US 6842702 B2 Abstract A method of multivariate spectral analysis, termed augmented classical least squares (ACLS), provides an improved CLS calibration model when unmodeled sources of spectral variation are contained in a calibration sample set. The ACLS methods use information derived from component or spectral residuals during the CLS calibration to provide an improved calibration-augmented CLS model. The ACLS methods are based on CLS so that they retain the qualitative benefits of CLS, yet they have the flexibility of PLS and other hybrid techniques in that they can define a prediction model even with unmodeled sources of spectral variation that are not explicitly included in the calibration model. The unmodeled sources of spectral variation may be unknown constituents, constituents with unknown concentrations, nonlinear responses, non-uniform and correlated errors, or other sources of spectral variation that are present in the calibration sample spectra. Also, since the various ACLS methods are based on CLS, they can incorporate the new prediction-augmented CLS (PACLS) method of updating the prediction model for new sources of spectral variation contained in the prediction sample set without having to return to the calibration process. The ACLS methods can also be applied to alternating least squares models. The ACLS methods can be applied to all types of multivariate data.
Claims(48) 1. A method of multivariate spectral analysis, comprising the steps of:
a) obtaining an estimate of spectral error covariance E
_{A }for measured set of multivariate spectral data A; b) decomposing the spectral error covariance E
_{A }according to E_{A}=TP+E, where T is a set of n×r scores and P is a set of r×p loading vectors obtained from factor analysis of the spectral error covariance E_{A}, and E is a set of n×p random errors and spectral variations not useful for prediction; c) guessing pure-component spectra K for the set of multivariate spectral data A;
d) predicting a set of component values Ĉ according to
ĈAK ^{T}(KK ^{T})^{−1} =A(K ^{T})^{+}; e) augmenting the set of predicted component values Ĉ with at least one vector of the T scores to obtain a first set of augmented component values
f) estimating augmented pure-component spectra
according to
g) testing for convergence according to
h) predicting a second set of augmented component values
according to
i) replacing the augmented portion of the second set of augmented component values
with the at least one vector of the T scores to obtain a third set of augmented component values
and
j) repeating steps f) through i) at least once.
2. The method of
and
3. The method of
with at least one vector of the P loading vectors prior to step h).
4. The method of
with at least one vector representing a spectral shape that is representative of at least one additional source of spectral variation prior to step h).
5. The method of
at step f).
6. The method of
7. The method of
at step h).
8. The method of
9. The method of
10. The method of claim wherein the measured set of multivariate spectral data A comprises image data.
11. The method of
^{A }is obtained from a shift difference generated from a single image.12. The method of
^{A }is obtained from repeat image spectra.13. A method of multivariate spectral analysis, comprising the steps of:
a) obtaining an estimate of spectral error covariance E
_{A }for measured set of multivariate spectral data A; b) decomposing the spectral error covariance E
_{A }according to E_{A}=TP+E, where T is a set of n×r scores and P is a set of r×p loading vectors obtained from factor analysis of the spectral error covariance E_{A}, and E is a set of n×p random errors and spectral variations not useful for prediction; c) guessing pure-component spectra K for the set of multivariate spectral data A;
d) augmenting the pure-component spectra K with at least one vector of the P loading vectors to obtain first augmented pure-component spectra {tilde over (K)};
e) predicting a first set of augmented component values
according to
f) estimating second augmented pure-component spectra
according to
g) testing for convergence according to
h) replacing the augmented portion of the second augmented pure-component spectra
with the at least one vector of the P loading vectors to obtain third augmented pure-component spectra
and
i) predicting a second set of augmented component values
according to
j) repeating steps f) through i) at least once.
14. The method of
and
15. The method of
16. The method of
with at least one vector of the T scores prior to step h).
17. The method of
with at least one vector representing a spectral shape that is representative of at least one additional source of spectral variation prior to step h).
18. The method of
at step f).
19. The method of
20. The method of
at step h).
21. The method of
22. The method of
23. The method of
24. The method of
_{A }is obtained from a shift difference generated from a single image.25. The method of
_{A }is obtained from repeat image spectra.26. The method of
_{A }is obtained from a shift difference generated from a single image.27. The method of
_{A }is obtained from repeat image spectra.28. A method of multivariate spectral analysis, comprising the steps of:
a) obtaining an estimate of the spectral error covariance E
_{A }for measured set of multivariate spectral data A; b) decomposing the spectral error covariance E
_{A }according to E_{A}=TP+E, where T is a set of n×r scores and P is a set of r×p loading vectors obtained from factor analysis of the spectral error covariance E_{A}, and E is a set of n×p random errors and spectral variations not useful for prediction; c) guessing a set of component values C for the set of multivariate spectral data A;
d) estimating pure-component spectra {circumflex over (K)} according to {circumflex over (K)}=(C
^{T}C)^{−1}C^{T}A=C^{+}A; e) augmenting the pure-component spectra {circumflex over (K)} with at least one vector of the P loading vectors to obtain first augmented pure-component spectra
f) predicting a first set of augmented component values
according to
g) testing for convergence according to
h) estimating second augmented pure-component spectra
according to
i) replacing the augmented portion of the second augmented pure-component spectra
with the at least one vector of the P loading vectors to obtain a third augmented pure-component spectra
and
j) repeating steps f) through i) at least once.
29. The method of
and
30. The method of
with at least one vector of the T scores prior to step h).
31. The method of
with at least one vector representing a spectral shape that is representative of at least one additional source of spectral variation prior to step h).
32. The method of
at step h).
33. The method of
34. The method of
at step f).
35. The method of
36. The method of
37. A method of multivariate spectral analysis, comprising the steps of:
a) obtaining an estimate of the spectral error covariance E
_{A }for measured set of multivariate spectral data A; _{A }according to E_{A}=TP+E, where T is a set of n×r scores and P is a set of r×p loading vectors obtained from factor analysis of the spectral error covariance E_{A}, and E is a set of n×p random errors and spectral variations not useful for prediction; c) guessing a set of component values C for the set of multivariate spectral data A;
d) augmenting the set of component values C with at least one vector of the T scores to obtain a first set of augmented component values {tilde over (C)};
e) estimating augmented pure-component spectra
according to
f) testing for convergence according to
g) predicting a second set of augmented component values
according to
h) replacing the augmented portion of the second set of augmented component values
with the at least one vector of the T scores to obtain a third set of augmented component values
and
i) repeating steps e) through h) at least once, using the augmented component values
in step f).
38. The method of
and
39. The method of
with at least one vector of the P loading vectors prior to step e).
40. The method of
with at least one vector representing a spectral shape that is representative of at least one additional source of spectral variation prior to step e).
41. The method of
at step e).
42. The method of
43. The method of
at step g).
44. The method of
45. The method of
46. The method of
47. The method of
_{A }is obtained from a shift difference generated from a single image.48. The method of
_{A }is obtained from repeat image spectra.Description This application is a divisional of application Ser. No. 10/209,841, filed Jul. 31, 2002 now U.S. Pat. No. 6,687,620, application Ser. No. 10/209,841 claims the benefit of Provisional Application No. 60/309,619, filed on Aug. 1, 2001, and Provisional Application No. 60/311,755, filed on Aug. 9, 2001. The Government has rights to this invention pursuant to Contract No. DE-AC04-94AL85000 awarded by the U.S. Department of Energy. The present invention relates generally to the field of multivariate spectral analysis and, more particularly, to a method for augmenting a classical least squares calibration model to provide improved predictions of component values in unknown samples having unmodeled sources of spectral variation. Over the past 20 years, quantitative multivariate spectral analysis has primarily shifted from the explicit classical least squares (CLS) method to the implicit principal component regression (PCR) and partial least squares (PLS) methods. The principle motivation for this shift is that CLS is based on an explicit linear additive model, e.g., the Beer-Lambert law. As such, CLS has the significant limitation that it requires the concentrations of all spectrally active components be known and included in the calibration model before an adequate prediction model can be developed. On the other hand, the PCR and PLS methods can achieve excellent predictions for multivariate spectral data sets where all of the spectrally active components have not been determined. Consequently, CLS has been relegated to solving a small set of well-defined linear problems with known spectrally active components that adhere to the Beer-Lambert law, e.g., infrared spectra of gas-phase samples. Nevertheless, PCR and PLS do not have the qualitative capabilities of CLS since they do not generate explicit estimated pure-component spectra that can be readily interpreted. Also, they are not well suited to the advantages of a newly developed prediction-augmented CLS (PACLS) technique as set forth in U.S. Pat. No. 6,415,233, which is incorporated by reference herein. The PACLS algorithm provides a basis for rapidly updating a CLS model during prediction of component values of the target unknown sample. PACLS adds spectral shapes (i.e., spectral intensity information) to the CLS estimate of the pure-component spectra during prediction to account for spectrally active components or other spectral effects present in the prediction samples that were not modeled during calibration. PACLS allows CLS models to be updated for the presence of spectrometer drift, changes in spectrometer parts or changes in whole spectrometers, unmodeled chemical or non-chemical spectral components, as well as updating for more generalized changes such as changes in starting materials, the presence of nonlinearities, chromatic aberrations, or stray light, etc. However, the PACLS algorithm is limited by the fact that accurate predictions require all interfering spectral components (including chemical and non-chemical sources of spectral variation) be explicitly included during calibration or prediction. If one or more spectral interferences were left out of the calibration, then their spectral influence would have to be explicitly added during prediction to correct for their absence in the calibration model. These limitations of the CLS model can be reduced and even eliminated by the development of a new generalized family of algorithms, hereinafter referred to as augmented classical least squares (ACLS). The ACLS model uses information derived from component values and spectral residuals during the CLS calibration to provide an improved calibration-augmented CLS model. When the new ACLS methods are combined with the PACLS prediction algorithm, a powerful set of new multivariate capabilities is realized such that analyses can be performed with incomplete knowledge of interferences in the calibration and the prediction data. The present invention further provides a generalization of ACLS methods for analyzing multivariate spectral data. Specific embodiments of the generalized ACLS methods are: spectral-residual augmented classical least squares (SRACLS), scores augmented classical least squares (SACLS), and concentration-residual augmented classical least squares (CRACLS) methods which all allow one to overcome the above deficiencies. The SRACLS, SACLS, and CRACLS methods are based on CLS so that they retain the qualitative benefits of CLS, yet they have the flexibility of PLS and other hybrid techniques in that they can define a prediction model even with unmodeled sources of spectral variation that are not explicitly included in the calibration model. The unmodeled sources of spectral variation may be unknown constituents, constituents with unknown concentrations, nonlinear responses, non-uniform and correlated errors, or other sources of spectral variation (e.g., temperature, spectrometer drift, etc.) that are present in the calibration sample spectra. Augmentation can also be applied to constrained alternating classical least squares methods (alternating between CLS calibration and CLS prediction) that are used when the reference variables, such as the pure-component spectra or component concentrations, are inadequately known for standard CLS method. The ACLS methods of the present invention can improve the component identification with such inadequately known data sets. Combining the present invention with the PACLS technique results in prediction models that are generally comparable or better in prediction ability to the standard PLS models. Also, since the various ACLS methods are based on CLS and unlike PLS, they can incorporate the PACLS feature of updating the prediction model for new sources of spectral variation without the need for time-consuming recalibration. These updated prediction models only require spectral information while PLS requires spectral and concentration information during recalibration. The present invention is not restricted to using continuous spectral information, but can also use any set of discontinuous spectral intensities that are selected in the calibration for the least squares analysis. Finally, the present invention generates better qualitative information about the analytes by generating better estimates of their pure-component spectra. A method of multivariate spectral analysis is provided that is able to generate accurate and precise prediction models from multivariate spectral data which includes unmodeled spectrally active components present in the calibration model. The method provides the improved qualitative information of CLS methods as well as the quantitative prediction ability of the implicit multivariate calibration methods by augmenting the calibration model with a measure of the residual resulting from unmodeled components. The present method of multivariate spectral analysis is most useful when an underlying calibration model includes unmodeled sources of spectral variation. In particular, reference variables (e.g., sample spectra and component values) are obtained for a set of calibration samples. An estimate is obtained for at least one reference variable for the set of calibration samples. Thereafter, a residual is obtained between the at least one reference variable and its corresponding estimated variable. A measure of the residual between the estimated and reference variable is used to augment its corresponding reference variable. Using the augmented reference variable, a value of at least one component in a set of unknown samples is predicted. Augmentation can be repeated until all sources of spectral variation are accounted for in the calibration samples. The iterative process allows the development of a CLS-type calibration model comparable in its quantitative prediction ability to implicit multivariate calibration methods, even when unmodeled spectrally active components are present in the calibration sample spectra. To better understand the present invention, the following introductory discussion is provided. The notation in the equations below uses upper-case bold letters for matrices, lower-case bold letters for vectors, and italicized letters for scalars. We use the ^ to indicate estimated values, As used herein, the term multivariate spectral data includes all types of multivariate data (i.e., data related to each sample or object is composed of responses from two or more variables). As used herein, spectral data to be used in the multivariate spectral analysis are considered to be any form of data where there are responses at two or more measured variables (e.g., wavelength, time, retention time, frequency, etc.) due to one or more perturbation sources (e.g., electromagnetic radiation, vibration, temperature, mechanical impulses, etc.). Therefore, the present invention can be applied to conventional spectroscopic methods that include, but are not limited to, infrared, near-infrared, visible, ultraviolet, X-ray, gamma-ray, Raman, mass spectroscopy, ion-mobility mass spectroscopy, Auger, fluorescence, phosphorescence, ESCA, far-infrared, microwave, x-ray fluorescence, NMR, energy loss spectroscopy, EDAX, ESR, and multi- and hyper-spectral imaging spectroscopy. These spectroscopic methods can be used in absorption, transmission, reflection, or emission modes. In addition, the methods of the present invention can be applied to other forms of multivariate data that include, but are not limited to, seismic data, chromatographic data, thermal gravimetric analysis, image data, and responses from multiple sensors. In particular, the methods of the present invention can be used to provide quantitative analysis of image data of the type provided with recently available commercial hyper-spectral imaging spectrometers. The methods can also be applied to the analysis of multispectral and hyperspectral image data obtained from remote sensors, such as satellite or airborne hyperspectral imaging sensors. The basic CLS method can be described by the following relationship:
In a CLS prediction step, the estimated pure-component spectra can then be used to predict the component values in an unknown set of samples according to:
Eq. (2) provides accurate estimates of the known pure-component spectra {circumflex over (K)}, and therefore accurate predictions of the unknown component values Ĉ, only if the data fit a linear model and C contains all the components that contribute to spectral variation in the calibration sample spectra A. The accuracy of the estimated pure-component spectra {circumflex over (K)} improves with each additional known reference value c The present invention provides a family of methods for analyzing multivariate data where there are unmodeled sources of spectral variation in the calibration model. The family of methods can be broadly described as first obtaining reference variables for a set of calibration samples (e.g., reference spectral data and reference values for components in the set of calibration samples). An estimate is obtained for at least one of the reference variables for the set of calibration samples. Thereafter, a residual is obtained between the at least one reference variable and its corresponding estimated variable. The residual between the estimated and reference variables is used to augment its corresponding reference variable. Using the augmented reference variable, a value of at least one component in a set of unknown samples is predicted. Augmentation can be repeated until all sources of spectral variation are accounted for in the set of calibration samples. The iterative process allows the development of a CLS-type calibration model comparable in quantitative prediction ability to implicit multivariate calibration methods even when unmodeled spectrally active components are present in the calibration sample spectra. The present invention generates an adaptable model that can achieve excellent quantitative and qualitative prediction results for samples of unknown composition that contain unmodeled sources of spectral variation. In the spectral-residual-augmented classical least squares (SRACLS) embodiment of the present invention, a spectral residual matrix E For an incompletely specified and/or nonlinear model, the spectral residual E Factor analysis methods can be applied to the spectral residual E In one embodiment of the present invention, Eq. (6) can be factor analyzed by any factor analysis method. Common methods include principal component analysis (PCA), partial least squares (PLS), or principal component regression (PCR). If PLS or PCR is used, then concentration errors (to be discussed later) will be included in the calibration process, as described in U.S. Pat. No. 6,341,257, which is incorporated herein by reference. The factor analysis can use either an orthogonal factor analysis method or any non-orthogonal factor analysis method. Using only the spectral residuals in the factor analysis, the scores, T, or the loading vectors, P, can be used to augment the CLS calibration model. If PCA is used as the factor analysis method applied to the calibration spectral residuals E The scores T can also be used to correct the CLS calibration model for unknown spectral components. In this case, the columns of T can augment the column dimension of the component values C matrix to form an augmented component value matrix {tilde over (C)}. Re-calibrating with the augmented CLS method as in Eq. (8) results in an augmented
With both SRACLS and SACLS, the spectral residuals E Another ACLS method involves the use of component value residuals E An n×m component residuals matrix, E To account for baseline variations, {circumflex over (K)} can be augmented with explicit baseline functions, i.e., vectors representing the potential variations in baselines in the spectra. These baseline functions can include an offset, general polynomials, orthogonal Legendre polynomials, or any expected functional form of the baselines. A row is added to {circumflex over (K)} for each additional order in the baseline function added. Using the PACLS technique, estimated or measured spectral shapes for other spectrally active components can be added to {circumflex over (K)} as well. A corresponding column must be added to Ĉ for each added row in {circumflex over (K)} before solving Eq. (10). If rows have been added to {circumflex over (K)}, then E To consider unmodeled sources of spectral variation, Eq. (1) can be rewritten:
The matrix, K Although D is not known, Eq. (15) still shows that each column, e In practice, D=0 will generally occur only if the spectral components do not overlap in the spectral region being analyzed. For calibration of the known components, the magnitudes for unit component value of the pure-component spectra for the unmodeled components are not required, since only the relative component values are needed to generate the correct net-analyte signals (NAS) for each of the known components. By including linear combinations of the unmodeled component values as additional columns in C and solving Eq. (16), the resulting additional pure-component spectra
Selecting one vector of the component residuals, e Using {tilde over (C)}, the augmented, (m+1)×p, pure-component spectra matrix
By iteration, additional augmentations can be used to remove additional unmodeled sources of spectral variation. The new augmented estimated pure-component spectra
The steps delineated by Eqs. (11, 16-17) remove one linear combination of the unmodeled spectral variations from the calibration model. These steps can be repeated again using one row from E Using the fully augmented {tilde over (C)}, augmented pure-component spectra
Because any of the ACLS models represent a CLS-type model, they are well suited to take advantage of the newly developed PACLS technique. The PACLS algorithm further augments the augmented pure-component spectra
The spectra added during prediction should account for all sources of spectral variation introduced during the collection of the prediction data that were not accounted for in the calibration model, as described in the previously referenced U.S. Pat. No. 6,415,233. If these sources of spectral variation can be attributed to a known component, a sample could be doped with the identified component to determine a spectral shape that can be added to the prediction. However, often the new sources of spectral variation (such as caused by instrument drift or sample insertion effects) cannot be so easily identified. Frequently, a repeat sample can be used to provide the required spectral information to augment the model for prediction. By repeatedly measuring a stable sample during calibration and prediction spectral measurements, spectral shapes for these new sources of spectral variation can be captured. The differences between repeated spectral measurements obtained during both calibration and prediction or an eigenvector analysis of those differences can provide the spectral information required for augmentation of the prediction model. Note that, unlike PLS, the spectral variation can be incorporated using the PACLS method without knowing the values of various components. When using the repeat sample spectra, the assumption is that the spectra can be used to provide a reasonable estimate of the error covariance matrix, i.e., that the unmodeled sources of spectral variation affect the repeat sample spectra and the prediction sample spectra in the same way. This assumption is valid if the spectra are similar across all samples, which is the case for the dilute aqueous solutions considered herein. However, if the range of component values causes large variations in the sample spectra, it may be necessary to use a subset of repeat samples to adequately capture the impact of the unknown sources of spectral variation. The following discussion of the SRACLS method of the present invention is made with reference to FIG. Initially, a set of calibration samples containing a plurality of spectrally active components is obtained at At step At step The spectral residuals can be decomposed per Eq. (6). All or selected spectral residual vectors from E In In The following discussion of the CRACLS embodiment of the present invention is made with reference to FIG. Initially, a set of calibration samples containing a plurality of spectrally active components is obtained at At step By extension at Each component residual vector e Alternatively, in another embodiment with iterations indicated by the circular arrow, steps After the desired number of augmentations, the augmented CLS prediction for component values
The CRACLS method of the present invention is demonstrated with both simulated and real data derived from a system of dilute aqueous solutions containing glucose, ethanol, and urea. The simulated data demonstrate the effectiveness of the CRACLS algorithm and help to elucidate the principles behind the method. Using experimental data, the prediction abilities of CRACLS and PLS are compared during cross-validated calibration. In combination with PACLS, the CRACLS predictions are comparable to PLS for the prediction of the glucose, ethanol, and urea components for validation samples collected when significant instrument drift was also present. The experimental samples consisted of a series of dilute aqueous solutions of glucose, ethanol, and urea each independently varied over the concentration range of 0-500 mg/dL. The samples were prepared by weight and volume in a pseudo D-optimal design that allowed each of the three analytes to be varied separately at 9 levels over the concentration range. The aqueous solvent was obtained from a single source of buffered saline solution. A detailed error analysis indicated that the samples were made to an accuracy of better than 1 mg/dL. The calibration data set consisted of 27 samples plus a repeat sample taken from the center of the design, i.e., the repeat sample contained approximately 250 mg/dL each of glucose, ethanol, and urea. The set of samples used for prediction (i.e., the validation samples) included the same repeat sample and 27 new samples from a design similar to the calibration set. The validation samples spanned the same 0 to 500 mg/dL concentration range for each of the analytes as in the calibration set. No sample from the new 27-sample prediction set had the same composition as any sample in the calibration set. Five of the prediction samples were removed from consideration when they were determined to be outlier samples contaminated by the epoxy used to seal the lids on the cuvettes. The samples were sealed with a magnetic stirring bar in 10-mm pathlength cuvettes. A temperature controller propelled the magnetic stirrer while holding the samples at a temperature constant to 0.05° C. (±1 σ). The samples were placed in the temperature controller and held in the beam of the spectrometer for 4 minutes with stirring to allow the sample temperature to equilibrate. The near-infrared spectra of the samples were obtained on a Nicolet Model 750 Fourier transform infrared (FT-IR) spectrometer. The spectrometer employed a 75 W tungsten-halogen lamp, quartz beam splitter, and liquid-N The run order of the calibration sample set was randomized. The spectrum of the repeat sample held at 32° C. was obtained after each group of two calibration or prediction samples. The spectra for the prediction samples were obtained approximately one month after the calibration spectra were obtained. Purge variation was introduced during the prediction data set to induce additional short-term instrument drift and to accentuate the difficulty in maintaining the calibration. Sample temperature was varied in random order in 2° C. steps over the range of 30 to 34° C. for both the calibration and prediction samples. A background of the empty sample holder was obtained after each sample. Transmittance spectra of the calibration and validation sample sets were obtained by ratioing each single-beam sample spectrum to either its corresponding background or to the average of the background for the day. The spectra were then converted to absorbance. Since the best calibration and prediction results were obtained with the spectra obtained using the averaged daily background, only the results obtained from these spectra are described herein. To demonstrate the effectiveness of the present invention, a simulated spectral data set was constructed using pure-component spectra derived from experimental dilute aqueous data using standard CLS methods. While these simulated spectra do not precisely match the true pure-component spectra (primarily because of significant baseline variations resulting from spectrometer drift during the collection of the calibration spectra), they can be used to demonstrate the capabilities of the present invention. The simulated spectral data were constructed from these pure-component spectra and a series of assumed concentrations to generate two different sets of 25 samples, one for calibration and the other for validation, containing 0-500 mg/dL each of glucose, ethanol, and urea and 98000-100000 mg/dL of water. The sum of all the components from each sample was constrained to 100000. In addition, normally distributed random spectral noise was added to the absorbance spectra at a level of 0.3% of the maximum spectral absorbance intensity. This level of noise amounted to approximately 5% of the absorbance caused by the minor components, which is slightly more noise than we observed in the measured spectral data. The spectral data were analyzed in the spectral region from 7500 to 11,000 cm The pure-component spectra K of the analytes and the water solvent used to generate the simulated spectral data are shown in FIG. Since water is the dominant component, the spectra of the simulated samples closely resemble that of water. However, if we remove the average spectrum, the spectral variation due to the analytes of each simulated sample is apparent as shown in FIG. The effect of augmentation on the prediction ability of the CRACLS model was examined for the simulation samples. If only two of the four components, glucose and ethanol, are used to build a CLS model based on the simulated data, the cross-validated standard errors of prediction (CVSEP) for glucose and ethanol are 234 mg and 93 mg, respectively. That is, there is almost no prediction ability for glucose. However, if a CRACLS model is built, augmenting the calibration model with the concentration residuals twice, CVSEPs of 17 mg for glucose and 3 mg for ethanol are achieved, demonstrating that the present invention method has clear prediction ability for both components. If only one component's concentrations is included in the CRACLS calibration model, either glucose or ethanol, and the model is augmented three times, we get essentially the same CVSEPs are obtained as above. Adding the concentration residuals E The CRACLS loading vectors qualitatively resemble the pure-component spectra. The estimated pure-component spectra or loading vectors (LV) that resulted by including water and urea concentrations in the calibration and augmenting with concentration residuals twice are shown as the solid lines in FIG. If the major component is left out of the CLS calibration, the qualitative information will not be as apparent. The loading vectors will generally match the shape of the major component due to its overwhelming influence. The extent of this influence on the simulated data is shown by the decomposition of the vectors in the next section. With mean centering, the loading vectors obtained from a CLS calibration without the solvent concentrations will reveal more of the distinctive spectral features of the non-solvent components. However, for a mixture system where the sum of the concentrations is constrained to a constant, the CLS estimated pure-component analyte spectrum represents the net change in the sample spectrum due to a unit change in the analyte concentration. Consequently, the estimated pure-component spectrum will include negative spectral changes due to the displacement of the solvent that will contaminate the mean-centered loading vectors. In every case, the qualitative information in the loading vectors for CLS will be equal to or better than PLS, since the first PLS weight-loading vector that retains the best qualitative information is generated from only one component. For CLS, and therefore for CRACLS, the qualitative information is improved as additional analyte concentrations are included in the calibration model. To illustrate this, a PLS calibration on glucose and a CLS calibration using only glucose and water were computed using the simulated data. The weight loading vector (WLV) from the PLS calibration and the loading vector (LV) from the CLS calibration for glucose are given in The CRACLS loading vectors may be decomposed to demonstrate the effective removal of unmodeled sources of spectral variation, even if only the minor components are included in the calibration. The decomposition of the first four CRACLS loading vectors provides insight into this effectiveness when only the minor components of glucose and ethanol are included in the calibration without mean centering. By defining a CLS prediction model using all four true pure-component spectra with unit concentration, shown in
The spectral residuals, after removing the given amount for each of the pure-component spectra, represent only random noise, indicating that all the spectral information contained in these loading vectors was removed. As shown in the Table I, the first and second loading vectors are the only ones that contain the glucose and ethanol pure-component spectra, respectively, corresponding to the order that their concentrations were placed in the concentration matrix. Note also that the pure-component spectra are present at unit concentration so they can provide quantitative information during prediction. In addition, the urea and water spectra contaminate these first two estimated pure-component spectra, as expected from Eq. (9). In Table I, the amount of water far exceeds the other components, so all of the loading vector shapes are very similar to the pure water spectrum. The last two estimated pure-component spectra are linear combinations of only urea and water, the components left out of the concentration matrix. Consequently, during prediction using these four loading vectors, the resulting NAS for glucose and ethanol (i.e., the part of each pure-component spectrum orthogonal to all other vectors) is the same as obtained from the fully specified CLS model. Basically, the contamination of both water and urea is removed from the prediction of the glucose and ethanol by the third and fourth loading vectors. Therefore, the prediction model correctly determines the glucose and ethanol concentrations even though the concentrations for the other two components were not explicitly included in the calibration. Now we consider the experimental data. The mean calibration spectrum and the mean-centered calibration sample spectra are shown in
All the chemical components and temperature were included in the model, and yet CLS still was unable to generate a precise calibration model. These results indicate why CLS is seldom used for prediction in multivariate calibrations. Also provided in Table II are the CVSEPs for PLS, with the optimum number of factors shown in parenthesis, and for CRACLS, using 10, 15 and 20 augmentations. Although there was a gradual decrease in the CVSEPs with more augmentations, the optimal number of augmentations was not apparent. Differences between the present invention calibration results and the PLS results for these data are not statistically significant. To predict the validation data, information from the repeat sample was included in both the PLS and the CRACLS models. For PLS, all the spectra from the repeat sample taken with the prediction data set were added to the original calibration data set along with its component concentrations, and the PLS model was recalibrated. The cross-validated calibrations were performed initially by using the standard approach of removing all the data for each sample during the rotations (sample-out rotation), so that all the spectra for the repeat sample were removed at once. The number of loading vectors or factors for the PLS model was determined using an F-test on the cross-validated calibrations. For the present invention, no changes were made in the CRACLS calibration model, i.e., no information from the repeat sample was given to the original CRACLS model to compute new pure-component spectra. Instead, the sources of spectral variation from the repeat sample shown in By mean centering, the spectral contribution of the repeat sample's specific concentration is removed, leaving only the variation from non-chemical sources. Note the sharp features in the mean-centered spectra at approximately 8800 cm Table III shows the standard error of predictions (SEPs) for all the minor constituents in the prediction data set from various CLS, PLS and the present invention methods using calibration models based upon the calibration data set and the repeat sample information.
As a comparison standard, the first column of Table III provides the CVSEP obtained from a separate PLS calibration obtained from the prediction data. As expected, CLS without the information about the interferents had poor prediction ability. PLS-A in Table III represents PLS recalibration with the optimal number of PLS factors selected during the sample-out cross-validated rotation. The results show that the model was inadequate, even though this method of rotating spectra out during cross-validation has generally been recommended to avoid overfitting the data. Therefore, the cross-validated rotation was modified during recalibration so that each of the repeat spectra from the prediction set was rotated out one at a time (spectrum-out rotation). By using the spectrum-out cross-validated rotation, a more correct number of factors are selected, and the PLS-B model in Table III predicted well for these data. Cross-validated rotation removing individual spectra was preferred in this case over rotation removing all repeat spectra for a given sample since the repeat spectra were added to capture the system drift, not the component concentrations. By rotating all repeat spectra out at once, the F-test on the cross-validated calibrations did not adequately gauge the impact of the drift on the residuals for factor selection. While the spectra-out rotation worked better for this data set, it can lead to severe overfitting in other data sets. Adding several spectra for the same sample can over emphasize that sample in the model, which could lead to overfitting. Consequently, a careful analysis on a case-by-case basis is required to determine the appropriate type of spectral rotation during cross-validation for PLS. The proper choice of rotation for PLS will depend on the relative effects of the concentrations and the spectrometer drift or other sources of spectral variations that need to be modeled. The results for present invention models using 10 and 20 augmentations during the CRACLS calibration are also presented in Table III. Again, it can be seen that the present invention compares favorably with PLS. Since the CRACLS model was developed using all components and the PLS model uses only one component at a time, it could be argued that the comparison is not valid. Therefore, the results were recalibrated using the present invention with 20 augmentations and the concentrations from only one component at a time to generate three separate models. Applying these models to the prediction data set, the SEPs for glucose, ethanol and urea were 20, 4, and 4 mg/dL, respectively. Essentially, these results matched the values derived when all the components were used in the CRACLS calibration model. Clearly, either way, the present invention method effectively removed the impact of all sources of spectral variation not represented by the given/known concentrations. Notice, however, that with CRACLS there is no problem with selecting a rotation method since the mean-centered repeat spectra are simply augmented to the estimated pure-component spectra during prediction. When there is more than one known component, the concentration residual vector to use for augmentation needs to be chosen. To do this, the impact that the unmodeled components will have on each of the concentration residuals needs to be considered. From Eq. (15), the concentration residual vector is related to the projection of the unmodeled pure-component spectra, K Returning now to the simulation, to demonstrate the applicability of the various component residuals, a calibration model was developed using the simulated data, leaving out only the glucose concentrations during calibration. The concentration residuals vs. the reference concentration for the three components included in the calibration are shown in Another choice for CRACLS is selection of the number of augmentations to use in the model. When using PLS, it is important to avoid using too many factors since overfitting the concentration data will degrade the prediction model. The excessive PLS factors have concentration residuals associated with them that can degrade the concentration predictions if they are included in the model. Finding the correct number of augmentations in CRACLS, however, may be less critical than choosing the optimal number of factors for PLS. Each additional loading vector generated by the concentration residual augmentation in CRACLS results in a reduction of the NASs for the analytes of interest. After the major unmodeled sources of spectral variation have been removed, the addition of more concentration residual vectors will generate estimated pure-component spectra that represent primarily random noise. Since random spectral noise is nearly orthogonal to the pure-component spectra of the known analytes, the impact on the NAS will be minimal. The insensitivity of CRACLS to the number of augmentations was demonstrated in the example above using real data. As shown in Tables II and III, the models generated using 10 or 20 augmentations gave equivalent results with no evidence of overfitting. A sufficient number of residual vectors need to be added to capture all of the unmodeled sources of spectral variation, but adding more residual vectors did not degrade the results. For these data, the CVSEPs for CRACLS dramatically decreased with increasing augmentations as real sources of spectral variation were modeled, and then the CVSEPs just gradually continued to decrease. The apparent insensitivity of CRACLS to the number of augmentations is an advantage over PLS since, at times, it is difficult to determine the optimal number of factors for PLS. The F-test for PLS can also be used on the cross-validated calibrations to select the number of augmentations in CRACLS. This F-test often results in selecting the maximum or near the maximum number of augmentations computed since the CVSEPs often just gradually decrease with each augmentation. An important observation is that the PACLS-augmented prediction models are mathematically identical whether the estimated of measured spectral shapes representing additional unmodeled sources of spectral variation in the prediction data set are added to {circumflex over (K)} of Eq. (10) during the cross validation in the calibration phase or to
Multivariate methods use a large number of spectral channels to take advantage of the signal averaging and resolving power inherent in spectral data. The multivariate nature of the spectra and the massive amount of data in hyperspectral images allow the characterization of systems with low signal-to-noise ratios and with multiple components whose spectral features overlap. Multivariate data also provide the opportunity to correct systematic spectral artifacts and aberrations introduced by the imaging instrument. The quantitative analysis of hyperspectral image data collected from FT-IR imaging systems that employ focal plane array (FPA) detectors can be greatly hampered by the presence of system artifacts in the spectral and image data. Methods are required to eliminate the detrimental effects of these systematic artifacts if they cannot be eliminated experimentally. The characteristics of the systematic artifacts can be captured in the error covariance structure of the data estimated from repeat spectral image differences. By coupling the generalized ACLS method of the present invention with improved multivariate curve resolution (MCR) techniques and estimates of the error covariance structure of the data, the detrimental effects of the systematic artifacts present in hyperspectral imaging FT-IR systems can be greatly minimized. MCR is a constrained alternating classical least squares method (alternating between CLS calibration and CLS prediction) used when the reference variables, such as pure-component spectra or component concentrations, are inadequately known for the standard CLS method. See R. Tauler et al., “Multivariate Curve Resolution Applied to Spectral Data from Multiple Runs of an Industrial Process,” Conventional MCR methods start with a guess for either the K or C matrices, using random numbers or using other methods to arrive at a reasonable first guess, such as SIMPLISMA. See W. Windig and D. A. Stephenson, For any alternating least-squares MCR method, the calibration and prediction CLS steps can be augmented with information that it is desired for the least-squares fit to ignore (e.g., vectors representative of known or estimated spectral components or any estimate of the error covariance of the system). An estimate of the error covariance can be obtained, for example, by taking repeat spectra or, in the case of spectral imaging, repeat spectral images. Each least-squares calibration step in MCR iteration can be augmented by the scores obtained from the factor analysis of the error covariance matrix or each least-squares prediction step can be augmented by the loading vectors from the same matrix, or both steps can be augmented. Additional constraints can be applied to the non-augmented portion only of the augmented matrices. In this manner, quantitative analysis of spectral images can proceed without the need for standards even in the presence of large error covariance in the data. In general, ACLS methods will be useful in the analysis of any type of spectral data where estimates of the error covariance of the data can be obtained. In These CLS prediction and calibration steps with constraints and augmentation can be repeated in a series of iterations indicated by the circular arrow until the set of iterations converges, minimizing the constrained augmented classical least squares solution. Alternatively, the initial guess can be C with the CLS calibration being performed first. The procedure is then similar to the above-described modified MCR method with only the starting point changing. These modified MCR methods can be applied to any set of multivariate data, and the use of augmented CLS methods is demonstrated below with spectroscopic image data. To demonstrate the improvements possible with augmented CLS methods, the MCR analysis with augmentation were applied to a FT-IR spectral image obtained from a sample of neoprene thermally aged in air at elevated temperatures containing system artifacts in the image data. With the ACLS method applied to the spectral image data, augmentation with scores improved the pure-component spectral estimates while augmentation with the corresponding eigenvectors improved primarily the relative concentration maps predicted from the image data. Augmentation with both scores and loading vectors (eigenvectors) improved both the pure-component spectra and the concentration maps, even without the requirement for calibration standards. Experimental A sample of neoprene rubber was investigated. To obtain adequate signal to noise in the FT-IR image data, the neoprene sample did not contain the normal carbon filler found in many neoprene rubbers. The sample that was used to illustrate the methods of the present invention was obtained from a 2 mm×4 mm×4 mm block of neoprene that was aged for 6 days in an air furnace held at a temperature of 140° C. The sample was then potted in epoxy and the center of the sample was cut with a microtome to an approximate thickness of 15 μm. However, independent estimates of the sample thickness as a function of position on the sample indicated that the thickness varied by a factor of greater than three across the sample. The spectrometer used in this study was a BioRad Stingray step-scan FT-IR imaging spectrometer equipped with a mercury-cadmium-telluride (MCT) FPA detector that had 4096 (64×64) detector elements. The sample was placed in the macro sampling compartment that provided a 4 mm×4 mm transmission image to be obtained of the thin 2 mm×4 mm aged neoprene sample. The sample was pressed flat onto a KBr window for support. Background single-beam images were obtained from a clear region of the KBr window next to the neoprene sample. Sample single-beam transmission image spectra were then obtained of the neoprene sample. The spectra were collected at a nominal spectral resolution of 16 cm Analysis The spectral data were collected with the use of the Bio-Rad Win-IR Pro software that came with the Stingray spectrometer. However, spectral analyses were performed using the Matlab code for the MCR, ACLS, and SIMPLISMA algorithms. Small numbers of spectra from each image were clear outliers and were removed from the analysis. These outliers generally exhibited spikes or steps in the raw interferograms. The sample images were trimmed to remove all pixels representing just the KBr window and mixed pixels at the extreme boundaries of the sample that contained sample and KBr window. Only pixels present in all single-beam images were retained for final analysis. The sample spectra were ratioed to backgrounds and converted to absorbance. Large, relatively complex baseline variations were present in these spectra. The presence of these baseline variations complicated attempts to extract pure-component spectra. Therefore, the spectra were preprocessed by isolating the spectral features of the sample and performing a separate linear baseline correction under each spectral band for each spectrum in the image. When pathlength varies over the imaged area of the sample, concentration information can be confounded with pathlength changes, since pathlength and concentration changes are generally indistinguishable. These pathlength variations can be largely corrected if the Beers-Lambert law is closely followed and if a spectral region can be identified that represents only pathlength variations. Since the neoprene rubber sample thickness was not uniform over the sample area, pathlength corrections to the data were required to obtain meaningful concentration maps of the various components across the sample. To perform pathlength corrections without direct measurements of the sample thickness, a spectral band that was most representative of sample pathlength was identified. This band was selected among the six baseline-corrected bands by performing a principal components analysis (PCA) on each spectral band and examining the results carefully. Necessary conditions for a spectral band to be representative of sample pathlength are that the band be present in all pixels of the image, that it has a consistent spectral shape in all pixels, and that only multiplicative changes are present in the band. Spectral bands that fulfill these conditions of only a multiplicative change will have the first PCA eigenvector represent nearly all the variance, and the first eigenvector will have the same shape as the average spectrum of the selected band. Of the six bands identified in the neoprene sample, the 1440 cm Once an appropriate spectral band has been selected for pathlength correction, pathlength correction terms can be determined for each spectrum in the hyperspectral image. Relative pathlength for each pixel was determined by assigning a relative pathlength of one to the average spectrum of the 1440 cm The primary goal of the analysis was to obtain the pure-component spectra and concentration maps of each component in the image directly from the spectral data, since standards were not available for these samples. As described above, quantitative analysis without standards can be accomplished with the MCR algorithm using constrained alternating least squares in an iterative process. Constraints were applied to limit the range of possible solutions in the analysis and minimize the detrimental effects of the system artifacts on the analyses. These constraints comprised both non-negativity of the concentrations and the pure-component spectra, since real concentrations and spectra will consist of all positive values, and equality constraints. The MCR analysis requires a starting point for the pure spectra and/or their corresponding concentrations. In the analysis described herein, pure component spectra obtained from the SIMPLISMA algorithm were used as starting points. These pure-component spectra are generated by finding and extracting the purest spectral bands in the spectral data sets one at a time. From these purest pixels, often reasonable initial estimates of the pure-component spectra can be obtained. Random numbers can also be used as starting points. Initially, MCR was performed starting with the SIMPLISMA estimated pure-component spectra using only non-negativity constraints. The constrained alternating least squares algorithm used was a modification of the fast non-negative least squares algorithm from the previously referenced Bro et al. Changes to Bro's algorithm were made to speed the convergence of the MCR analysis. The spectral data were heavily contaminated by systematic errors, probably due to readout problems from the FPA detector, that severely degraded the quality of the both the pure-component spectra and the concentration maps generated from the MCR analysis. The result is the presence of large correlated residuals in the spectral domain and a banding artifact in the concentration maps. The standard method of applying MCR with non-negativity constraints to spectral data can be improved by augmentation of the calibration and prediction solutions with estimates of the error covariance structure in the spectra. All of the current implementations of the MCR algorithms assume that the spectral errors are random, of constant variance and independent (i.e., uncorrelated). These assumptions were violated in the FT-IR image data. The ACLS methods of the present invention enable the MCR analysis to be corrected for the presence of non-constant and correlated spectral errors in the data. To correct the MCR analysis using the ACLS methods, an estimate of the error covariance structure of the spectral image data is required. This estimate can be obtained from the repeat image spectral differences of the sample. If the sample does not move and does not change over time, then any changes in the image spectra at a given pixel are related to systematic changes in the system response as well as random noise. In the past, E In Eq. (6), TP is the product of scores and eigenvectors from the PCA of E A visible video image of the approximate portion of the aged neoprene sample imaged by the FT-IR spectrometer is displayed in As mentioned in the Experimental section, systematic artifacts were present in the FT-IR spectral image data. These artifacts were not always readily observable in the raw interferograms, single-beam spectra, or absorbance spectra. However, a PCA analysis of the spectral image data revealed the presence of the artifact as a systematic striped or banded pattern in the score images from the PCA analysis. As indicated in Not only is the artifact clearly present in the PCA score images, it is also present in the spectral data. In The non-uniform and correlated nature of the noise in the spectral images can best be observed by forming the error covariance matrix from the mean-centered repeat image spectral differences. This two-dimensional representation of the spectral error in the image is shown in FIG. The non-constant noise variance for infrared absorbance spectra is expected, since the log transformation required to obtain absorbance spectra takes noise variance that is expected to be constant at all frequencies for MCT detectors and makes the noise nonuniform over the spectral frequencies. The expected form of the noise variance of the difference of two repeat absorbance spectra at a single frequency can be estimated as outlined below.
Conventional MCR methods were applied to the spectral image data presented in Pathlength-corrected relative concentration maps representing the three pure components are presented in FIG. MCR was then applied to the same spectral image data with the concentration matrix augmented by the first five sets of scores from the mean-centered repeat image spectral differences, according to the ACLS method of the present invention. The score-augmented MCR was implemented with equality constraints on these five sets of image scores. In applying these equality constraints on the scores, the alternating classical least-squares generation of the pure-component spectra is corrected for the presence of score-related concentration errors in the image data. MCR was then applied to the same spectral image data with the first five sets of eigenvectors from the mean-centered repeat image spectral differences added to the pure-component spectral matrix. The eigenvector-augmented MCR was implemented with equality constraints on these five eigenvectors. With eigenvector augmentation only, there was some improvement in noise of the high-energy side of the hydroxyl-dominated pure component, but there was little, if any, improvement in the noise on the other pure-component spectra. However, the concentration maps exhibit dramatic improvement in the quality. The striping artifact was nearly completely removed when the eigenvectors augment the MCR, as shown in FIG. Finally, the MCR analysis was augmented with both scores and eigenvectors from the repeat image spectral differences. In this embodiment of the present invention, one simply augments with scores during the calibration phase of the MCR and augments with the eigenvectors during the prediction phase. Thus, at each iteration in the MCR procedure, any estimates related to the augmented vectors are replaced with the original eigenvectors from the PCA decomposition of the repeat image differences. The combined augmentation with scores and eigenvectors generates both improved pure-component spectra and corrected concentration maps. The pure-component spectra with scores and eigenvector augmentation are presented in FIG. Having thus described the present invention with particularity and demonstrated its utility with respect to both simulated and actual experimental data, those skilled in the art will appreciate that the present invention has broad applicability to a wide range of opportunities for analyzing multivariate spectral data and obtaining both qualitative and quantitative results. Many preprocessing procedures such as centering, scaling, path-length correction, smoothing, derivatives, multiplicative signal correction, PCA, wavelet compression, and covariance filtering have been developed that can be performed on the measured spectra when using the present invention. Patent Citations
Referenced by
Classifications
Legal Events
Rotate |