for a variety of errors in the system by requiring that the processor indicate zero iodine content for the reference signal.
A variety of mechanical scanning systems have been shown for providing scanned x-ray beams having different configurations. These mechanical systems can be eliminated by using electronically scanned electron beams. These scanned electron beams, in special purpose x-ray tubes, can be scanned over anodes to provide an electronically movable x-ray source. This moving source, in connection with fixed collimators, provides a scanned x-ray beam without mechanical motion.
In the system using changing anode voltage, the anode voltage was stepped from one value to the next, as shown in FIGS. 1 and 3. This can be difficult to achieve if it is done rapidly. Since the mean energy is the important parameter being changed, it is not essential for the anode voltage to remain constant during the photon collection interval. For example, sinusoidal anode voltages can be used with the data for the higher energy collected during the positive half cycle and that for the lower energy collected during the negative half cycle. These sinusoidal anode voltages are much easier to generate. Another alternative involves grid control within the x-ray tube. A varying anode voltage can be used with the grid-controlled x-ray tube turned on during the desired voltage ranges. For example, using a sinusoidal anode voltage, the x-ray tube can be switched on during the region of the maximum and minimum values and turned off during the transition regions of the sine wave.
The temporal requirements for this system are different from conventional x-ray imaging systems using imaging detectors such as film or television fluoroscopy. With imaging detectors, when viewing moving structures such as the coronary arteries, it is essential that the exposure take place during a time interval of under 50 milliseconds where the structures are relatively stationary. In the systems described in this application each subsection of the volume is exposed in sequence. This results in a "focal-plane shutter" effect where each region is exposed at a different time. In this case it is essential that each subsection exposure take place during a time interval where the structures of interest are relatively stationary. The entire exposure, however, can take place over a time interval long as compared to the motion, as with the focal-plane shutter. The resultant image will not be blurred or smeared due to the motion since the structure is essentially stationary during each subsection exposure. The image will, however, be distorted in that different portions of the anatomy will be viewed at different times. For example, a vertical vessel moving horizontally will result in an image of a somewhat tilted vessel since each line will be exposed at a slightly different time. The diagnostic value of this skewed image of the vessel is not impaired since narrowing or stenosis of the vessel can be observed with good clarity since no blurring or smearing has taken place. If an imaging detector were used with a long exposure interval the image of the moving vessel would be significantly blurred by the motion, thus greatly reducing its diagnostic value. Using the configuration of FIG. 4 typical exposure times would be 1-10 milliseconds per line resulting in 0.2-5 seconds per total image. The short line exposure interval makes it excellent for cardiovascular visualization despite the long image time.
In this system only a subsection of the volume is irradiated at one time greatly reducing scatter and allowing the use of efficient non-imaging detectors. These detectors have negligible additive noise thus greatly increasing the sensitivity of the measurements. The unique combination of this x-ray imaging system with multiple-energy measurements allows the imaging of small amounts of specific materials. These detectors also facilitate the use of higher energies with their improved quantum statistics and immunity to bone. These properties are unique in that they allow the non-invasive imaging of small vessels such as the coronary arteries.
|Cited Patent||Filing date||Publication date||Applicant||Title|
|US2730566 *||Dec 27, 1949||Jan 10, 1956||Bartow Beacons Inc||Method and apparatus for x-ray fluoroscopy|
|US3101407 *||Apr 9, 1959||Aug 20, 1963||Jr John Daniel Shipman||Fluoroscope system utilizing an image storage tube|
|US3633029 *||Oct 27, 1970||Jan 4, 1972||Cgr Medical Corp||Pulsed x-ray control system with improved film darkening|
|US3848130 *||Jun 25, 1973||Nov 12, 1974||A Macovski||Selective material x-ray imaging system|
|US3866047 *||Apr 9, 1973||Feb 11, 1975||Emi Ltd||Penetrating radiation examining apparatus having a scanning collimator|
|US4029963 *||Jul 30, 1976||Jun 14, 1977||The Board Of Trustees Of Leland Stanford Junior University||X-ray spectral decomposition imaging system|
|US4096391 *||May 31, 1977||Jun 20, 1978||The Board Of Trustees Of The University Of Alabama||Method and apparatus for reduction of scatter in diagnostic radiology|
|US4196352 *||Apr 28, 1978||Apr 1, 1980||General Electric Company||Multiple purpose high speed tomographic x-ray scanner|
|US4247774 *||Jun 26, 1978||Jan 27, 1981||The United States Of America As Represented By The Department Of Health, Education And Welfare||Simultaneous dual-energy computer assisted tomography|
|US4260898 *||Sep 28, 1978||Apr 7, 1981||American Science And Engineering, Inc.||X-ray imaging variable resolution|
|1||"Radiological Contrast Enhancing Methods", Jacobson et al., Advances in Biological and Medical Physics, Academic Press, date unknown, pp. 201-260.|
|2||*||Radiological Contrast Enhancing Methods , Jacobson et al., Advances in Biological and Medical Physics, Academic Press, date unknown, pp. 201 260.|
|3||*||X Ray Spectrophotometry in Vivo , Jacobson, American Journal of Roentgenology, vol. 91, 1/64, pp. 202 210.|
|4||X-Ray Spectrophotometry in Vivo", Jacobson, American Journal of Roentgenology, vol. 91, 1/64, pp. 202-210.|
|Citing Patent||Filing date||Publication date||Applicant||Title|
|US4763345 *||Jul 3, 1986||Aug 9, 1988||The Regents Of The University Of California||Slit scanning and deteching system|
|US4780897 *||May 6, 1986||Oct 25, 1988||General Electric Company||Dual energy imaging with kinestatic charge detector|
|US4813061 *||Aug 5, 1987||Mar 14, 1989||Kabushiki Kaisha Toshiba||X-ray diagnosis method and apparatus|
|US4987584 *||Apr 3, 1990||Jan 22, 1991||Heiman Gmbh||Materials inspection system using x-ray imaging|
|US5020085 *||Apr 18, 1990||May 28, 1991||Matsushita Electric Industrial Co., Ltd.||X-ray image processing device|
|US5123037 *||Dec 14, 1990||Jun 16, 1992||General Electric Cgr S.A.||Method for calibrating the measuring system of an x-ray apparatus|
|US5132998 *||Mar 1, 1990||Jul 21, 1992||Matsushita Electric Industrial Co., Ltd.||Radiographic image processing method and photographic imaging apparatus therefor|
|US5204888 *||Dec 14, 1990||Apr 20, 1993||Aloka Co., Ltd.||Bone mineral content measuring apparatus|
|US5319547 *||Aug 10, 1990||Jun 7, 1994||Vivid Technologies, Inc.||Device and method for inspection of baggage and other objects|
|US5335255 *||Mar 24, 1992||Aug 2, 1994||Seppi Edward J||X-ray scanner with a source emitting plurality of fan beams|
|US5465284 *||Dec 5, 1990||Nov 7, 1995||University Of Massachusetts Medical Center||System for quantitative radiographic imaging|
|US5481584 *||Nov 23, 1994||Jan 2, 1996||Tang; Jihong||Device for material separation using nondestructive inspection imaging|
|US5485492 *||Mar 22, 1994||Jan 16, 1996||Lunar Corporation||Reduced field-of-view CT system for imaging compact embedded structures|
|US5490218 *||Dec 10, 1993||Feb 6, 1996||Vivid Technologies, Inc.||Device and method for inspection of baggage and other objects|
|US5511106 *||Jun 30, 1994||Apr 23, 1996||Siemens Aktiengesellschaft||X-ray diagnostics installation for producing x-ray exposures of body parts of a patient|
|US5533080 *||May 8, 1995||Jul 2, 1996||Lunar Corporation||Reduced field-of-view CT system for imaging compact embedded structures|
|US5838758 *||Mar 13, 1995||Nov 17, 1998||Vivid Technologies||Device and method for inspection of baggage and other objects|
|US5841832 *||Sep 26, 1997||Nov 24, 1998||Lunar Corporation||Dual-energy x-ray detector providing spatial and temporal interpolation|
|US5864146 *||Nov 13, 1996||Jan 26, 1999||University Of Massachusetts Medical Center||System for quantitative radiographic imaging|
|US6088423 *||Jun 5, 1998||Jul 11, 2000||Vivid Technologies, Inc.||Multiview x-ray based system for detecting contraband such as in baggage|
|US6195413||Jun 14, 1999||Feb 27, 2001||Heimann Systems Gmbh||Method and arrangement for detecting X-rays|
|US6445767||Sep 2, 1999||Sep 3, 2002||University Of Massachussetts Medical Center||System for quantitative radiographic imaging|
|US6507633||Feb 15, 2001||Jan 14, 2003||The Regents Of The University Of Michigan||Method for statistically reconstructing a polyenergetic X-ray computed tomography image and image reconstructor apparatus utilizing the method|
|US6717174||Jun 29, 2001||Apr 6, 2004||University Of Massachusetts Medical Center||System for quantitative radiographic imaging|
|US6891918 *||Nov 27, 2002||May 10, 2005||Ge Medical Systems Global Technology Company, Llc||Methods and apparatus for acquiring perfusion data|
|US6920397 *||Oct 1, 2003||Jul 19, 2005||Vialogy Corp.||Repeatable software-based active signal processing technique|
|US6922462 *||Jul 31, 2002||Jul 26, 2005||Ge Medical Systems Global Technology Company, Llc||Method, system and computer product for plaque characterization|
|US6963806 *||Aug 12, 2003||Nov 8, 2005||Vialogy Corp.||Technique for analyzing arrayed signals using quantum expressor functions|
|US6987833 *||Oct 16, 2003||Jan 17, 2006||General Electric Company||Methods and apparatus for identification and imaging of specific materials|
|US7027561 *||May 21, 2003||Apr 11, 2006||Xcounter Ab||Dual-energy scanning-based detection of ionizing radiation|
|US7031425 *||Nov 27, 2002||Apr 18, 2006||Ge Medical Systems Global Technology Company, Llc||Methods and apparatus for generating CT scout images|
|US7046756 *||May 20, 2003||May 16, 2006||General Electric Company||Rotatable filter for a pre-subject CT collimator having multiple filtering profiles|
|US7047136 *||Oct 1, 2003||May 16, 2006||Vialogy Corp.||Interferometric technique for quantitating biological indicators|
|US7050529 *||Jul 23, 2002||May 23, 2006||Ge Medical Systems Global Technolgy Company, Llc||Methods and apparatus for performing a computed tomography scan|
|US7120222 *||Jun 5, 2003||Oct 10, 2006||General Electric Company||CT imaging system with multiple peak x-ray source|
|US7158611 *||Aug 21, 2002||Jan 2, 2007||Siemens Aktiengesellschaft||Method for determining density distributions and atomic number distributions during radiographic examination methods|
|US7236572 *||May 28, 2004||Jun 26, 2007||Siemens Aktiengesellschaft||X-ray imaging apparatus with continuous, periodic movement of the radiation detector in the exposure plane|
|US7238951 *||Mar 24, 2003||Jul 3, 2007||Commissariat A L'energie Atomique||Two-dimensional ionising particle detector|
|US7260173 *||Feb 25, 2003||Aug 21, 2007||Hitachi, Ltd.||Apparatus and method for detecting threats|
|US7330531||May 11, 1995||Feb 12, 2008||University Of Massachusetts Medical Center||System for quantitative radiographic imaging|
|US7333590||Jun 7, 2006||Feb 19, 2008||Xcounter Ab||Dual-source scanning-based detection of ionizing radiation|
|US7356118 *||Oct 24, 2005||Apr 8, 2008||Scantech Holdings, Llc||Angled-beam detection system for container inspection|
|US7431500 *||Mar 31, 2004||Oct 7, 2008||Analogic Corporation||Dynamic exposure control in radiography|
|US7583779 *||Nov 24, 2004||Sep 1, 2009||General Electric Company||System and method for acquisition and reconstruction of contrast-enhanced, artifact-reduced CT images|
|US7599471 *||Oct 24, 2007||Oct 6, 2009||The Boeing Company||Method and apparatus for rotating an anode in an x-ray system|
|US7606349 *||Feb 9, 2007||Oct 20, 2009||L-3 Communications Security and Detection Systems Inc.||Selective generation of radiation at multiple energy levels|
|US7636413 *||Dec 22, 2009||General Electric Company||Method and apparatus of multi-energy imaging|
|US7660453||Aug 20, 2002||Feb 9, 2010||Imaging Therapeutics, Inc.||Methods and devices for analysis of x-ray images|
|US7672431 *||Oct 12, 2004||Mar 2, 2010||Bayer Schering Pharma Aktiengesellschaft||X-ray arrangement and x-ray contrast process for imaging an object under examination that contains at least one radiopaque element as well as use of the x-ray arrangement|
|US7688935 *||Aug 21, 2006||Mar 30, 2010||General Electric Company||Method and apparatus of multi-energy imaging|
|US7688936 *||Mar 30, 2010||General Electric Company||Method and apparatus of multi-energy imaging|
|US7697657 *||Mar 23, 2007||Apr 13, 2010||General Electric Company||System and method of density and effective atomic number imaging|
|US7702075 *||Apr 23, 2007||Apr 20, 2010||Tsinghu University||Energy spectrum modulation apparatus, material discrimination method and device, image processing method|
|US7734076||Dec 11, 2006||Jun 8, 2010||General Electric Company||Material decomposition image noise reduction|
|US7756239 *||Dec 7, 2006||Jul 13, 2010||General Electric Company||Diagnostic imaging two non K-edge basis materials plus N K-edge contrast agents|
|US7778382 *||Aug 17, 2010||General Electric Company||CT imaging system with multiple peak x-ray source|
|US7840247||Nov 23, 2010||Imatx, Inc.||Methods of predicting musculoskeletal disease|
|US7869560||Jan 11, 2011||Ge Medical Systems Global Technology Company, Llc||X-ray CT apparatus and image processing apparatus|
|US7885372||Dec 7, 2007||Feb 8, 2011||Morpho Detection, Inc.||System and method for energy sensitive computed tomography|
|US7889834 *||Feb 15, 2011||Siemens Aktiengesellschaft||Method for preparing reconstructed CT image data records and CT system|
|US7894569 *||Nov 14, 2008||Feb 22, 2011||Koninklijke Philips Electronics N.V.||Medical X-ray examination apparatus for performing K-edge imaging|
|US7983382 *||Nov 26, 2008||Jul 19, 2011||General Electric Company||System and method for material segmentation utilizing computed tomography scans|
|US7995822||Dec 31, 2009||Aug 9, 2011||Imatx, Inc.||Methods for the compensation of imaging technique in the processing of radiographic images|
|US8000441||Jul 20, 2009||Aug 16, 2011||Imatx, Inc.||Calibration devices and methods of use thereof|
|US8000766||Aug 16, 2011||Imatx, Inc.||Methods to diagnose treat and prevent bone loss|
|US8019044 *||Sep 13, 2011||Shkumat Nick A||Image acquisition for dual energy imaging|
|US8031836||Oct 4, 2011||Imatx, Inc.||Methods and devices for quantitative analysis of x-ray images|
|US8068580||Feb 2, 2009||Nov 29, 2011||Imatx, Inc.||Methods and devices for quantitative analysis of x-ray images|
|US8073521||Dec 6, 2011||Imatx, Inc.||Method for bone structure prognosis and simulated bone remodeling|
|US8111803||Apr 29, 2009||Feb 7, 2012||General Electric Company||Method for energy sensitive computed tomography using checkerboard filtering|
|US8180016||Aug 7, 2008||May 15, 2012||Kyoto University||X-ray CT apparatus and method thereof|
|US8213566 *||Jun 22, 2009||Jul 3, 2012||Koninklijke Philips Electronics N.V.||K-edge imaging|
|US8290564||Aug 28, 2007||Oct 16, 2012||Imatx, Inc.||Method for bone structure prognosis and simulated bone remodeling|
|US8306180 *||Nov 6, 2012||Tsinghua University||Image reconstruction method for high-energy, dual-energy CT system|
|US8311181 *||Nov 13, 2012||General Electric Company||Apparatus and method of visualizing multi-energy imaging data|
|US8311182 *||Sep 22, 2010||Nov 13, 2012||General Electric Company||System and method of notch filtration for dual energy CT|
|US8452068||Nov 2, 2011||May 28, 2013||Covidien Lp||Hybrid registration method|
|US8467589||Jun 18, 2013||Covidien Lp||Hybrid registration method|
|US8473032||Jun 2, 2009||Jun 25, 2013||Superdimension, Ltd.||Feature-based registration method|
|US8484000||Sep 2, 2005||Jul 9, 2013||Vialogy Llc||Detecting events of interest using quantum resonance interferometry|
|US8515147||Dec 11, 2007||Aug 20, 2013||Koninklijke Philips N.V.||Spectrally resolving X-ray imaging device|
|US8565858||Jun 23, 2009||Oct 22, 2013||Covidien Lp||Methods and systems for performing medical procedures with reference to determining estimated dispositions for actual dispositions of projective images to transform projective images into an image volume|
|US8571178 *||Sep 24, 2010||Oct 29, 2013||Fujifilm Corporation||Radiation imaging apparatus and imaging control device controlling a filter based on subject information|
|US8588365||Aug 15, 2011||Nov 19, 2013||Imatx, Inc.||Calibration devices and methods of use thereof|
|US8600124||Sep 16, 2005||Dec 3, 2013||Imatx, Inc.||System and method of predicting future fractures|
|US8625874||Dec 22, 2009||Jan 7, 2014||Imatx, Inc.||Methods and devices for analysis of x-ray images|
|US8639009||Mar 29, 2010||Jan 28, 2014||Imatx, Inc.||Methods and devices for evaluating and treating a bone condition based on x-ray image analysis|
|US8649481||Oct 3, 2011||Feb 11, 2014||Imatx, Inc.||Methods and devices for quantitative analysis of X-ray images|
|US8660239||Nov 16, 2009||Feb 25, 2014||Kromek Limited||Detector apparatus and method|
|US8686372 *||Nov 30, 2011||Apr 1, 2014||Ushio Denki Kabushiki Kaisha||Method for the spatially resolved measurement of parameters in a cross section of a beam bundle of high-energy radiation of high intensity|
|US8705688 *||Dec 19, 2011||Apr 22, 2014||Kabushiki Kaisha Toshiba||Medical image processing apparatus, X-ray computed tomography apparatus, and medical image processing method|
|US8781191||Aug 16, 2012||Jul 15, 2014||Imatx, Inc.||Methods for the compensation of imaging technique in the processing of radiographic images|
|US8792617 *||Mar 2, 2012||Jul 29, 2014||Siemens Aktiengesellschaft||Method and x-ray system to create a dual energy x-ray image|
|US8818484||Nov 17, 2010||Aug 26, 2014||Imatx, Inc.||Methods of predicting musculoskeletal disease|
|US8908829 *||Feb 23, 2011||Dec 9, 2014||Canon Kabushiki Kaisha||X-ray imaging apparatus and X-ray imaging method|
|US8913818 *||Jan 27, 2014||Dec 16, 2014||Imatx, Inc.||Methods and devices for evaluating and treating a bone condition based on X-ray image analysis|
|US8939917||Feb 16, 2010||Jan 27, 2015||Imatx, Inc.||Methods and devices for quantitative analysis of bone and cartilage|
|US8965075||May 13, 2010||Feb 24, 2015||Imatx, Inc.||System and method for predicting future fractures|
|US8965087||Dec 2, 2013||Feb 24, 2015||Imatx, Inc.||System and method of predicting future fractures|
|US9069092||Mar 6, 2012||Jun 30, 2015||L-3 Communication Security and Detection Systems Corp.||X-ray imager with sparse detector array|
|US9113839 *||Feb 22, 2011||Aug 25, 2015||Rapiscon Systems, Inc.||X-ray inspection system and method|
|US9117258||May 20, 2013||Aug 25, 2015||Covidien Lp||Feature-based registration method|
|US9155501||Jul 8, 2014||Oct 13, 2015||Imatx, Inc.||Methods for the compensation of imaging technique in the processing of radiographic images|
|US9267955||Aug 15, 2011||Feb 23, 2016||Imatx, Inc.||Methods to diagnose treat and prevent bone loss|
|US9271803||May 2, 2013||Mar 1, 2016||Covidien Lp||Hybrid registration method|
|US9275469||Dec 4, 2014||Mar 1, 2016||Imatx, Inc.||Methods and devices for evaluating and treating a bone condition on x-ray image analysis|
|US20020070365 *||Jun 29, 2001||Jun 13, 2002||University Of Massachusetts Medical Center||System for quantitative radiographic imaging|
|US20020193686 *||Jan 2, 2001||Dec 19, 2002||Pinhas Gilboa||Methods and systems for performing medical procedures with reference to projective image and with respect to pre-stored images|
|US20020196899 *||Aug 5, 2002||Dec 26, 2002||University Of Massachusetts Medical Center||System for quantitative radiographic imaging|
|US20030015208 *||May 28, 2002||Jan 23, 2003||Philipp Lang||Methods to diagnose treat and prevent bone loss|
|US20030112921 *||Aug 20, 2002||Jun 19, 2003||Philipp Lang||Methods and devices for analysis of x-ray images|
|US20030195416 *||Apr 16, 2002||Oct 16, 2003||Toth Thomas L.||Method and apparatus of multi-energy imaging|
|US20040017879 *||Jul 23, 2002||Jan 29, 2004||Hoffman David M.||Methods and apparatus for performing a computed tomography scan|
|US20040022359 *||Jul 31, 2002||Feb 5, 2004||Kishore Acharya||Method, system and computer product for plaque characterization|
|US20040063136 *||Oct 1, 2003||Apr 1, 2004||Sandeep Gulati||Repeatable software-based active signal processing technique|
|US20040064261 *||Aug 12, 2003||Apr 1, 2004||Sandeep Gulati||Technique for analyzing arrayed signals using quantum expressor functions|
|US20040072232 *||Oct 1, 2003||Apr 15, 2004||Sandeep Gulati||Interferometric technique for quantitating biological indicators|
|US20040101087 *||Nov 27, 2002||May 27, 2004||Jiang Hsieh||Methods and apparatus for generating CT scout images|
|US20040101090 *||Nov 27, 2002||May 27, 2004||Danielle Drummond||Methods and apparatus for acquiring perfusion data|
|US20040101097 *||Feb 25, 2003||May 27, 2004||Kyoichiro Wakayama||Apparatus and method for detecting threats|
|US20040111219 *||Jul 9, 2003||Jun 10, 2004||Sandeep Gulati||Active interferometric signal analysis in software|
|US20040120457 *||Dec 20, 2002||Jun 24, 2004||University Of Massachusetts Medical Center||Scatter reducing device for imaging|
|US20040223585 *||Aug 21, 2002||Nov 11, 2004||Bjorn Heismann||Method for determining density distributions and atomic number distributions during radiographic examination methods|
|US20040234021 *||May 20, 2003||Nov 25, 2004||Hoffman David M.||Rotatable filter for a pre-subject ct collimator having multiple filtering profiles|
|US20040234031 *||May 21, 2003||Nov 25, 2004||Tom Francke||Dual-energy scanning-based detection of ionizing radiation|
|US20040247082 *||Jun 5, 2003||Dec 9, 2004||Ge Medical Systems Global Technology Company, Llc||Ct imaging system with multiple peak x-ray source|
|US20050002486 *||May 28, 2004||Jan 6, 2005||Michael Maschke||X-ray imaging apparatus with continuous, periodic movement of the radiation detector in the exposure plane|
|US20050069086 *||Mar 31, 2004||Mar 31, 2005||Analogic Corporation||Dynamic exposure control in radiography|
|US20050084069 *||Oct 16, 2003||Apr 21, 2005||Yanfeng Du||Methods and apparatus for identification and imaging of specific materials|
|US20050123093 *||Oct 12, 2004||Jun 9, 2005||Ruediger Lawaczeck||X-ray arrangement and x-ray contrast process for imaging an object under examination that contains at least one radiopaque element as well as use of the x-ray arrangement|
|US20050148860 *||Sep 17, 2004||Jul 7, 2005||Imaging Therapeutics, Inc.||Method for bone structure prognosis and simulated bone remodeling|
|US20050161611 *||Mar 24, 2003||Jul 28, 2005||Commissariat A L'energie||Two-dimensional ionising particle detector|
|US20060013355 *||Jul 15, 2005||Jan 19, 2006||Bjoern Heismann||Method for the reconstruction of sectional images from detector measurement data of a tomography device|
|US20060053005 *||Sep 2, 2005||Mar 9, 2006||Sandeep Gulati||Detecting events of interest using quantum resonance interferometry|
|US20060062442 *||Sep 16, 2005||Mar 23, 2006||Imaging Therapeutics, Inc.||System and method of predicting future fractures|
|US20060109949 *||Nov 24, 2004||May 25, 2006||Tkaczyk J E||System and method for acquisition and reconstruction of contrast-enhanced, artifact-reduced ct images|
|US20060233302 *||Oct 24, 2005||Oct 19, 2006||Might Matthew B||Angled-beam detection system for container inspection|
|US20060280281 *||Jun 13, 2006||Dec 14, 2006||Thomas Flohr||Method for calculating absorber-specific weighting coefficients and method for improving a contrast-to-noise ratio, dependent on an absorber, in an x-ray image, produced by an x-ray machine, of an object to be examined|
|US20060280282 *||Aug 21, 2006||Dec 14, 2006||Toth Thomas L||Method and apparatus of multi-energy imaging|
|US20060285645 *||Aug 18, 2006||Dec 21, 2006||Hoffman David M||CT imaging system with multiple peak X-ray source|
|US20070047794 *||Aug 31, 2006||Mar 1, 2007||Philipp Lang||Methods and devices for analysis of x-ray images|
|US20070078336 *||Aug 21, 2006||Apr 5, 2007||Toth Thomas L||Method and apparatus of multi-energy imaging|
|US20070092056 *||Oct 16, 2006||Apr 26, 2007||Thomas Flohr||Method for generating CT displays in x ray computed tomography|
|US20070223653 *||Jun 7, 2006||Sep 27, 2007||Xcounter Ab||Dual-source scanning-based detection of ionizing radiation|
|US20070286329 *||Apr 23, 2007||Dec 13, 2007||Xuewu Wang||Energy spectrum modulation apparatus, material discrimination method and device, image processing method|
|US20080031412 *||Aug 28, 2007||Feb 7, 2008||Imaging Therapeutics, Inc.||Method for Bone Structure Prognosis and Simulated Bone Remodeling|
|US20080043917 *||Feb 9, 2007||Feb 21, 2008||L-3 Communications Security and Detection Systems Inc.||Selective generation of radiation at multiple energy levels|
|US20080049897 *||May 24, 2005||Feb 28, 2008||Molloy Janelle A||System and Method for Temporally Precise Intensity Modulated Radiation Therapy (Imrt)|
|US20080095303 *||Aug 17, 2005||Apr 24, 2008||Koninklijke Philips Electronics, N.V.||Apparatus For The Evaluation Of Rotational X-Ray Projections|
|US20080097794 *||Dec 10, 2007||Apr 24, 2008||Imaging Therapeutics, Inc.||System and Method for Building and Manipulating a Centralized Measurement Value Database|
|US20080135789 *||Dec 11, 2006||Jun 12, 2008||Yanfeng Du||Material decomposition image noise reduction|
|US20080137803 *||Dec 7, 2006||Jun 12, 2008||Xiaoye Wu||Diagnostic imaging two non k-edge basis materials plus n k-edge contrast agents|
|US20080273666 *||Mar 23, 2007||Nov 6, 2008||Deborah Joy Walter||System and method of density and effective atomic number imaging|
|US20090015119 *||Jul 11, 2007||Jan 15, 2009||Maytag Corp.||Outer support body for a drawer-type dishwasher|
|US20090041185 *||Mar 28, 2008||Feb 12, 2009||Scantech Holdings, Llc||Angled-beam detection system for container inspection|
|US20090052621 *||Aug 23, 2007||Feb 26, 2009||Deborah Joy Walter||Method and apparatus for basis material decomposition with k-edge materials|
|US20090080597 *||Sep 26, 2007||Mar 26, 2009||Samit Kumar Basu||System and method for performing material decomposition using an overdetermined system of equations|
|US20090110147 *||Oct 24, 2007||Apr 30, 2009||Morteza Safai||Method and apparatus for rotating an anode in an x-ray system|
|US20090122953 *||Nov 10, 2008||May 14, 2009||Yasuhiro Imai||X-ray ct apparatus and image processing apparatus|
|US20090147910 *||Dec 7, 2007||Jun 11, 2009||General Electric Company||System and method for energy sensitive computed tomography|
|US20090207966 *||Feb 13, 2009||Aug 20, 2009||Shkumat Nick A||Image acquisition for dual energy imaging|
|US20090207970 *||Feb 2, 2009||Aug 20, 2009||Imaging Therapeutics, Inc.||Methods and Devices for Quantitative Analysis of X-Ray Images|
|US20090257549 *||Apr 9, 2009||Oct 15, 2009||Bjorn Heismann||Method for preparing reconstructed CT image data records and CT system|
|US20090264729 *||Oct 22, 2009||Pinhas Gilboa||Methods And Systems For Performing Medical Procedures With Reference To Projective Image And With Respect To Pre Stored Images|
|US20100008558 *||Dec 11, 2007||Jan 14, 2010||Koninklijke Philips Electronics N. V.||Spectrally resolving x-ray imaging device|
|US20100040192 *||Jul 27, 2009||Feb 18, 2010||Tsinghua University||Image reconstruction method for high-energy, dual-energy ct system|
|US20100098212 *||Dec 30, 2009||Apr 22, 2010||Imaging Therapeutics, Inc.||Methods and Devices for Quantitative Analysis of X-Ray Images|
|US20100128844 *||Nov 26, 2008||May 27, 2010||Brian Thomsen||System and Method for Material Segmentation Utilizing Computed Tomography Scans|
|US20100135564 *||Dec 22, 2009||Jun 3, 2010||Brian William Thomsen||Apparatus for and method of selecting material triplets for a multi-material decomposition|
|US20100135565 *||Feb 1, 2010||Jun 3, 2010||Brian William Thomsen||Apparatus and method of visualizing multi-energy imaging data|
|US20100197639 *||Dec 22, 2009||Aug 5, 2010||Imaging Therapeutics, Inc.||Methods and Devices for Analysis of X-Ray Images|
|US20100223016 *||Aug 15, 2008||Sep 2, 2010||Gary Gibson||Method and apparatus for identification and detection of liquids|
|US20100278296 *||Apr 29, 2009||Nov 4, 2010||General Electric Company||Method for energy sensitive computed tomography using checkerboard filtering|
|US20100310035 *||Nov 14, 2008||Dec 9, 2010||Koninklijke Philips Electronics N.V.||Medical x-ray examination apparatus for performing k-edge imaging|
|US20100310036 *||Dec 9, 2010||General Electric Company||Computed tomography method and apparatus|
|US20110019891 *||Jul 20, 2010||Jan 27, 2011||Sylvie Puong||Medical imaging process for triple-energy modeling, and device for implementing such a process|
|US20110036360 *||Mar 29, 2010||Feb 17, 2011||Imaging Therapeutics, Inc.||Methods and Devices for Analysis of X-Ray Images|
|US20110075810 *||Sep 24, 2010||Mar 31, 2011||Fujifilm Corporation||Radiation imaging apparatus and imaging control device|
|US20110096905 *||Jun 22, 2009||Apr 28, 2011||Koninklijke Philips Electronics N.V.||K-edge imaging|
|US20110194668 *||Aug 7, 2008||Aug 11, 2011||Kyoto University||X-ray ct apparatus and method thereof|
|US20110211667 *||Sep 1, 2011||Abdelaziz Ikhlef||De-populated detector for computed tomography and method of making same|
|US20110222658 *||Nov 16, 2009||Sep 15, 2011||Kromek Limited||Detector Apparatus and Method|
|US20110243382 *||Oct 6, 2011||Edward James Morton||X-Ray Inspection System and Method|
|US20120069953 *||Sep 22, 2010||Mar 22, 2012||Naveen Chandra||System and method of notch filtration for dual energy ct|
|US20120093279 *||Apr 19, 2012||Toshiba Medical Systems Corporation||Medical image processing apparatus, x-ray computed tomography apparatus, and medical image processing method|
|US20120138805 *||Nov 30, 2011||Jun 7, 2012||Xtreme Technologies Gmbh||Method for the Spatially Resolved Measurement of Parameters in a Cross Section of a Beam Bundle of High-Energy Radiation of High Intensity|
|US20120224668 *||Mar 2, 2012||Sep 6, 2012||Lothar Baetz||Method and x-ray system to create a dual energy x-ray image|
|US20120321042 *||Feb 23, 2011||Dec 20, 2012||Canon Kabushiki Kaisha||X-ray imaging apparatus and x-ray imaging method|
|US20140126694 *||Jun 14, 2011||May 8, 2014||Daniel Abenaim||Security Scanner|
|CN1880949B||Jun 14, 2006||Dec 22, 2010||西门子公司||Method for calculating absorber-specific weighting coefficients and method for improving a contrast-to-noise ratio|
|CN101557762B||Dec 11, 2007||Jun 27, 2012||皇家飞利浦电子股份有限公司||Spectrally resolving x-ray imaging device|
|CN102215754B||Aug 7, 2008||Jun 5, 2013||国立大学法人京都大学||X-ray ct apparatus, and its method|
|DE19826062B4 *||Jun 12, 1998||Dec 14, 2006||Smiths Heimann Gmbh||Verfahren und Anordnung zur Detektion von Röntgenstrahlen|
|DE102005027436A1 *||Jun 14, 2005||Dec 28, 2006||Siemens Ag||Verfahren zur Berechnung von absorberspezifischen Gewichtungskoeffizienten und Verfahren zur Verbesserung eines von einem Absorber abhängigen Kontrast-zu-Rausch-Verhältnisses in einem von einer Röntgeneinrichtung erzeugten Röntgenbild eines zu untersuchenden Objektes|
|DE102005027436B4 *||Jun 14, 2005||Sep 4, 2008||Siemens Ag||Verfahren zur Berechnung von absorberspezifischen Gewichtungskoeffizienten und Verfahren zur Verbesserung eines von einem Absorber abhängigen Kontrast-zu-Rausch-Verhältnisses in einem von einer Röntgeneinrichtung erzeugten Röntgenbild eines zu untersuchenden Objektes|
|EP0385505A2 *||Mar 2, 1990||Sep 5, 1990||Matsushita Electric Industrial Co., Ltd.||Radiographic image processing method and photographic imaging apparatus therefor|
|EP0398029A1 *||Apr 18, 1990||Nov 22, 1990||Matsushita Electric Industrial Co., Ltd.||An X-ray image processing device|
|EP0434506A1 *||Dec 12, 1990||Jun 26, 1991||General Electric Cgr S.A.||Calibration method for the measuring chain of an X-ray apparatus|
|EP0549858A2 *||Oct 6, 1992||Jul 7, 1993||Matsushita Electric Industrial Co., Ltd.||Substance quantitative analysis method|
|EP2189115A1 *||Aug 7, 2008||May 26, 2010||Kyoto University||X-ray ct apparatus, and its method|
|EP2189115A4 *||Aug 7, 2008||Aug 18, 2010||Univ Kyoto||X-ray ct apparatus, and its method|
|WO1994024939A1 *||Apr 23, 1993||Nov 10, 1994||Lunar Corporation||Reduced field-of-view ct system for imaging compact embedded structures|
|WO2007108736A1 *||Mar 13, 2007||Sep 27, 2007||Xcounter Ab||Dual-source scanning-based detection of ionizing radiation|
|WO2008072175A1 *||Dec 11, 2007||Jun 19, 2008||Koninklijke Philips Electronics N.V.||Spectrally resolving x-ray imaging device|
|WO2009073284A1 *||Oct 20, 2008||Jun 11, 2009||General Electric Company||Filter with alternating pattern for use in energy sensitive computed tomography|
|WO2010001281A1||Jun 1, 2009||Jan 7, 2010||Koninklijke Philips Electronics N.V.||Spectral ct|
|WO2010058201A2 *||Nov 16, 2009||May 27, 2010||Durham Scientific Crystals Limited||Detector apparatus and method|
|WO2010058201A3 *||Nov 16, 2009||Feb 24, 2011||Kromek Limited||Detector apparatus and method|
|U.S. Classification||378/146, 378/5, 378/98.11, 348/E05.089|
|International Classification||H04N5/32, H05G1/60, G01N23/04, A61B6/03, G06T11/00|
|Cooperative Classification||A61B6/4291, H05G1/60, A61B6/4035, A61B6/4241, G01N23/043, G06T2211/408, H04N5/3205, G06T2211/412, A61B6/032, A61B6/405, A61B6/482, G06T11/005|
|European Classification||A61B6/42B8, A61B6/48D, A61B6/03B, A61B6/40H, A61B6/40F, G06T11/00T1, G01N23/04C, H04N5/32S, H05G1/60|
|May 1, 2002||AS||Assignment|
Owner name: COOPERSURGICAL ACQUISITION CORP., CALIFORNIA
Free format text: ASSIGNMENT OF ASSIGNORS INTEREST;ASSIGNOR:NORLAND MEDICAL SYSTEMS, INC.;REEL/FRAME:012841/0988
Effective date: 20020411