|Publication number||US7669438 B2|
|Application number||US 11/432,152|
|Publication date||Mar 2, 2010|
|Priority date||May 13, 2005|
|Also published as||CN101212998A, CN101212998B, US7644594, US20060254589, US20060254590, WO2006124578A2, WO2006124578A3, WO2006124578B1|
|Publication number||11432152, 432152, US 7669438 B2, US 7669438B2, US-B2-7669438, US7669438 B2, US7669438B2|
|Inventors||James M. Berry, Steve Morris|
|Original Assignee||Anesthetic Gas Reclamation, Llc|
|Export Citation||BiBTeX, EndNote, RefMan|
|Patent Citations (99), Non-Patent Citations (20), Referenced by (4), Classifications (14), Legal Events (2)|
|External Links: USPTO, USPTO Assignment, Espacenet|
This application is based upon provisional application 60/680,644 filed on May 13, 2005, the priority of which is claimed. On Nov. 4, 2005, Applicants filed related non-provisional application Ser. No. 11/266,966, which claims the benefit of U.S. provisional patent application 60/680,644 filed on May 13, 2005. On May 11, 2006, Applicants filed related non-provisional application Ser. No. 11/432,192, which claims the benefit of U.S. provisional patent application 60/680,644 filed on May 13, 2005. On May 11, 2006, Applicants filed related non-provisional application Ser. No. 11/432,189, which claims the benefit of U.S. provisional patent applications 60/680,644 filed on May 13, 2005 and 60/682,249 filed on May 18, 2005.
1. Field of the Invention
This invention concerns the treatment of waste anesthetic gases produced by one or more anesthesia delivery systems of a healthcare or other facility that use inhaled anesthetics for medical, dental, or veterinary purposes. In order to prevent atmospheric pollution, the invention pertains to the removal and reclamation of nitrous oxide, fluoro-ethers, and other halocarbons from a stream of waste anesthetic gases prior to its discharge to the atmosphere. In particular, this invention involves the removal and reclamation of anesthetic gases at elevated pressures, which allows the removal and reclamation process by condensation to be conducted at higher temperatures.
2. Backround Art
Anesthesia delivery systems in surgical facilities (medical, dental, and veterinary) produce significant quantities of waste anesthetic gases. Currently these gases are collected from the patients' exhalation by a dedicated or shared vacuum system. The healthcare facilities typically employ one or more centrally-located vacuum pumps to collect waste gases from individual anesthetizing locations. These vacuum pumps are usually oversized, because they are designed to collect exhaled anesthetics over a wide range of flow rates. Because these pumps operate continuously, the waste anesthetic gas suction system also entrains large amounts of surrounding room air from the anesthetizing locations, significantly diluting the waste anesthetic gases therein. At the central vacuum pump(s), the gas stream is often admixed with additional room air to further dilute it prior to its ejection from the facility. This dilute waste anesthetic gas/air mixture is typically pumped to a location outside of the surgical facility, where it is vented to the open atmosphere.
The waste anesthetic gases are generally collected at about 20-30° C. with a relative humidity ranging between 10 to 60 percent. The average composition of the waste gases is estimated to be (in volume percent): 25-32 percent oxygen, 60-65 percent nitrogen, 5-10 percent nitrous oxide, and 0.1-0.5 percent volatile halocarbons, including fluoro-ethers, such as isoflurane, desflurane and sevoflurane. The waste anesthetic gas may also contain trace amounts of lubricating oil vapor from the vacuum pumps.
An increasingly significant source of environmental concern, waste anesthetic gas halocarbons (similar in composition to Freon-12® and other refrigerants) have been linked to ozone depletion and to a lesser degree, global warming. The halocarbons used in anesthesia (primarily halogenated methyl ethyl ethers) now represent a significant emissions source, because other industrial and commercial halocarbon emissions have been greatly reduced by legislation and other initiatives in recent years. Although waste anesthetic gas emissions have so far escaped environmental regulation in the United States, legislative initiatives to strictly regulate waste anesthetic gas emissions will likely occur in the near future.
Several techniques have been proposed to treat waste anesthetic gases in an attempt to mitigate the growing problem of waste anesthetic gas emissions. For example, U.S. Pat. No. 4,259,303 describes the treatment of laughing gas with a catalyst, U.S. Pat. No. 5,044,363 describes the adsorption of anesthetic gases by charcoal granules, U.S. Pat. No. 5,759,504 details the destruction of anesthetic gases by heating in the presence of a catalyst, U.S. Pat. No. 5,928,411 discloses absorption of anesthetic gases by a molecular sieve, and U.S. Pat. No. 6,134,914 describes the separation of xenon from exhaled anesthetic gas. A cryogenic method for scrubbing volatile halocarbons from waste anesthetic gas is disclosed by Berry in U.S. Pat. No. 6,729,329, which is incorporated herein by reference.
Another cryogenic waste anesthetic gas condensation system has recently been disclosed by Berry, et al. in co-pending application Ser. No. 11/432,189, entitled “Anesthetic Gas Reclamation System and Method.” This system uses a batch-mode frost fractionation process whereby the temperatures of the individual anesthetic gases are lowered to a point such that they condense and collect as frost on the cooling surfaces of a cold trap/fractionator. This co-pending application, filed on May 11, 2006, is incorporated herein by reference.
The current methods for scavenging waste anesthetic gases from anesthetizing locations 15A, 15B, 15C in healthcare facilities generally involve drawing high flows of room air into the dedicated or shared vacuum collection manifold 16 to entrain waste anesthetic gases. The collection manifold 16 may also continuously draw in air through a number of idle anesthetizing machines 12A, 12B, 12C. On average, the collection system manifold 16 extracts between 20-30 liters of waste anesthetic gas and/or room air per minute at each anesthetizing location 15A, 15B, 15C. For a large hospital having between 20-30 operating rooms, it is estimated that waste anesthetic reclamation system 10 flow rate ranges between 500-1000 l/min. (14-35 scf/min.).
The advantages of a high-flow dilute waste gas system are that the system easily accommodates a wide range of anesthetic exhaust flows, the system is safer because little anesthetic can escape the system, and the system is more trouble-free because little maintenance is required. However, high-flow systems are energy-intensive, generally requiring large vacuum pumps 20 in order to maintain sufficient suction at a large number of anesthetizing stations 15A, 15B, 15C. For example, in order to maintain a vacuum of about 200 mm Hg at a flow rate of 1-2cubic feet per minute (cfm) at each anesthetizing station 15A, 15B, 15C, vacuum pumps of 100-200 cfm capacity are not uncommon.
Additionally, a diluted waste anesthetic gas stream is thermally inefficient. Removal of a waste gas component by condensation requires lowering the temperature of the entire flow stream to a point where the partial pressure of the gaseous waste component is equal to or greater than its saturated vapor pressure (at that temperature). Therefore, to cool the large volume of diluted waste anesthetic gas to a temperature below the saturated vapor pressure of its components, a sizeable cooling utility (i.e. a greater quantity of liquid oxygen, liquid nitrogen, etc.) is required. A method and system for increasing the efficacy and efficiency of condensation-type waste anesthetic gas scavenging and reclamation systems are thus desirable.
A low-flow scavenging system provides a more efficient means of waste anesthetic gas recovery through condensation, because a smaller volume of gas has to be cooled to the condensation temperatures of the individual gases. A low flow scavenging method, facilitated by a dynamic waste anesthetic gas collection apparatus, has recently been disclosed by Berry et al. in co-pending application Ser. No. 11/266,966, entitled “Method of Low Flow Anesthetic Gas Scavenging and Dynamic Collection Apparatus Therefor.” This co-pending application, filed on Nov. 4, 2005, is incorporated herein by reference.
Typically, anesthetic gases are highly volatile substances. For a given temperature, they have a higher vapor pressure than the vapor pressure of water and other lower volatile substances. Substances with higher vapor pressures generally require greater cooling to achieve the same or similar condensate recovery as substances with lower vapor pressures. Thus, anesthetic gases need to be cooled to extremely low temperatures, i.e. cryogenic temperatures, in order to recover appreciable amounts of anesthetic as condensate. However, these extremely low temperatures approach, and in many cases, fall below the freeze point of many anesthetics. In such situations, the waste anesthetic gas stream may still contain anesthetic concentrations that could be condensed except for the undesirable freezing of the system.
Pressure, in addition to temperature, can greatly influence condensation. Elevating the pressure of the condensation system is advantageous, because it allows condensation to occur at significantly higher temperatures than would otherwise occur at lower operating pressures. This also avoids the risk and problems associated with freezing of the condensate. For these types of vapor/liquid phase equilibrium systems, the most beneficial thermodynamic characteristic is that pressure has a much larger effect on the dew point of the vapor than the freezing point of the liquid. Thus, the dew point temperature of a typical anesthetic-laden vapor stream increases with increasing pressure while its freezing point temperature stays relatively constant for varying system pressures.
The increased temperature span between the dew point of the vapor and the freeze point of the condensate, due to increases in system pressure, provides greater operational flexibilities for condensation systems. For example, less cryogenic refrigerant is needed to effect the same amount of condensation, because condensation can occur at higher temperatures. Furthermore, if a more complete separation of the anesthetic from the waste gas stream is desired, the system temperature can be lowered while maintaining an elevated pressure. This permits additional anesthetic to be condensed from the vapor phase without the associated risk of condensate freezing. Thus, a strategy may be developed to achieve the optimum separation of anesthetic by simply adjusting the condensation system pressure relative to the condensation system temperature. Of course, the relative refrigeration versus compression costs should also be considered in any cost optimization strategy.
3. Identification of Objects of the Invention
A primary object of the invention is to provide an economical system and method for removing fluoro-ethers, nitrous oxide, and other volatile halocarbons from waste anesthetic gases from a surgical or other healthcare facility before such gases are vented to the atmosphere.
Another object of the invention is to provide an economical system and method for substantially preventing atmospheric venting of fluoro-ethers and other volatile halocarbons of waste anesthetic gas while eliminating the need of prior art catalysts, charcoal granules and heating techniques.
Another object of the invention is to provide a system and method which reclaims and allows re-distillation and/or reuse of a large percentage of the nitrous oxide and/or anesthetic halocarbons used in the facility.
Another object of the invention is to provide an economical system and method which utilizes and enhances existing waste anesthetic gas reclamation systems of healthcare facilities for minimal impact and cost.
Another object of the invention is to provide an economical system and method which utilizes existing liquid oxygen and/or liquid nitrogen storage and delivery systems of healthcare facilities for energy efficiency and minimal impact during the reclamation system installation.
Another object of the invention is to provide an economical system and method for separating various removed nitrous oxide, fluoro-ethers, and other volatile halocarbon components based on their characteristic bubble and dew points.
Another object of the invention is to provide an economical system and method for increasing the efficacy and efficiency of condensation-type waste anesthetic scavenging systems.
Another object of the invention is to provide a flexible system and method for increasing the efficacy and efficiency of condensation-type waste anesthetic scavenging system by operating the system under varying pressures and temperatures.
Other objects, features, and advantages of the invention will be apparent to one skilled in the art from the following specification and drawings.
The objects identified above, as well as other advantages and features, are preferably embodied in a system and method for the removal of nitrous oxide and volatile halocarbon gas components from waste anesthetic gases using one or more compression stages to elevate the pressure of the waste anesthetic gas stream prior to anesthetic reclamation by condensation. The waste anesthetic gas stream at elevated pressure is cooled in a condenser such that the temperature of the nitrous oxide and other anesthetic halocarbons are lowered to a point where the vapors condense as a removal liquid or collect as frost on the cooling surfaces of the condenser. In other words, to recover the anesthetic components from the effluent gas, the nitrous oxide and halocarbon components in the waste anesthetic gas are compressed and cooled to either condense them into a removable liquid condensate or solidify them onto the cooling coil surfaces of a heat exchanger/condenser. Whether the anesthetic gas components condense as a liquid or deposit as a solid depends on the operational temperature and pressure of the condenser/heat exchanger.
Compression of the waste anesthetic gas to a level above atmospheric pressure is advantageous, because the higher pressure essentially elevates the temperature at which saturation and condensation of the anesthetic gas can occur. Thus, compression of the gas above atmospheric pressure allows the same fraction of anesthetic to be removed by condensation at a higher temperature as would otherwise have occurred by condensation at atmospheric pressure and at a lower temperature. Moreover, a greater fraction of anesthetic may be condensed from the vapor phase as the temperature of the compressed waste anesthetic gas is lowered from this higher temperature. A strategy may be developed to achieve the optimum separation of anesthetic by simply manipulating the condensation system pressure relative to the condensation system temperature. Furthermore, energy and cost saving may be possible when the relative refrigeration versus compression costs are factored into the strategy.
In a preferred embodiment of the invention, a compressor unit consisting of one or more compression stages is located within the waste anesthetic gas scavenger system between the waste anesthetic gas collection unit and the condensation unit. The compressor unit is sized to compress the anesthetic waste gas from the collection unit to a pressure up to 50 psig for subsequent treatment in a condensation system using a refrigerant, i.e. liquid oxygen, liquid nitrogen, etc., supplied by a hospital or other medical, dental, or veterinary facility. In an alternative embodiment, the waste anesthetic gas stream is compressed to pressures well above 50 psig to take advantage of attendant increases in separation efficiency and fractional extraction. However, in this alternative embodiment, a separate refrigerant supply for the condenser is desirable in order to avoid the risk of contaminating the healthcare facility's gas supplies should an internal leak within the condenser occur.
In a preferred embodiment, the compressed waste anesthetic gas is passed through a multi-stage condenser/heat exchanger wherein heat from the waste anesthetic gas stream is exchanged with the liquid refrigerant. In the first condenser stage, any water vapor in the waste anesthetic gas stream is condensed and extracted. In subsequent condenser stages, the temperature of the compressed gas stream is reduced to a point where the partial pressure of each gaseous waste component is equal to or greater than its saturated vapor pressure (at that temperature and elevated pressure). At atmospheric pressure, the anesthetics are extracted from the waste gas into their purified components most efficiently at temperatures near their individual freezing points. However, separation of the anesthetic vapor mixture is achieved at temperatures higher than their individual freezing points when the waste anesthetic gas stream is condensed at elevated pressure. After condensation and extraction of the anesthetic gases into their purified components, the remainder of the anesthetic gas is vented to atmosphere. However, in a preferred embodiment, the waste gas stream is passed through an expansion valve to further cool the gas via the Joule-Thompson effect. This may also induce additional condensation of anesthetic components from the waste gas. More preferably, the waste gas stream is passed through a small turbine, which recovers the potential energy of compression and may induce additional condensation.
The invention is described in detail hereinafter on the basis of the embodiments represented in the accompanying figures, in which:
As shown in
In a preferred embodiment, the compressor 42 is sized to compress the anesthetic waste gas from the collection units 15A, 15B, 15C to a pressure up to 50 psig for subsequent treatment in a condensation unit 22. Pressures above 50 psig are preferable in order to take advantage of attendant increases in separation efficiency and fractional extraction. Multistage compressors are used to avoid the problems associated with high compression ratios, such as high discharge temperatures and increased mechanical breakdowns. As a result, compressor manufacturers recommend a compression ratio of no more than 10:1, especially for low-temperature applications. Multistage compressors can also be more economical than single stage compressors because of the attendant power cost savings attributable to compression stages having smaller compression ratios. However, the compressor 42 of system 200 needs only a single compression stage, because a compression ratio of no more than 10:1 is anticipated.
The condenser 22 preferably uses a liquid oxygen, liquid nitrogen, or similar refrigerant obtained from the common supply of these liquefied gases normally available at a hospital or other medical, dental, or veterinary facility. If the waste anesthetic gas is compressed above the facility's gas supply pressure (e.g. 50 psig), then contamination of the common refrigerant supply with waste anesthetic is possible should an internal leak occur within the condenser unit 22. In an alternative preferred embodiment, the waste anesthetic gas stream is compressed to pressures well above 50 psig to take advantage of the attendant increases in separation efficiency and fractional extraction. For compression above 50 psig, however, a separate supply of liquid oxygen, liquid nitrogen, or similar refrigerant, is recommended in order to avoid the risk of contaminating the common gas supplies of the healthcare facility with waste anesthetic should an internal leak occur within the condenser unit 22.
After compression, the waste anesthetic gas flows through a collection vessel or receiver 26 which allows any liquid condensed due to compression to be removed and separated from the compressed waste anesthetic gas stream. Prior to condensation recovery of the anesthetic components, any water vapor in the gas stream should be removed to prevent freezing of the liquid water condensate in the condenser 22. A preferred method to remove water vapor from the waste anesthetic gas stream is to use a first condenser stage 222A (
The compressed waste anesthetic gas stream is then cooled in a single or multiple stage condenser 22 such that the temperature of the nitrous oxide and other anesthetic halocarbons are lowered to a point where the vapors either condense as a removal liquid on the condenser coils 236B (
After the anesthetic components are removed through condensation, the remaining waste gas (mainly composed of entrained air) may be vented to the atmosphere 46. Preferably, the compressed waste gas is first passed through an expansion valve 43 and a receiver 45 prior to atmospheric venting 46. The expansion valve 43 reduces the compressed waste gas to atmospheric pressure and further cools the compressed waste gas via the Joule-Thompson effect. Any additional anesthetic components in the waste gas may be condensed through Joule-Thompson adiabatic expansion. These anesthetic condensates are collected in the receiver 45 prior to the atmospheric discharge 46 of the waste gas. More preferably, however, the compressed waste gas is first throttled through a small turbine 44 (
Moreover, prior to atmospheric discharge, heat integration of the cooled waste gas with streams to be cooled may reduce the overall cooling utility of the method and system. For example, compression of the waste anesthetic gas stream causes the temperature of the gas stream to increase. The cooled waste gas stream to be vented 46 could be used to cool this compressed waste anesthetic gas stream prior to condensation in order to reduce the overall refrigerant requirement of the heat exchanger/condenser 22.
Berry discloses two cryogenic methods for recovering volatile halocarbons from waste anesthetic gas. First, U.S. Pat. No. 6,729,329 discloses the use of liquid oxygen to condense anesthetic gas components into recoverable liquid condensates.
A condenser unit 22 is provided which includes first and second condensers 222A and 222B. The outlet line 221 for liquid oxygen from supply tank 220 is fluidly connected to the condensing coils 236B of the second vessel 222B. The outlet of condensing coils 236B is fluidly connected via flow line 225 to the inlet of coils 236A of first vessel 222A. The outlet of coils 236A is fluidly connected via flow line 227 to valve 214 and flow lines connected thereto (not shown) of the healthcare facility.
A flow line 239 connects the waste anesthetic gas flow lines from receiver 26 (
The cooled compressed gas near the bottom of vessel 222A is conducted via flow line 241 to the top or entrance of heat exchanger/condenser 222B where it is applied at a temperature greater than 0° C. The cooled compressed gas applied to the top of heat exchanger/condenser 222B passes over coils 236B wherein it exchanges heat with the liquid oxygen flowing countercurrently through the coils 236B. The oxygen from flow line 221 enters the coils 236B at a temperature of approximately −150° C. and leaves the coils 236B via flow line 225 at an increased temperature. If necessary, an intermediate bypass valve 235 may be provided in line 221 to bring the temperature in line 225 at the inlet of coils 236A to approximately 0° C. The temperature of the compressed waste anesthetic gas from flow line 241 is lowered while passing over the coils 236B such that the halocarbons of the waste gas are liquefied and discharged into a collection tank 24. The remainder of the compressed waste gas, i.e., those components which are not harmful to the atmosphere, are vented to the atmosphere via flow line 46, throttled through an expansion valve 43 (
Second, co-pending application Ser. No. 11/432,189, entitled “Anesthetic Gas Reclamation System and Method,” discloses the use of a batch-mode frost fractionation process whereby the temperatures of the individual anesthetic gases are lowered to a point such that they collect as frost on the cooling surfaces of a cold trap/fractionator. The cold trap/fractionator is periodically cycled through a thawing stage, during which the cooling surfaces, caked with frost gas components deposited from the waste anesthetic gas passing thereby, are gently warmed to sequentially separate and collect the trapped components.
As shown in
A flow line 139 connects the waste anesthetic gas flow lines from receiver 26 (
This countercurrent heat exchanger arrangement results in a temperature gradient where the top of the cold trap/fractionator 125 is the warmest and where the bottom of the cold trap/fractionator 125 is the coldest. The upper region 160 of the cooling coils 136 of the cold trap/fractionator 125 cools the compressed waste anesthetic gas to a temperature of approximately −5° C. to extract water vapor as frost on the coils 136. The upper middle region 162 of the cooling coils 136 next cools the compressed waste anesthetic gas to a temperature of approximately −60° C. which allows sevoflurane to condense and solidify onto the coils 136. Next, the lower middle region 163 extracts nitrous oxide by condensation and solidification at a temperature of approximately −90° C., and finally the lower region 164 of the cooling coils 136 extracts isoflurane and desflurane by condensation and solidification onto the coils 136 at the lowest temperature (between approximately −100° C. and −110° C.). Alternatively, if heat exchanger/condenser 125 is operated under low pressure (i.e. vacuum pressure), then the anesthetic components may be desublimated/deposited directly onto coils 136 without first entering a liquid phase. The remainder of the compressed waste anesthetic gas, i.e., those components which are not harmful to the atmosphere, are vented to the atmosphere via flow line 46, throttled through an expansion valve 43 (
The cold trap/fractionator 125 is periodically cycled through a thaw process in order to defrost the cooling coils 136. Thawing of coils 136 is effectuated by reducing or prohibiting the flow of liquid oxygen therethrough by thermostatic control valve 133. This allows the cold trap/fractionator 125 to warm to room temperature through heat transfer with its ambient surroundings. In an alternate embodiment, warmed oxygen from heat exchanger 122 may be partially or completely directed through the cooling coils 136 by simultaneously opening valve 159 and closing valves 133, 154. In yet a third embodiment, another fluid (not shown) may be directed through cooling coils 136 to achieve a controlled thaw.
A funnel-shaped hopper 157 forms the lowest point of heat exchanger/condenser 125 and preferably drains into a 4-way selector valve 158, which in turn is fluidly coupled to anesthetic collection tanks 24A, 24B and a water collection tank 23. As the temperature of the coils 136 increases above approximately −100° C. during the thawing stage, desflurane (melting point of approximately −108° C. at atmospheric pressure) and isoflurane (melting point of approximately −103° C. at atmospheric pressure) melt from the lower region 164 of cold trap/fractionator 125 and collect in the hopper 157. Selector valve 158 is concurrently aligned to allow the liquid desflurane and isoflurane to gravity feed into one of the collection tanks 24A, 24B. As the cold trap/fractionator continues to warm above −90° C., the trapped nitrous oxide melts from the lower middle region 163 of cold trap/fractionator 125 and collects in the hopper 157. Selector valve 158 is concurrently aligned to allow the liquid nitrous oxide to gravity feed into one of the collection tanks 24A, 24B. As the temperature warms further still above −65° C., sevoflurane (melting point of approximately −67° C. at atmospheric pressure) melts from the upper middle region 162 of the cooling coils 136 and collects in the hopper 157. Selector valve 158 is concurrently aligned to allow the liquid sevoflurane to gravity feed into one of the collection tanks 24A, 24B. Likewise, as the cold trap/fractionator 125 continues to warm, the water vapor frost will melt above 0° C. from the upper region 160 and be channeled by selector valve 158 into the water collection tank 23. By this method, the fluoro-ethers are fractionated as they are removed from the waste anesthetic gas.
The waste anesthetic gas collection manifold 16 operates at a slight vacuum pressure, e.g. 5 cm, which is generated by compressor 42. Therefore, isolating the collection manifold 16 from entraining room air when no waste anesthetic gas is being produced reduces the average anesthetic scavenging flow by approximately 90 percent and subsequently reduces the necessary capacity of the compressor 42, heat exchanger/condenser 22, piping, and associated other hardware. For a large hospital having between 20-30 operating rooms, it is estimated that waste anesthetic gas flow rate of 500-1000 l/min with the prior art reclamation system 10 of
From vacuum manifold 16, the collected waste gas stream is passed through a check valve 35 to compressor 42. Compressor 42 has a single compression stage sized to elevate the pressure of the anesthetic waste gas from collection units 30A, 30B, 30C to a pressure above atmospheric pressure for subsequent treatment in a condensation unit 22. After compression, the waste anesthetic gas flows through a collection vessel or receiver 26 which allows any liquid condensed due to compression to be removed and separated from the compressed waste anesthetic gas stream. The compressed waste anesthetic gas stream is then cooled in a multi-stage condenser 22 such that the temperature of the nitrous oxide and other anesthetic halocarbons are lowered to a point where the vapors condense on the condenser coils 236B as a removal liquid (see disclosure with respect to
As previously disclosed, the compressed waste gas is preferably first throttled through a small turbine 44 or similar device prior to atmospheric release 46 in order to capture the potential energy of the compressed waste gas. The captured energy may then be used to power the compressor 42 or supply other energy requirements of the method and system. Reducing the pressure of the compressed work anesthetic gas stream through turbine 44 may induce additional condensation of anesthetic components. Therefore, receiver 45 is provided to collect these anesthetic condensate prior to atmospheric discharge 46 of the remaining waste gas.
The Abstract of the disclosure is written solely for providing the United States Patent and Trademark Office and the public at large with a means by which to determine quickly from a cursory inspection the nature and gist of the technical disclosure, and it represents solely a preferred embodiment and is not indicative of the nature of the invention as a whole.
While some embodiments of the invention have been illustrated in detail, the invention is not limited to the embodiments shown; modifications and adaptations of the above embodiment may occur to those skilled in the art. Such modifications and adaptations are in the spirit and scope of the invention as set forth herein:
|Cited Patent||Filing date||Publication date||Applicant||Title|
|US575714||Dec 28, 1893||Jan 26, 1897||Christian heinzerling|
|US1040886||Mar 5, 1909||Oct 8, 1912||Air Liquide||Process of recovery of volatile liquids.|
|US3348538||Nov 5, 1964||Oct 24, 1967||Scott Aviation Corp||Breathing apparatus exhalation valve with suction control|
|US3517521||Jan 24, 1968||Jun 30, 1970||Us Interior||Method and apparatus for separating neon from a mixture of gases|
|US3592191||Jan 17, 1969||Jul 13, 1971||Jackson Richard R||Recovery of anesthetic agents|
|US3714942||Feb 3, 1969||Feb 6, 1973||Sub Marine Syst Inc||Cryogenic gas processing system|
|US3721239||Jul 9, 1969||Mar 20, 1973||R Myers||Anesthetic gas exhaust system|
|US3800793||Dec 23, 1971||Apr 2, 1974||R Marrese||Anesthesia apparatus having negative pressure relief means|
|US3867936||Jan 11, 1974||Feb 25, 1975||Shirley Aldred & Co Ltd||Anaesthetic gas safety equipment|
|US3941573||May 2, 1974||Mar 2, 1976||James Frederick Chapel||Apparatus for removing anesthetic gases|
|US4004585||Nov 18, 1975||Jan 25, 1977||Boehringer John R||Safety interface for anesthesia vacuum scavenging|
|US4127163||Aug 18, 1976||Nov 28, 1978||Bechtel International Corp.||Freezer-sublimer for gaseous diffusion plant|
|US4180066||Oct 20, 1977||Dec 25, 1979||Vital Signs, Inc.||Safety interface and valve for anesthesia gas scavenging|
|US4181508||Nov 30, 1977||Jan 1, 1980||Gesellschaft Fur Kernforschung M.B.H.||Method and apparatus for separating desublimatable components from gas mixtures|
|US4205095||Nov 6, 1978||May 27, 1980||Barr Iain G||Glycerides|
|US4219020||Aug 9, 1978||Aug 26, 1980||Fraser Sweatman, Inc.||Scavenger valve attachment for inhalation sedation system mask|
|US4246015||Dec 31, 1979||Jan 20, 1981||Atlantic Richfield Company||Freeze-wash method for separating carbon dioxide and ethane|
|US4259303||Oct 15, 1979||Mar 31, 1981||Kuraray Co., Ltd.||Method of and system for treating waste anesthetic gas|
|US4261178||Jan 19, 1979||Apr 14, 1981||Robinair Manufacturing Corporation||Environmental protection refrigeration disposal and charging system|
|US4265239||Nov 27, 1978||May 5, 1981||Fischer Jr Charles M||Gas scavenging exhaust system|
|US4281518||Jan 21, 1980||Aug 4, 1981||Messerschmitt-Bolkow-Blohm Gmbh||Method and apparatus for separating particular components of a gas mixture|
|US4291689||Oct 19, 1979||Sep 29, 1981||Airco, Inc.||Evacuation manifold for medical anesthesia circuits|
|US4312339||Mar 31, 1980||Jan 26, 1982||Porter Instrument Co., Inc.||Device for administering an anesthetic gas|
|US4378984 *||Aug 25, 1981||Apr 5, 1983||Cheng Chen Yen||Distillative freezing process for separating volatile mixtures|
|US4447462||Jul 18, 1983||May 8, 1984||The Procter & Gamble Company||Structural fat and method for making same|
|US4451273||Mar 31, 1983||May 29, 1984||Cheng Chen Yen||Distillative freezing process for separating volatile mixtures and apparatuses for use therein|
|US4527558||Jun 24, 1983||Jul 9, 1985||The Boc Group, Inc.||Scavenger system|
|US4538605||Mar 8, 1983||Sep 3, 1985||Gambro Engstrom Ab||Anesthetic apparatus|
|US4609388||May 24, 1985||Sep 2, 1986||Cng Research Company||Gas separation process|
|US4633890||Dec 4, 1984||Jan 6, 1987||Edward Carden||Anaesthetist's gas pollution inhibitor|
|US4653493||Feb 8, 1985||Mar 31, 1987||Hoppough John M||Ventilator unit exhalation contamination control device|
|US4676239||Jul 17, 1986||Jun 30, 1987||David Humphrey||Anesthetic system|
|US4755201||Apr 17, 1986||Jul 5, 1988||Messer. Griesheim Gmbh||Process for removing lighter volatile impurities from gases|
|US4768347||Nov 4, 1987||Sep 6, 1988||Kent-Moore Corporation||Refrigerant recovery and purification system|
|US4832042||Aug 19, 1987||May 23, 1989||Emory University||Ventilator hood system for indirect calorimetry|
|US4895172||Jun 4, 1987||Jan 23, 1990||Lindkvist Allan Erik||Gas collection device|
|US4905685||Apr 13, 1988||Mar 6, 1990||Siemens Aktiengesellschaft||Inhalation anaesthesia equipment|
|US4928685||Oct 5, 1988||May 29, 1990||Cairns & Brother Inc.||Closed-circuit positive pressure breathing apparatus with pneumatically operated storage chamber|
|US4949714||Jul 26, 1989||Aug 21, 1990||Viratek Inc.||Scavenging medical hood|
|US5033464||Sep 20, 1989||Jul 23, 1991||Delcatilho Raymond A||Combination oral suction and scavenging apparatus|
|US5044361||Apr 12, 1988||Sep 3, 1991||Zenova Aktiebolag||Method and apparatus for reuse of anesthetics|
|US5044363||May 24, 1990||Sep 3, 1991||The United States Of America As Represented By The Department Of Health And Human Services||Adsorption system for scavenging anesthetic agents from waste gas released during surgical activity|
|US5046491||Oct 17, 1990||Sep 10, 1991||Derrick Steven J||Apparatus and method for respired gas collection and analysis|
|US5046492||Jul 15, 1988||Sep 10, 1991||Stackhouse Wyman H||Clean room helmet system|
|US5062270||Aug 31, 1990||Nov 5, 1991||Exxon Production Research Company||Method and apparatus to start-up controlled freezing zone process and purify the product stream|
|US5152812||May 10, 1991||Oct 6, 1992||Kovach J Louis||Recovery of condensable organic compounds from inert gas streams laden|
|US5205843||Feb 14, 1992||Apr 27, 1993||Membrane Technology And Research, Inc.||Process for removing condensable components from gas streams|
|US5253641||Oct 3, 1991||Oct 19, 1993||Choate Thomas V||Respiratory filter apparatus and method|
|US5311862||May 4, 1992||May 17, 1994||Blasdell Richard J||Inhalation apparatus|
|US5323623||Jun 25, 1993||Jun 28, 1994||Nestec S.A.||Recovery of aroma gases|
|US5339642||Mar 11, 1993||Aug 23, 1994||Spx Corporation||Refrigerant recovery to multiple refrigerant storage containers|
|US5345928||Feb 6, 1991||Sep 13, 1994||Medicvent Ab||System arrangement for the evacuation of anaesthesia or analgesia gas|
|US5370110||Aug 31, 1992||Dec 6, 1994||Children's Medical Center Corporation||Anesthetic scavenging hood|
|US5398675||Oct 14, 1992||Mar 21, 1995||Henkin; Melvyn L.||Anesthesia rebreathing system|
|US5419317||Apr 1, 1994||May 30, 1995||Accutron||Inhalation apparatus|
|US5450728 *||Nov 30, 1993||Sep 19, 1995||Air Products And Chemicals, Inc.||Recovery of volatile organic compounds from gas streams|
|US5482033||Feb 8, 1994||Jan 9, 1996||Engle; John R.||Anesthetic waste gas evacuation system|
|US5507282||Feb 4, 1994||Apr 16, 1996||University Of Manitoba||Control of airway pressure during mechanical ventilation|
|US5520169 *||Mar 20, 1995||May 28, 1996||Dragerwerk Aktiengesellschaft||Anesthesia arrangement for recovering gaseous anesthetic agents|
|US5568910||Mar 2, 1995||Oct 29, 1996||Delmarva Laboratories, Inc.||Anesthesia machine|
|US5676133||Jun 4, 1996||Oct 14, 1997||Apotheus Laboratories, Inc.||Expiratory scavenging method and apparatus and oxygen control system for post anesthesia care patients|
|US5678540||Mar 23, 1995||Oct 21, 1997||Siemens Elema Ab||Breathing gas system|
|US5694924||Oct 10, 1996||Dec 9, 1997||Siemens-Elema Ab||Anesthetic administration system with active regulation of the volume of the gas reservoir during a breathing cycle|
|US5715813||Jan 31, 1994||Feb 10, 1998||Guevrekian; Lawrence||Capture system for waste anesthetic gas|
|US5740682||Jan 23, 1995||Apr 21, 1998||Ram Lavie||Method for the recovery of organic vapors|
|US5759504||Dec 18, 1995||Jun 2, 1998||Hitachi, Ltd.||Method for treating organohalogen compounds with catalyst|
|US5769072||Mar 18, 1997||Jun 23, 1998||Siemens Elema Ab||Anesthetic system with multiple valve-controlled bellows|
|US5819555 *||Aug 19, 1996||Oct 13, 1998||Engdahl; Gerald||Removal of carbon dioxide from a feed stream by carbon dioxide solids separation|
|US5928411||Jul 21, 1997||Jul 27, 1999||Dragerwerk Ag||Device for absorbing a gas mixture consisting of laughing gas and anesthetic vapor|
|US6030591||Jan 2, 1998||Feb 29, 2000||Atmi Ecosys Corporation||Process for removing and recovering halocarbons from effluent process streams|
|US6076524||Jun 23, 1994||Jun 20, 2000||Children's Medical Center Corporation||Anesthetic scavenging hood|
|US6080226||Sep 30, 1998||Jun 27, 2000||Uop Llc||Nitrous oxide purification by pressure swing adsorption|
|US6082133||Feb 5, 1999||Jul 4, 2000||Cryo Fuel Systems, Inc||Apparatus and method for purifying natural gas via cryogenic separation|
|US6131571||Apr 30, 1997||Oct 17, 2000||University Of Florida||Ventilation apparatus and anesthesia delivery system|
|US6134914 *||Jul 19, 1997||Oct 24, 2000||Messer Griesheim Gmbh||On-line recovery of xenon from anaesthetic gas|
|US6158434||Feb 27, 1996||Dec 12, 2000||Henk W. Koster||Ventilatory system with additional gas administrator|
|US6206002||May 13, 1999||Mar 27, 2001||Hudson Rci Ab||Device and method for recovering anaesthetic during the use of inhaled anaesthetics|
|US6237596||Nov 8, 1996||May 29, 2001||George L. Bohmfalk||Disposable mask and suction catheter|
|US6328036||Apr 15, 1999||Dec 11, 2001||Siemens-Elema Ab||Anesthetic apparatus and method for operating same|
|US6357437||Feb 11, 2000||Mar 19, 2002||Vortex Recoveries Inc.||Waste gas recovery apparatus|
|US6374635||Oct 27, 1999||Apr 23, 2002||Ebara Corporation||PFC gas recovery process and apparatus|
|US6405539||Nov 3, 1998||Jun 18, 2002||Pneumatic Berlin Gmbh||Method and device for recovering gases|
|US6475266||Mar 16, 2001||Nov 5, 2002||Fuji Photo Film Co., Ltd.||Exhaust gas recovery method and apparatus|
|US6488028||Oct 10, 1996||Dec 3, 2002||Hudson Rci Ab||Device for recovering anaesthetic|
|US6490883||Jun 12, 2001||Dec 10, 2002||Air Products And Chemicals, Inc.||Process and apparatus for removal of volatile compounds from process gases|
|US6513345||Dec 29, 1999||Feb 4, 2003||Shell Oil Company||Nozzle for supersonic gas flow and an inertia separator|
|US6536430||Oct 10, 2000||Mar 25, 2003||Charles A. Smith||Portable anesthesia rebreathing system|
|US6729329 *||Aug 6, 2001||May 4, 2004||James M. Berry||System for removal of halocarbon gas from waste anesthetic gases|
|US6736140||Mar 13, 2003||May 18, 2004||Matrx Medical Inc.||Exhalation evacuator|
|US6776158||Jul 23, 2002||Aug 17, 2004||Euthanex Corporation||System for anesthetizing laboratory animals|
|US6863067||Nov 14, 2003||Mar 8, 2005||Maquet Critical Care Ab||Device for exchange of gas volume|
|US20030185735 *||Sep 27, 2001||Oct 2, 2003||Masatoshi Hotta||Process and apparatus for treating waste anesthetic gas|
|US20050155380 *||Feb 5, 2004||Jul 21, 2005||Michael Rock||Anesthetic agent recovery|
|US20060254586||Nov 4, 2005||Nov 16, 2006||Anesthetic Gas Reclamation, Llc||Method of low flow anesthetic gas scavenging and dynamic collection apparatus therefor|
|US20060254587||May 11, 2006||Nov 16, 2006||Anesthetic Gas Reclamation, Llc||Anesthetic gas reclamation system and method|
|US20060254590||May 11, 2006||Nov 16, 2006||Anesthetic Gas Reclamation, Llc||Method and apparatus for self-contained anesthetic gas reclamation|
|USRE36460||May 1, 1998||Dec 21, 1999||Life Science Holdings, Inc.||Method of providing circulation via lung expansion and deflation|
|WO1998008583A1||Jul 19, 1997||Mar 5, 1998||Messer Griesheim Gmbh||On-line recovery of xenon from anaesthetic gas|
|WO2001024858A2||Sep 19, 2000||Apr 12, 2001||Messer Griesheim Gmbh||Gas collection device|
|1||Applicant's Response to Final Office Action and Terminal Disclaimer Filed Aug. 4, 2009 in connection with U.S. Appl. No. 11/432,192.|
|2||Applicant's Response to First Office Action filed Apr. 9, 2009 in connection with U.S. Appl. No. 11/432,189.|
|3||Applicant's Response to First Office Action filed Apr. 9, 2009 in connection with U.S. Appl. No. 11/432,192.|
|4||Applicant's Response to First Office Action filed Nov. 17, 2008 in connection with U.S. Appl. No. 11/266,966.|
|5||Applicant's Response to Non-Final Office Action filed Jun. 1, 2009 in connection with U.S. Appl. No. 11/266,966.|
|6||Applicant's Supplemental Response to First Office Action filed Apr. 10, 2009 in connection with U.S. Appl. No. 11/432,189.|
|7||Applicant's Supplemental Response to First Office Action filed Apr. 10, 2009 in connection with U.S. Appl. No. 11/432,192.|
|8||Brown AC, Canosa-Mas CE, Parr AD, et al.: Tropospheric lifetimes of halogenated anaesthetics. Nature 1989; 341: 635-637.|
|9||Examiner's Final Office Action mailed May 4, 2009 in connection with U.S. Appl. No. 11/432,192.|
|10||Examiner's First Office Action mailed Jul. 17, 2008 in connection with U.S. Appl. No. 11/266,966.|
|11||Examiner's First Office Action mailed Oct. 10, 2008 in connection with U.S. Appl. 11/432,189.|
|12||Examiner's Second Office Action mailed Dec. 30, 2008 in connection with U.S. Appl. No. 11/266,966.|
|13||Langbein T, Sonntag H, Trapp D, et al.: Volatile anaesthetics and the atmosphere: atmospheric lifetimes and atmospheric effects of halothane, enflurane, isoflurane, desflurane and sevoflurane. Br J Anaesth 1999; 82: 66-73.|
|14||McCulloch, A.: Letter to Editor regarding Langbein, et al. 1999 paper. Br J Anaesth 2000; 84 (4): 534-36.|
|15||Notice of Allowance mailed Aug. 24, 2009 in connection with U.S. Appl. No. 11/432,192.|
|16||Notice of Allowance mailed Jul. 23, 2009 in connection with U.S. Appl. No. 11/266,966.|
|17||Notice of Allowance mailed May 20, 2009 in connection with U.S. Appl. No. 11/432,189.|
|18||R. F. Dunn, M. Zhu, B.K. Srinivas and M. M. El-Halwagi (1995), Optimal Design of Energy-Induced Separation Networks for VOC Recovery, AIChE Symp. Ser., 90(303), 74-85, NY: AIChE.|
|19||U.S. Appl. No. 11/266,966, filed Nov. 4, 2005, Berry et al.|
|20||Written Opinion of International Search Authority for PCT/US2006/18416 mailed on Sep. 24, 2007.|
|Citing Patent||Filing date||Publication date||Applicant||Title|
|US8267081||Feb 20, 2009||Sep 18, 2012||Baxter International Inc.||Inhaled anesthetic agent therapy and delivery system|
|US8596494 *||Jan 16, 2009||Dec 3, 2013||Stanwell Technic Limited||Gas recovery systems for beverage dispensing|
|US20100212668 *||Feb 20, 2009||Aug 26, 2010||Baxter International Inc.||Inhaled anesthetic agent therapy and delivery system|
|US20110204089 *||Jan 16, 2009||Aug 25, 2011||Robert Walter Shettle||Improvements to gas recovery system|
|U.S. Classification||62/617, 128/205.12, 62/532, 128/204.16|
|International Classification||B01D9/04, A62B19/00, F24F5/00, C02F1/22, F25J3/00, A62B23/02|
|Cooperative Classification||A61M16/009, A61M2205/3606, A61M16/0093|
|May 11, 2006||AS||Assignment|
Owner name: ANESTHETIC GAS RECLAMATION, L.L.C., TENNESSEE
Free format text: ASSIGNMENT OF ASSIGNORS INTEREST;ASSIGNORS:BERRY, JAMES M.;MORRIS, STEVE;REEL/FRAME:017891/0001
Effective date: 20060511
Owner name: ANESTHETIC GAS RECLAMATION, L.L.C.,TENNESSEE
Free format text: ASSIGNMENT OF ASSIGNORS INTEREST;ASSIGNORS:BERRY, JAMES M.;MORRIS, STEVE;REEL/FRAME:017891/0001
Effective date: 20060511
|Jul 12, 2013||FPAY||Fee payment|
Year of fee payment: 4