|Publication number||US6467113 B2|
|Application number||US 09/946,886|
|Publication date||Oct 22, 2002|
|Filing date||Sep 5, 2001|
|Priority date||Aug 25, 1997|
|Also published as||CA2301609A1, DE69808941D1, DE69808941T2, DE69841600D1, EP1011391A1, EP1011391B1, EP1234565A2, EP1234565A3, EP1234565B1, US6021533, US6295675, US6760939, US7111348, US7353556, US7398573, US20020029423, US20030019042, US20040261185, US20070011817, US20070017032, WO1999009865A1|
|Publication number||09946886, 946886, US 6467113 B2, US 6467113B2, US-B2-6467113, US6467113 B2, US6467113B2|
|Inventors||Craig D. Ellis, Kenith W. Chambers, Stephen E. Glover, Kerry J. Mensching|
|Original Assignee||Hill-Rom Services, Inc.|
|Export Citation||BiBTeX, EndNote, RefMan|
|Patent Citations (42), Non-Patent Citations (16), Referenced by (53), Classifications (17), Legal Events (5)|
|External Links: USPTO, USPTO Assignment, Espacenet|
This application is continuation of U.S. Application Ser. No. 09/465,872, filed on Dec. 16, 1999, now U.S. Pat. No. 6,295,675 which is a divisional of U.S. Application Ser. No. 08/917,145 filed on Aug. 25, 1997, now U.S. Pat. No. 6,021,533, the disclosures of which are expressly incorporated by reference herein.
The present invention relates to a mattress assembly for use on a hospital bed. More particularly, the present invention relates to a replacement mattress assembly which can be used on various types of bed frames to provide improved patient support and therapies.
According to present invention, a patient support apparatus is provided that includes a cover, at least one air bladder, an air supply, a valve, a valve control, and a tube. The cover has a top patient rest surface and a bottom surface and defines an interior region. The at least one air bladder is located in the interior region of the cover. The valve is configured to control the flow of air to the at least one air bladder. The valve control is configured to control operation of the valve. The tube has an interior region configured to conduct air supplied by the air supply to the valve. The patient support further includes an electrical cable coupled to the valve control and the valve. The electrical cable is located at least partially within the interior region of the tube.
According to another aspect of the invention, a patient support apparatus is provided including a cover, at least one air bladder, and an air supply. The cover defines an interior region. The at least one air bladder is located in the interior region. The patient support apparatus further includes a cloth tube configured to deliver air from the air supply to the at least one air bladder.
Additional features of the invention will become apparent to those skilled in the art upon consideration of the following detailed description of the preferred embodiment exemplifying the best mode of carrying out the invention as presently perceived.
The detailed description particularly refers to the accompanying figures in which:
FIG. 1 is an exploded perspective view of the mattress assembly of the present disclosure illustrating a plurality of air cushions, air tubes, and control valves located between top and bottom covers;
FIG. 2 is a diagrammatic view illustrating connection between the valves and the air cushions of the present disclosure;
FIG. 3 is an exploded perspective view illustrating a bottom cover and a plurality of low friction plastic transfer plates configured to be coupled to the bottom cover to facilitate transfer of the mattress assembly from one bed frame to another;
FIG. 4 is a perspective view illustrating a blower housing coupled to a foot board of a bed for supplying air to the mattress assembly;
FIG. 5 is a sectional view taken along lines 5—5 of FIG. 4 illustrating an air intake manifold coupled to the blower housing;
FIG. 6 is a partial sectional view illustrating a slot formed in the blower housing for receiving a corresponding pin formed on the air intake manifold;
FIG. 7 is a perspective view illustrating further details of the air intake manifold;
FIG. 8 is an exploded perspective view illustrating details of an air hose assembly extending between the blower housing and the mattress assembly which includes an internal electrical cord for transmitting control signals from the blower housing control panel to the mattress assembly;
FIG. 9 is a partial side elevation view illustrating a siderail of a bed and a siderail down sensor coupled to a frame below the siderail;
FIG. 10 is an enlarged side elevation view illustrating a switch of the siderail down sensor which is closed when the siderail is in its upwardly pivoted position;
FIG. 11 is a side elevation view similar to FIG. 10 illustrating the sensor switch in an open position when the siderail is pivoted downwardly;
FIG. 12 is a sectional view taken through another embodiment of the siderail down indicator which clips on a frame member of the siderail; and
FIG. 13 is a side elevation view of the siderail down indicator of FIG. 12.
Referring now to the drawings, FIG. 1 illustrates a mattress assembly 10 of the present disclosure. The mattress assembly 10 includes a bottom cover 12 having a bottom surface 14 and upwardly extending sidewall 16 surrounding bottom surface 14 to define an interior region 18. Straps 20 are coupled to bottom cover 12 for securing the mattress assembly 10 to a bed frame (not shown) if desired.
A plurality of air cushions are configured to be located within the interior region 18 of mattress assembly 10. A pair of rotation cushions 22 are located on bottom surface 14. Cushions 22 are stored in a normally deflated configuration on surface 14. Rotation cushions 22 are selectively inflated and deflated to control rotation therapy of a patient located on the mattress assembly 10.
The mattress assembly 10 includes a head end 24 and a foot end 26. A pair of proportional valve assemblies 28 and 30 are located in interior region 18 adjacent head end 24. A lower head cushion 32 is located within interior region 18 adjacent head end 24. Lower body cushions 34 and 36 are located in the interior region 18 spaced toward the foot end 26 from lower head bladder 32.
Transversely-extending support surface bladders 38 are located on top of bladders 32, 34, and 36 within a continuous interior volume of interior region 18. Support surface cushions 38 include a head cushion 40, a chest cushion 42, a seat cushion 44, and a foot cushion 46. Support cushions 40, 44, and 46 include inner bladder sections 48 and outer bladder sections 50 and 51 which are separately controllable from an air supply source as discussed below.
Air enters the mattress assembly 10 from a blower 52 of an air system through inlet 54. Inlet 54 is coupled to an inlet 55 of a percussion/vibration valve 56. Air supply through inlet 54 is also coupled to valves 28 and 30 via flexible, cloth tubes 58 and 60, respectively. Cloth tube 58 includes a first end 62 coupled to an outlet 57 of the manifold of valve 56 and a second end 64 coupled to a manifold inlet 66 of valve 28. Cloth tube 60 has a first end 68 coupled to an outlet 69 of the manifold of valve 56 and a second end 70 coupled to a manifold inlet 72 of valve 30 as shown in FIG. 2. A mesh tube liner is located within and extends the length of each of the cloth tubes 58 and 60 to permit a vacuum to be applied to the tubes 58 and 60 to deflate the air bladders rapidly as discussed below.
The cloth tubes 58 and 60 are illustratively two-inch diameter tubes which transfer air from the blower unit 52 to the valve assemblies 28 and 30. Cloth tubes 58 and 60 are very flexible and reduce the likelihood of kinking when moved or articulated with the mattress assembly 10 compared to conventional plastic tubes.
The mattress assembly 10 further includes width extension cushions 74, 76, 78, and 80 which are positioned outside bottom cover 12. Cushions 74 and 78 are located on opposite sides of the mattress assembly 10 near head end 24. Cushions 76 and 80 are located on opposite sides of the mattress assembly 10 near foot end 26. As best illustrated in FIG. 2, the width extension cushions 74, 76, 78, and 80 are all coupled together and coupled to a valve 82 of the air system located near foot end 26 of mattress assembly 10. Width extension cushions 74, 76, 78, and 80 are normally inflated during operation of the mattress assembly 10. However, valve 82 may be manually opened to release air from the width extension cushions 74, 76, 78, and 80 to permit the mattress assembly 10 to be moved to a narrower frame. In other words, when a wide frame is used, the width extension bladders 74, 76, 78, and 80 are inflated. Therefore, the mattress assembly 10 can be used to fit on frames having various widths without creating a gap between siderails of the frame and the edges of the mattress assembly 10. Typically, Med/Surg frames are wider frames. Critical care frames are typically narrower frames. Therefore, mattress assembly 10 can be used on both Med/Surg frames and critical care frames by manually opening and closing valve 82.
A top cover 84 is located all over the sidewall 16 of bottom cover 12. Top cover 84 is illustratively a washable cover. The remainder of the cushions, hoses, and bottom cover are wipeable for cleaning.
FIG. 2 illustrates air flow between the valves and various cushions of the mattress assembly 10. Rotation bladders 22 are coupled to valves 28 and 30 by air supply lines 88 and 90, respectively. Lower head cushion 32 is coupled to line 106 from valve 30. Lower body cushions 34 and 36 include internal bladders 94 and 96, respectively, which are each coupled to a supply line 92 from valve 30. When operation of the mattress assembly is initiated, air is supplied through supply line 92 to inflate the internal bladders 94 and 96 automatically to a predetermined pressure to reduce the likelihood that a patient will bottom out against a bed frame. Internal bladders 94 and 96 are surrounded by external bladders of lower body cushions 34 and 36. The external bladders of cushions 34 and 36 are coupled to outlets of valves 28 and 30 by supply lines 98 and 100, respectively. Therefore, external bladders of cushions 34 and 36 can be controlled by lines 98 and 100 while the internal bladders 94 and 96 remain inflated by supply line 92.
Central section 48 of head support surface cushion 40 is coupled to an. outlet of valve 28 by line 102. Opposite side sections 50 and 51 of head support surface cushion 40 are coupled to valves 28 and 30 by lines 104 and 106, respectively.
Chest support surface cushion 42 is coupled to valve 28 by line 108. Chest support surface cushion includes internal percussion/vibration (P/V) bladders 110, 112, and 114. P/V bladder 110 is coupled to a first outlet of P/V valve 56 by line 116. P/V bladder 112 is coupled to a second outlet of P/V valve 56 by line 118. P/V bladder 114 is coupled to a third outlet of P/V valve 56 by line 120.
Side portions 50 and 51 of seat support surface cushion 44 are coupled to lines 104 and 106 extending from valves 28 and 30, respectively. Central portion 48 of seat support surface cushion 44 is coupled to valve 30 by line 122.
Opposite side sections 50 and 51 of foot support surface cushion 46 are coupled to supply lines 104 and 106 of valves 28 and 30, respectively. Central section 48 of foot support surface cushion 46 is coupled to valve assembly 30 by supply line 124. Supply line 104 from valve 28 is also coupled to an inlet of valve 82. An outlet of valve 82 is coupled to width extension cushions 74, 76, 78, and 80 as discussed above. Outlet line 125 is a vent hose.
If it is desired to transport a bed with a patient on the mattress assembly 10, the valves 28 and 30 are actuated to deflate the inner sections 48 of cushions 40, 44, and 46 to a reduced pressure compared to outer sections 50 and 51. The outer sections 50 and 51 of cushions 40, 44, and 46 remain inflated. Cushions 34 and 35 remain inflated. This helps cradle the patient to maintain the patient on the mattress assembly 10 during transport of the bed.
Details of the valves 28, 30, and 56 are disclosed in U.S. Application Ser. No. 09/093,303, now U.S. Pat. No. 6,202,672 which is based on U.S. Provisional Application No. 60/056,763, the disclosure of which is incorporated herein by reference.
FIG. 3 illustrates a plurality of transfer plates 130 which are coupled to bottom surface 14 of bottom cover 12 to facilitate transfer of the mattress assembly 10 from one bed frame to another bed frame. Transfer plates 130 include a foot plate 132, a thigh plate 134, a seat plate 136, a chest plate 138, and a head plate 140. Plates 132, 134, 136, 138, and 140 are each formed from a low friction plastic material. Plates are mounted to bottom surface 14 with suitable fasteners such as screws 142. It is understood that a plurality of fasteners 142 are used to couple each transfer plate 132, 134, 136, 138, and 140 to the bottom cover 10. It is also understood that other suitable fasteners such as rivets, snaps, etc. may be used for the plates 130. Each plate 132, 134, 136, 138, and 140 is formed to include a pair of apertures 144 which provide handle grips to facilitate transfer of the mattress assembly 10. Each plate 132, 134, 136, 138, and 140 is also formed to include a plurality of elongated apertures 145. The transfer plates 130 are used to reduce the friction while sliding the mattress assembly 10 from one bed frame to another to permit transfer without disrupting a patient lying on the mattress assembly 10.
Blower assembly 52 is configured to hang on to a foot board 146 of a bed 148 as shown in FIG. 4. The blower assembly 52 includes a handle 150, blower housing 153, and a touch screen control display 152. The touch screen control display or valve control 152 permits an operator to control operation of the blower assembly 52 and valves 28, 30, and 56 to control therapies of the mattress assembly 10. A main microprocessor of the assembly is included within the blower housing. In addition, a blower motor and a power supply are located within the blower housing.
Air enters the blower housing 153 through intake manifold 154 in the direction of arrows 156. Air exits blower assembly 52 through outlet connector 158 and passes through air hose 160 to the inlet of manifold of valve 56. Manifold 154 is configured to reduce air intake noise into blower assembly 52. Manifold 154 includes a rear wall 162 defining an inlet 164 along a bottom surface of manifold 154. Pegs 166 on opposite sides if manifold 154 are configured to couple the manifold 154 to the blower housing 153 by entering slots 168 as shown in FIG. 6.
Manifold 154 includes an internal lip 170 to retain a filter 172 in the manifold 154. In the illustrated embodiment, the blower housing 153 includes a recessed portion 174 for receiving the manifold 154. A grate 176 permits inlet air to pass into the blower housing 153 in the direction of arrows 178. The grate 176 is not required. In other words, an opening can be formed in blower housing 153 without the grate 176.
As best illustrated in FIG. 5, manifold 154 deflects inlet air entering the blower housing 153 in the direction of arrows 156 by an angle of 90°. This directional change reduces air intake noise. A layer of sound foam 180 is located along rear wall 162 to further reduce air intake noise.
Another feature of the present disclosure is illustrated in FIG. 8. The air supply hose 160 includes air connectors 158 at each end. Connectors include a hose fitting 182, an outer sleeve 184, and an O-ring 186. A spring release 188 is provided to lock the fittings 158 in place. An electrical cable 190 includes electrical connectors 192 at opposite ends. Cable 190 is inserted through openings 194 and fittings 182 so that the cable 190 extends through the air tube 160 from the blower housing 153 into the inside of mattress assembly 10. Therefore, cable 190 is not exposed. One connector 192 is coupled to the electrical circuit of the blower assembly 52 and the other connector 192 is coupled to the electrical circuit within the mattress assembly 10. When the fittings 182 and 184 are assembled, the fittings 182 and 184 clamp the cable 190 to provide strain relief for the cable 190.
If it is desired to quickly deflate the plurality of air cushions within the mattress assembly 10, the fitting 158 can be removed from an air outlet of the housing 153 and the manifold 154 can be removed from the air inlet of the blower housing 153. The fitting 158 coupled to air hose 160 is then connected to a female receptacle molded into the housing 153 at the air inlet so that air may be removed rapidly from the plurality of air cushions of the mattress assembly 10.
Another feature of the present disclosure is illustrated in FIGS. 9-13. A siderail down sensor 200 is provided coupled to a frame 202 of bed 148. The siderail down sensor 200 is configured to provide an output signal over signal line 204 when the siderail 206 of bed 148 is moved downwardly in the direction of arrows 208.
As illustrated in the enlarged views in FIGS. 10 and 11, the frame includes a support member 210 movable from the position over sensor apparatus 200 when the siderail is up to the position spaced apart from sensor apparatus 200 when the siderail is down. Sensor 200 includes a body 212 and fasteners 214 for securing the body 212 to the frame 202. Sensor 200 also includes a switch assembly 216 having an actuator arm 218 which closes and opens a switch 220 as the siderail 206 moves from its up position illustrated in FIG. 9 to the down position. In other words, when the switch 220 is open as shown in FIG. 11, an output signal is generated to indicate that the siderail 206 is down. When the controller 227 receives a siderail down signal from sensor 200, certain therapies of the mattress assembly 10 are disabled. For instance, rotational therapy is discontinued upon detection of the siderail being down by sensor 200.
Another embodiment of the siderail down sensor is illustrated in FIGS. 12 and 13. In this embodiment, a clip assembly 222 is provided for securing the sensor 224 to the siderail 206. Specifically, the clip assembly 222 is configured to mount the sensor 224 to a support frame 226 of siderail 206. Clip assembly 222 includes a first body portion 228 slidably coupled to a second body portion 230. First and second body portions 228 and 230 are biased toward each other by springs 232. Illustratively, sensor 224 is a ball switch or a mercury switch.
Angle sensors are provided within the mattress assembly 10 so that the microprocessor can determine the articulation angle for a head section 24 of the mattress assembly 10. A first sensor such as an accelerometer is located in a seat section of the mattress assembly 10. A second sensor such as an accelerometer is coupled to a bottom surface of one of the valves 28 or 30 located within the head section 24 of the mattress assembly 10. The seat section accelerometer provides a reference output since the seat section does not articulate. Therefore, a zero reading can be taken from the seat sensor. As the head of the bed is articulated, the head sensor detects such movement and compares its new position to the reference position from the sensor in the seat section. The seat section sensor can accommodate movement to the Trendelenburg and reverse-Trendelenburg position so that the angle of the head section of the mattress relative to the seat section can always be detected during articulation of the mattress assembly 10 on a bed frame.
Although the invention has been described in detail with reference to a certain illustrated embodiment, variations and modifications exist within the scope and spirit of the present invention as described and defined in the following claims.
|Cited Patent||Filing date||Publication date||Applicant||Title|
|US779576||Sep 11, 1903||Jan 10, 1905||Benjamin F Berryman||Mattress.|
|US3303518||Sep 8, 1964||Feb 14, 1967||Ingram George||Inflatable mattresses, pillows and cushions|
|US3772717||Feb 11, 1971||Nov 20, 1973||K Yuen||Inflatable mattresses and cushions|
|US3978530||Nov 21, 1975||Sep 7, 1976||Amarantos John G||Air inflatable bed-like device with adjustable back support|
|US4477935||Jan 8, 1982||Oct 23, 1984||Griffin Gordon D||Mattress support system|
|US4483029||Jul 21, 1982||Nov 20, 1984||Support Systems International, Inc.||Fluidized supporting apparatus|
|US4525885||Nov 16, 1984||Jul 2, 1985||Mediscus Products Limited||Support appliance for mounting on a standard hospital bed|
|US4527298||Mar 18, 1982||Jul 9, 1985||Moulton Lee A||Electro pneumatic bed|
|US4541135||Apr 16, 1984||Sep 17, 1985||Victor Karpov||Air mattress|
|US4637083||Mar 13, 1985||Jan 20, 1987||Support Systems International, Inc.||Fluidized patient support apparatus|
|US4638519||Apr 4, 1985||Jan 27, 1987||Air Plus, Inc.||Fluidized hospital bed|
|US4825486||Dec 24, 1987||May 2, 1989||Matsushita Electric Works, Ltd.||Bedsore-preventing air mattress controller|
|US4944060||Mar 3, 1989||Jul 31, 1990||Peery John R||Mattress assembly for the prevention and treatment of decubitus ulcers|
|US4951335||Jun 5, 1989||Aug 28, 1990||Donan Marketing Corporation||Mattress assembly|
|US4993920||Apr 7, 1989||Feb 19, 1991||Harkleroad Barry A||Air mattress pumping and venting system|
|US5020176||Oct 20, 1989||Jun 4, 1991||Angel Echevarria Co., Inc.||Control system for fluid-filled beds|
|US5029352||Feb 14, 1990||Jul 9, 1991||Ssi Medical Services, Inc.||Dual support surface patient support|
|US5036559||Dec 6, 1989||Aug 6, 1991||SSI Medical Sevices, Inc.||Method of dual mode patient support|
|US5067189||Apr 11, 1990||Nov 26, 1991||Weedling Robert E||Air chamber type patient mover air pallet with multiple control features|
|US5121512||Nov 1, 1989||Jun 16, 1992||Irene Kaufmann||Auxiliary inflatable device serving as mattress|
|US5168589||Jun 11, 1991||Dec 8, 1992||Kinetic Concepts, Inc.||Pressure reduction air mattress and overlay|
|US5267364||Aug 11, 1992||Dec 7, 1993||Kinetic Concepts, Inc.||Therapeutic wave mattress|
|US5325551||Jun 16, 1992||Jul 5, 1994||Stryker Corporation||Mattress for retarding development of decubitus ulcers|
|US5483709||Apr 1, 1994||Jan 16, 1996||Hill-Rom Company, Inc.||Low air loss mattress with rigid internal bladder and lower air pallet|
|US5539942||Dec 17, 1993||Jul 30, 1996||Melou; Yves||Continuous airflow patient support with automatic pressure adjustment|
|US5542136||Aug 5, 1994||Aug 6, 1996||Stryker Corporation||Portable mattress for treating decubitus ulcers|
|US5561873||Jun 22, 1995||Oct 8, 1996||Patient Transfer Systems, Inc.||Air chamber-type patient mover air pallet with multiple control features|
|US5564142||May 11, 1995||Oct 15, 1996||Liu; Tsung-Hsi||Air mattress collaboratively cushioned with pulsative and static symbiotic sacs|
|US5586346||Feb 15, 1994||Dec 24, 1996||Support Systems, International||Method and apparatus for supporting and for supplying therapy to a patient|
|US5611096||May 9, 1994||Mar 18, 1997||Kinetic Concepts, Inc.||Positional feedback system for medical mattress systems|
|US5623736||Dec 9, 1994||Apr 29, 1997||Suport Systems, International||Modular inflatable/air fluidized bed|
|US5634225||May 25, 1995||Jun 3, 1997||Foamex L.P.||Modular air bed|
|US5699570||Jun 14, 1996||Dec 23, 1997||Span-America Medical Systems, Inc.||Pressure relief valve vent line mattress system and method|
|US5787531||Jul 23, 1996||Aug 4, 1998||Pepe; Michael Francis||Inflatable pad or mattress|
|US5794288||Jun 14, 1996||Aug 18, 1998||Hill-Rom, Inc.||Pressure control assembly for an air mattress|
|US5815865||Nov 30, 1995||Oct 6, 1998||Sleep Options, Inc.||Mattress structure|
|GB159299A||Title not available|
|GB2092439A||Title not available|
|GB2199803A||Title not available|
|WO1994009686A1||Oct 29, 1993||May 11, 1994||Geomarine Systems, Inc.||Lateral rotation therapy mattress system and method|
|WO1995031920A1||May 25, 1995||Nov 30, 1995||Egerton Hospital Equipment Limited||Improvements in and relating to low air-loss mattresses|
|WO1996033641A1||Apr 24, 1996||Oct 31, 1996||Kinetic Concepts, Inc.||Air bed with fluidized bead surface and related methods|
|1||Air Flow 5000 Mattress Replacement System, Atlantis Medical, Milltown, NJ, date unknown.|
|2||Apropos, CRS-8500, National Patient Care Systems, date unknown.|
|3||ASAP II Therapy System, DynaMedics Corporation, London, ON, Canada, Mar. 1995.|
|4||Bazooka, Innovative Medical Systems, Manchester, NH, 1995.|
|5||Economic Relief, Bio Therapy(R) Plus, Sunrise Medical Bio Clinic, Ontario, CA, date unknown.|
|6||Economic Relief, Bio Therapy® Plus, Sunrise Medical Bio Clinic, Ontario, CA, date unknown.|
|7||First Step, Mattress Replacement System, KCI, San Antonio, TX, 1991.|
|8||Gaymar Sof-Care Plus(R) Companion(TM) System, Gamar Industries, Inc., 1994.|
|9||Gaymar Sof-Care Plus® Companion™ System, Gamar Industries, Inc., 1994.|
|10||Impression, Pressure Relief Therapy, KCI, date unknown.|
|11||Lumex AkroTech 4000, Lumex, date unknown.|
|12||microAir(TM) 1000, GSI Medical Systems, Carmel, NY, 1989.|
|13||microAir™ 1000, GSI Medical Systems, Carmel, NY, 1989.|
|14||PRO 2000 MRS, Pneu-Care Series, Cardio Systems, Dallas, TX, date unknown|
|15||Z,701enaissance(TM), Therapeutic Mattress Replacement System, Pegausus Airwave Inc., date unknown.|
|16||Z,701enaissance™, Therapeutic Mattress Replacement System, Pegausus Airwave Inc., date unknown.|
|Citing Patent||Filing date||Publication date||Applicant||Title|
|US6684434 *||Dec 4, 2002||Feb 3, 2004||Hill-Rom Services, Inc.||Mattress assembly|
|US6689077||May 6, 2002||Feb 10, 2004||Reza R. Dabir||Apparatus and method for pressure management having temperature controlled air flow|
|US6829796||Oct 1, 2002||Dec 14, 2004||Hill-Rom Services, Inc.||Integrated barrier and fluid supply for a hospital bed|
|US7037278||Apr 14, 2003||May 2, 2006||Dabir Reza R||Apparatus and method for pressure management having temperature controlled air flow|
|US7114204||Jan 14, 2005||Oct 3, 2006||Smart Medical Technology, Inc.||Method and apparatus for transferring patients|
|US7260860||Mar 7, 2005||Aug 28, 2007||Hill-Rom Services, Inc.||Mattress system for a hospital bed|
|US7310839||Dec 14, 2004||Dec 25, 2007||Hill-Rom Services, Inc.||Patient support apparatus|
|US7322947||Dec 5, 2003||Jan 29, 2008||Gaymar Industries, Inc.||Vibrational and pulsating cushioning device|
|US7363663||Jul 9, 2007||Apr 29, 2008||Hill-Rom Services, Inc.||Mattress with automatic width adjustment|
|US7461425||Jul 9, 2007||Dec 9, 2008||Hill-Rom Services, Inc.||Bed with automatically identifiable mattress type|
|US7565710||Jul 9, 2007||Jul 28, 2009||Hill-Rom Services, Inc.||Support surface with inflatable width adjustment portion|
|US7648392||Dec 13, 2005||Jan 19, 2010||Hill-Rom Services, Inc.||Quick connector for multi-media|
|US7735164||Dec 5, 2007||Jun 15, 2010||Smart Medical Technology, Inc.||Disposable patient transfer mattress|
|US7805782||Mar 6, 2006||Oct 5, 2010||Hill-Rom Services, Inc.||Siderail for a hospital bed|
|US7832039||Jun 17, 2009||Nov 16, 2010||Hill-Rom Services, Inc.||Support surface with inflatable core zones|
|US7904976||Apr 27, 2007||Mar 15, 2011||Hill-Rom Services, Inc.||Endboard for a patient support|
|US8006333 *||Nov 15, 2007||Aug 30, 2011||Stryker Corporation||Patient support surface with turn-assist|
|US8038632||Dec 24, 2007||Oct 18, 2011||Stryker Corporation||Vibrational and pulsating cushion device|
|US8104122||Dec 18, 2006||Jan 31, 2012||Hill-Rom Services, Inc.||Patient support having an extendable foot section|
|US8122546||Nov 15, 2010||Feb 28, 2012||Hill-Rom Services, Inc.||Adjustable width mattress with relief portions|
|US8201292 *||Nov 14, 2007||Jun 19, 2012||Stryker Corporation||Patient support surface with turn-assist|
|US8276222||Oct 3, 2006||Oct 2, 2012||Smart Medical Technology, Inc.||Patient transfer kit|
|US8429774||Aug 13, 2010||Apr 30, 2013||Hill-Rom Industries Sa||Lateral tilt device|
|US8601622||Apr 5, 2013||Dec 10, 2013||Hill-Rom Industries S.A.||Patient support apparatus including a lateral tilt device|
|US8677536||Nov 18, 2009||Mar 25, 2014||Hill-Rom Services, Inc.||Method and apparatus for sensing foot retraction in a mattress replacement system|
|US8887326||Sep 7, 2012||Nov 18, 2014||Smart Medical Technology, Inc.||Patient transfer kit|
|US9125777||Nov 22, 2011||Sep 8, 2015||Sage Products, Llc||Body transport apparatus|
|US9241580||Nov 26, 2013||Jan 26, 2016||Sage Products, Llc||Body transport apparatus with integrated handles|
|US9314388||Jul 27, 2015||Apr 19, 2016||Sage Products, Llc||Body transport apparatus|
|US9526348 *||Oct 7, 2014||Dec 27, 2016||Hill-Rom Services, Inc.||Person support systems|
|US20040193078 *||Dec 5, 2003||Sep 30, 2004||Flick Roland E.||Vibrational and pulsating cushioning device|
|US20050091753 *||Dec 14, 2004||May 5, 2005||Hill-Rom Services, Inc.||Patient support apparatus|
|US20060026767 *||Mar 7, 2005||Feb 9, 2006||Chambers Kenith W||Mattress system for a hospital bed|
|US20060117482 *||Dec 2, 2005||Jun 8, 2006||Branson Gregory W||Touch screen control for lateral rotation of a hospital bed mattress|
|US20060156468 *||Jan 14, 2005||Jul 20, 2006||Patrick James E||Method and apparatus for transferring patients|
|US20060156473 *||Dec 13, 2005||Jul 20, 2006||Chambers Kenith W||Quick connector for multi-media|
|US20060195984 *||Mar 6, 2006||Sep 7, 2006||Reza Hakamiun||Siderail for a hospital bed|
|US20060195986 *||Mar 6, 2006||Sep 7, 2006||Reza Hakamiun||Footboard for a hospital bed|
|US20080005848 *||Jul 9, 2007||Jan 10, 2008||Chambers Kenith W||Support Surface with Inflatable Width Adjustment Portion|
|US20080010752 *||Jul 9, 2007||Jan 17, 2008||Chambers Kenith W||Bed with Automatically Identifiable Mattress Type|
|US20080092295 *||Dec 24, 2007||Apr 24, 2008||Gaymar Industries, Inc.||Vibrational and Pulsating Cushioning Device|
|US20080097259 *||Dec 24, 2007||Apr 24, 2008||Gaymar Industries, Inc.||Vibrational and Pulsating Cushion Device|
|US20080120780 *||Nov 15, 2007||May 29, 2008||Stryker Corporation||Patient support surface with turn-assist|
|US20080141463 *||Nov 14, 2007||Jun 19, 2008||Stryker Corporation||Patient support surface with turn-assist|
|US20080201858 *||Feb 25, 2008||Aug 28, 2008||Jean-Luc Caminade||Mattress type support device including at least one solenoid valve for controlling fluid feed/vent to or from compartments of the mattress|
|US20080235875 *||Mar 28, 2008||Oct 2, 2008||Stryker Corporation||Maternity bed and patient lying surface therefor|
|US20080263771 *||Apr 27, 2007||Oct 30, 2008||Hill-Rom Services, Inc.||Endboard for a patient support|
|US20090249552 *||Jun 17, 2009||Oct 8, 2009||Hill-Rom Services, Inc.||Support Surface with Inflatable Core Zones|
|US20110099723 *||Nov 15, 2010||May 5, 2011||Chambers Kenith W||Adjustable Width Mattress with Relief Portions|
|US20110113561 *||Nov 18, 2009||May 19, 2011||Douglas Stephen L||Method and apparatus for sensing foot retraction in a mattress replacement system|
|US20110113562 *||Nov 8, 2010||May 19, 2011||Uzzle Thomas E||Endboard for person support apparatus|
|US20110162142 *||Mar 14, 2011||Jul 7, 2011||Reza Hakamiun||Endboard for a patient support|
|US20150089749 *||Oct 7, 2014||Apr 2, 2015||Hill-Rom Services, Inc.||Person support systems|
|International Classification||A61G7/10, A61G7/05, A47C27/10, A61G7/057|
|Cooperative Classification||A61G7/052, A61G7/0509, A61G7/0524, A61G2203/20, A61G7/0506, A61G7/0507, A61G7/103, A61G2200/32, A61G7/05, A61G7/05769|
|European Classification||A61G7/057K, A61G7/05S|
|Apr 13, 2006||FPAY||Fee payment|
Year of fee payment: 4
|Apr 14, 2010||FPAY||Fee payment|
Year of fee payment: 8
|May 30, 2014||REMI||Maintenance fee reminder mailed|
|Oct 22, 2014||LAPS||Lapse for failure to pay maintenance fees|
|Dec 9, 2014||FP||Expired due to failure to pay maintenance fee|
Effective date: 20141022