Publication number | US7819011 B2 |
Publication type | Grant |
Application number | US 11/450,186 |
Publication date | Oct 26, 2010 |
Filing date | Jun 9, 2006 |
Priority date | Aug 23, 2005 |
Fee status | Paid |
Also published as | US20070089519 |
Publication number | 11450186, 450186, US 7819011 B2, US 7819011B2, US-B2-7819011, US7819011 B2, US7819011B2 |
Inventors | Zhili Hao, Farrokh Ayazi |
Original Assignee | Georgia Tech Research Corporation |
Export Citation | BiBTeX, EndNote, RefMan |
Patent Citations (13), Referenced by (4), Classifications (6), Legal Events (3) | |
External Links: USPTO, USPTO Assignment, Espacenet | |
This application claims the benefit of U.S. Provisional Application No. 60/710,713, filed Aug. 23, 2005.
This invention was made in part with government support under Contract Number DAAH01-01-1-R004 awarded by the United States Army. Therefore, the government may have certain rights in this invention.
The present invention relates generally to micromachined block resonators, and more particularly, to high-Q longitudinal block resonators with annexed platforms for use in mass sensing applications.
Resonant micro- and nano-cantilevers as mass sensors have demonstrated adsorption-induced stiffness variation, complicating interpretation of experimental results for calculating adsorbed mass from frequency shift. This is discussed by S. Cherian and T. Thundat, in Applied Physics Letters, 80 (2002), pp. 2219-2221, for example. Moreover, cantilevers need to be scaled down to the nanometer range for high sensitivity, imposing fabrication difficulty and long-term stability. This is discussed by A. Gupta, J. Denton, et. al., in J. Microelectromech. Syst., 12, (2003), pp. 185-192, for example. While FBAR-based mass sensors can achieve high sensitivity at micron scale, the dependence of the mass sensitivity on the FBAR design, such as acoustic isolation for separating the vibrations from the substrate, complicates the fabrication processes. This is discussed by R. Gabi, E. Green, et. al., in IEEE Sensors 2003, pp. 1184-1188.
MEMS/NEMS (MicroElectroMechanical Systems or NanoElectroMechanical Systems) based resonators coated with selective binding layers are of great interest for detecting chemical or biological species. This is discussed by P. G. Datskos, et al., in “Micro and Nanocantilever Sensors”, Encyclopedia of Nanoscience and Technology, Edited by H. S. Nalwa, American Publishers (2004). They have found numerous potential applications. For instance, they can be used to monitor patients at home, provide tools for life science study, monitor environmental conditions, and contribute to homeland security.
By monitoring its resonant frequency variation, a mechanical resonant structure coated with a particular sensitive binding layer can detect the existence and measure the concentration of a particular target species in the analyte of interest, to be specific, the adsorbed mass of that species. Although this basic concept of species or mass sensing has been demonstrated for a long time, some key technical challenges prevent it from practical applications. For instance, most of the MEMS/NEMS based mass sensors developed so far have utilized a flexural-mode cantilever as the resonant structure. Cantilever-based mass sensors have demonstrated attogram-level sensitivity in vacuum and femtogram-level sensitivity in air. This is discussed by B. Llic, et al., in “Attomgram Detection Using Nanoelectromechanical Oscillators”, Journal of Applied Physics, Vol. 95, No. 7, April 2004, pp. 3694-3703, K. L. Ekinci, et al., in “Ultrasensitive Nanoelectromechanical Mass Detection”, Applied Physics Letters, Vol. 84, No. 22, May 2004, pp. 4469-4471, and Nickolay V. Lavrik et al., in “Femtogram Mass Detection Using photothermally Actuated Nanomechanical Resonators”, Applied Physics Letters, Vol. 82, No. 16, April 2003, pp. 2697-2699. However, their low quality factors (Q) demonstrated in air make them unsuitable for real-world detection. Especially, a higher Q in a resonant mass sensor is desirable in that it translates to a higher signal-to-noise ratio, lower motional resistance for its interface circuit, and a lower minimum detectable mass. This is discussed by Yu-Wei Lin, et al., in “Series-Resonant VHF Micromechanical Resonator Reference Oscillators,” Resistance” IEEE Journal of Solid-State Circuits, Vol. 39, No. 12, December 2004, pp. 2477-2491, and K. L. Ekinci, Y et al., in “Ultimate Limits to Inertial Mass Sensing Based Upon Nanoelectromechanical Systems”, Journal of Applied Physics, Vol. 95, No. 5, March 2004, pp. 2682-2689. Due to their complicated assembly of photothermal actuation and optical readout disclosed in the Lavrik et al. paper, integration and packaging of cantilever-based sensors present a major bottle block to the implementation of sensor-array configurations.
Besides the above-mentioned challenges, there are other technical issues pertinent to cantilever-based mass sensors. For instance, resonant frequency shift should ideally be caused only by the mass of an absorbed species. However, the adsorption process introduces stiffness variation of a cantilever, and complicates the interpretation of experimental results. This is discussed by G. Y. Chen, et al., “Adsorption-Induced Surface Stress and Its Effects on Resonance Frequency of Microcantilevers”, Vol. 77, No. 8, Journal of Applied Physics, April 1995, pp. 3618-3622, and Suman Cherian and Thomas Thundat, in “Determination of Adsorption-Induced Variation in the Spring Constant of a Microcantilever”, Applied Physics Letters, Vol. 80, No. 12, March 2002, pp. 2219-2221. Since the same amount of adsorbed mass at different locations along the length of a cantilever has different distributed modal mass, non-uniform adsorption of a target species will further introduce error in experimental measurement. This is discussed by Don L. Devoe, in “Piezoelectric Thin Film Micromechanical Beam Resonators”, Sensors and Actuators, Vol. 88, 2001, pp. 263-272.
It would be desirable to have micromachined longitudinal block resonators that have improved sensitivity and reliability.
The various features and advantages of the present invention may be more readily understood with reference to the following detailed description taken in conjunction with the accompanying drawings, wherein like reference numerals designate like structural elements, and in which:
Disclosed are high-Q length-extensional mass sensors, featuring both annexed sensing platforms and on-chip integrated transducers. The annexed sensing platforms incorporated into the sensors greatly alleviate the technical issues pertinent to cantilever-based mass sensors. The utilization of the length-extensional bulk-mode vibrations of a resonant microstructure enables the advantages as follows: 1) higher mass sensitivity at a large scale, relative to cantilever-based sensors; 2) high-Q in air; 3) on-chip integrated transducers; and 4) compatibility with sensor array configurations.
Concept
Working Principle
Referring to the drawing figures,
Initially, the resonant structure with the selective binding 16 layer vibrates at its original resonant frequency, f_{o}. Once a target species appears in the analyte under test, the binding layer 16 will adsorb the species 17 and give rise to a lower resonant frequency, f_{s}. By measuring resonant frequency shift, Δf=f_{o}−f_{s}, this mass sensor monitors mass loading, Δm, from species adsorption.
Annexed Sensing Platforms
As illustrated in
Due to the near-uniform motion of the platforms 12, especially in areas sufficiently away from the separation beams 13, the adsorbed species 17 across the platforms 12 has the same distributed modal mass, and hence non-uniform adsorption does not complicate the interpretation of experimental results. Moreover, since the annexed platforms 12 experience the maximum vibration amplitude in the whole structure, the adsorbed species introduces the largest distributed modal mass and maximizes the shift in resonant frequency of the device.
On-Chip Integrated Transducers
Referring to
As is shown in
As is shown in
As is illustrated in
Alternatively, the mass sensor 10 can be operated with on-chip integrated piezoelectric transducers 18, as shown in
Design and Modeling
Length-Extensional Bulk-Mode Vibrations
The length-extensional bulk-mode vibrations of the mass sensor 10 will now be discussed.
Close-up views of modal displacement distribution in different regions of the microstructure are illustrated in
Therefore, the equivalent modal stiffness, k, of the length-extensional vibrations is from the central block 11 and the separating beams 13. The annexed platforms 12 undergo primarily translational-motion while suffering slightly deformation in the area close to the separating beams 13. Because the annexed platforms 12 experience the maximum vibration amplitude, it is reasonable to consider that the equivalent modal mass, M, is mainly from them.
Simulated in ANSYS/Multiphysics,
A comparison of the modal displacement distribution along the length of the structures used for mass sensing is illustrated in
m(x)=m _{s}·φ(x) (1)
where m_{s }and φ(x) denote the adsorbed mass and modal displacement at the location, x.
Based on the above equation, the equivalent distributed modal mass of the same amount of adsorbed mass can vary from zero at x=0 to twice of its physical value at the edge of either a cantilever or a rectangular block. Thus, the nonuniform adsorption of a target species 17 on such types of resonant structures will complicate the interpretation of the adsorbed mass. In contrast, the adsorbed species across the platforms 12 has the same contribution to the equivalent distributed modal mass, greatly alleviating the non-uniform absorption of any species 17.
As may be observed from
Since the sensing platform 12 undergoes negligible deformation, the 1-D governing equation for the length-extensional bulk-mode vibrations in the microstructure may be expressed as:
where E and ρ denote Young's modulus and density of the resonant structure's material, respectively; A_{b}=b_{b}·h and A_{c}=b·h denote the cross-section areas of the separation beam 13 and the central block 11, respectively.
When the resonator undergoes time-harmonic vibrations, it can be assumed that:
u(x,t)=U _{0}(x)·e ^{iω·t} (3)
where ω denotes the angular resonant frequency.
By using the boundary condition at x=0, requiring that the displacement be zero, the solution to Region I may be written as:
U _{c} =U _{c0}·sin(c·x)Region I (4)
where c=ω/√{square root over (E/ρ)} and U_{c0 }is the vibration amplitude in Region I
The solution to Region II may be written as:
U _{b} =U _{b1}·cos(c·x)+U _{b2}·sin(c·x)Region II (5)
where U_{b1 }and U_{b2 }are the vibration amplitudes in Region II.
For Region II, the boundary condition at x=L+Lb requires that the normal force be equal to the acceleration of the mass of the annexed sensing platform 12:
where m_{a}=ρ·L_{a}·b·h is the physical mass of the annexed sensing platform 12.
The boundary condition at x=L requires that the normal force and the longitudinal displacement from the central block 11 be equal to the corresponding values from the separation beam 13:
The combination of Equations (6) and (7) yields the expression for the following relation:
where the matrix is associated with the geometrical parameters and the physical properties of the structural material used.
In order to obtain nontrivial solutions to U_{c0}, U_{b1}, and U_{b2}, the determinant of the matrix in Equation (8) must be set to zero. It is the eigenvalue causing the determinant to vanish that corresponds to the resonant frequency of the bulk-mode vibrations of the microstructure. Therefore, the eigenvalue equation for the resonant frequency can be expressed as:
where γ_{A}=A_{b}/A_{c }denotes the ratio of the cross-section area of the separation beam 13 to that of the central block 11.
By expanding tangent and cotangent functions into series and neglecting higher orders (c·L_{b})^{2 }and (c·L)^{2}, Equation (9) can be simplified as:
where f_{th }is the theoretical resonant frequency and γL=L_{b}/L is the ratio of the length of the separation beam 13 to half the length of the central block 11.
As illustrated in
Equivalent Lumped-Element Model
The equivalent modal stiffness of the device can be obtained using two numerical simulations. As shown in
Given in either
Through combining Equations (11) and (12), the accurate equivalent modal stiffness of this mass sensor 10 can be obtained using the following expression:
Then, the equivalent modal mass of the mass sensor 10 can be written as:
where λ_{eff }is a coefficient, which is larger than the value of 1 and can be determined from numerical simulation. The dynamic vibration behavior of the microstructure can be described by an equivalent lumped-element model:
where D is the damping coefficient, which is determined by experiment. The capacitive force, denoted by F, is from the on-chip integrated capacitive transducers 18.
Capacitive Transducers
With the assumption of a parallel-plate model, which neglects the fringing effect, the capacitive force with the resonant frequency, ω, applied to the two ends of the sensing platforms 12 can be calculated as below:
where F_{d }and F_{s}, denote the terms from the drive electrode and sense electrode, respectively; and v_{d }is an ac voltage of frequency, ω. C_{do }and C_{so }are the static capacitances of the drive and sense electrodes 23, 22, respectively.
Electro-Mechanical Coupling
Incorporating the capacitive force into Equation (15) leads to the expression modeling electromechanical coupling behavior of the mass sensor 10:
where the term at the right side of the equation is the driving force, and k_{e }is the electrostatic stiffness, expressed as below:
Therefore, taking electrostatic stiffness into account, the resonant frequency of the capacitive mass sensor 10 is calculated as:
To aid in the design and analysis of the device, an admittance model is derived, by combining the capacitive equations with the second-order equation of the resonant microstructure. The overall transfer function describing the admittance, Y_{sd}, between the drive electrode 23 (input) and sensed electrode 22 (output) of the mass resonator depicted in
where v_{d }is the drive voltage, and i_{s }the current measured at the sense electrode 22.
Equation (20) can also be expressed as the product of the mechanical force-displacement transfer function for the microstructure, x(jω)/F(jω), the electromechanical coupling at the input, η_{d}=F(jω)/v_{d}(jω), and the electromechanical coupling at the output port, η_{s}(jω)=i_{s}(jω)/[jω·x(jω)]. Therefore, Equation (20) can be rewritten as:
While Equation (17) can be written in the following format:
Based on the above equations, an equivalent electrical model describing the dynamic behavior of the two-port configuration of the capacitive mass sensors 10 is illustrated in
where TX_{1}, and TX_{2 }denote the coupling of the drive side and the sense side, respectively, which are both equal to 1. R_{net }is the resistor of 50Ω in a network analyzer, which is connected to the mass sensor 10 for performance characterization.
According to Equation (23b), a high quality factor is critical for reducing the motional resistance of the mass sensor 10. Correspondingly, a high resonant sense current going through the mass sensor 10 can be expected:
The vibration amplitude at the end of the annexed sensing platform 12 is calculated as:
Figures of Merit
The mass sensitivity (S_{m}) of this device can be expressed as:
According to Equation (26), by varying the length (L_{s}) of the annexed sensing platform 12, this design can cover a large range of mass sensitivity within one single sensor-array chip.
The theoretical minimum detectable mass (Δm_{min}) of this mass sensor 10 may be expressed as:
where k_{b }and T denote the Boltzman constant and the environment temperature, respectively; B is Bandwidth; and q is the vibration amplitude. Both a high-Q and a large vibration amplitude help reduce the Δm_{min}.
With the known design and operation parameters, k, k_{e}, and f_{tune}, and the measured parameter, f_{s}, the actual loaded mass, Δm, can be calculated using the following relation:
Experimental Verification
Experimental Procedure
The length-extensional capacitive mass sensors 10 with on-chip integrated capacitive transducers 18 have been fabricated on a 4.3 μm-thick SOI (silicon-on-insulator) wafer using a one-mask fabrication process. Exemplary one-mask fabrication processes are disclosed by Reza Abdolvand and Farrokh Ayazi, in “A Gap Reduction and Manufacturing Technique for Thick Oxide Mask Layers with Multiple-Size Sub-Micron Openings,” to be published in Journal of Microelectromechanical Systems, and in “Single-Mask Reduced-Gap Capacitive Micromachined Devices,” Proc. IEEE MicroElectro Mechanical Systems Conference (MEMS2005), Miami, Fla., 2005, pp. 151-154, for example.
To evaluate the performance of this mass sensor 10, certain amount of nanoparticles (Ceria from nGimat Co., diameter<20 nm) was placed on its annexed platforms 12, using a very fine probe tip (radius=2.5 μm) under a microscope. This has the same effect as mass loading due to species adsorption.
To measure the mass sensitivity of this device, the resonant frequencies of the mass sensor 10 were measured before and after mass loading, as illustrated in
R _{measured}=50·10^{Loss/20} (29)
Experimental Results
The performance of exemplary mass sensors 10 shown in
TABLE 1 | ||||
Performance summary of the high-Q bulk-mode | ||||
mass sensor with annexed sensing platforms | ||||
Sensing platform length (μm) | ||||
35 | 40 | 45 | ||
Resonant frequency (MHz) | 13.392 | 12.245 | 11.534 | |
Bias voltage (V) | 140 | 110 | 110 | |
Drive and sense gap (nm) | 680 | 560 | 570 | |
Quality factor (Q) | 4400 | 3800 | 4000 | |
Motional resistance (MΩ) | 2.43 | 2.4 | 2.43 | |
Sensing current (μA) | 0.45 | 0.51 | 0.48 | |
Vibration amplitude (nm) | 12 | 12 | 13 | |
Stiffness (N/m) | 2.20 × 10^{5} | 2.05 × 10^{5} | 2.00 × 10^{5} | |
Effective mass (pg) | 3.10 × 10^{4} | 3.46 × 10^{4} | 3.80 × 10^{4} | |
Mass sensitivity (Hz/pg) | 215 | 177 | 151 | |
Minimum detectable mass (attogram) | 1.17 | 1.52 | 1.57 | |
Both theoretical (23 b) and measured (29) motional resistances of the mass sensors are 2.4 MΩ or so. The vibration amplitude at ˜12 nm leads to leads to a sense current of 0.5 μA going through the mass sensors. A stiffness in the table is the equivalent modal stiffness, and varies with fabrication tolerance of the width of the separation beams. By varying the length of the sensing platform from 35 μm to 45 μm, the mass sensitivity can be tuned from 215 Hz/pg to 151 Hz/pg. Both the mass sensitivity and the quality factor of this device are much larger than the corresponding values (66 Hz/pg and Q<100 at freq<100 kHz) of a sub-micron thick cantilever, such as is disclosed by Amit Gupta, et al., in “Novel Fabrication Method for Surface Micromachined Thin Single-Crystal Silicon Cantilever Beams”, Journal of Microelectromechanical Systems, Vol. 12, No. 2, April 2003, pp. 185-192. Based on Equation (27), the theoretical Δm_{min }of this device in air is at the attogram level, which is comparable to that of a nanocantilever in vacuum.
Thus, high-Q length-extensional mass sensors 10 have been disclosed, that exhibit numerous advantages over micro/nano-cantilever-based mass sensors. With no need of a transducer assembly and with a large Q, the mass sensors 10 show better performance in air and compatibility with sensor array configuration. To lower the bias voltage, a HARPSS-on-SOI process can be used to reduce the capacitive gaps (<200 nm). Alternatively, piezoelectric transduction 18 can be used on the central block 11 of the sensor 10, providing more flexibility for using the device in various detection environments. The disclosed mass sensors 10 can be further integrated into oscillator circuits for sensor array configurations and further parallel detection.
The above-disclosed high quality factor (Q) mass sensors 10 having annexed platforms 12 offer the advantages of improved mass sensitivity at micron scale, high-Q in air, simplicity of fabrication, and improved reliability. The mass sensors 10 are capable of detecting sub-picogram mass change in air and in liquid. The sensitivity of such mass sensors 10 depends on lateral dimensions, and is independent of thickness. Un-deformed annexed platforms 12, coated with the selective binding layer 16 adsorbs agents, and may be used to avoid adsorption-induced stiffness changes, which lead to improved reliability of the mass sensors 10. The adsorption-induced stiffness change is negligible and the mass change is solely the reason for the frequency shift of the mass sensors 10.
The operating resonant mode shape of the mass sensors 10 shows that the annexed platforms 13 do not experience deformation but only translational movement. This feature is enabled by the use of slim separating beams 13, which prevent longitudinal waves from propagating to the platforms 12.
Compared with either cantilever-based or FBAR-based mass sensors, the high-Q mass sensors 10 having annexed platforms 12 offers the advantages of improved mass sensitivity at micron scale, high-Q in air, simplicity of fabrication, and improved reliability. The sensitivity of 60 Hz/picogram has been demonstrated at ˜12 MHz. The mass sensors 10 use the longitudinal bulk-mode of a very simple structure with relatively large dimensions (208 μm×48 μm×10 μm) to achieve picogram-level sensitivity.
Thus, improved micromachined mass sensors have been disclosed. It is to be understood that the above-described embodiments are merely illustrative of some of the many specific embodiments that represent applications of the principles discussed above. Clearly, numerous and other arrangements can be readily devised by those skilled in the art without departing from the scope of the invention.
Cited Patent | Filing date | Publication date | Applicant | Title |
---|---|---|---|---|
US3164004 * | May 15, 1961 | Jan 5, 1965 | Exxon Research Engineering Co | Coated piezoelectric analyzers |
US4215570 * | Apr 20, 1979 | Aug 5, 1980 | The United States Of America As Represented By The United States Department Of Energy | Miniature quartz resonator force transducer |
US4517841 * | Jan 6, 1983 | May 21, 1985 | Sundstrand Data Control, Inc. | Accelerometer with beam resonator force transducer |
US4548079 * | Dec 5, 1983 | Oct 22, 1985 | Wabco Westinghouse Fahrzeugbremsen Gmbh | Method and apparatus to automatically determine the weight or mass of a moving vehicle |
US5121180 * | Jun 21, 1991 | Jun 9, 1992 | Texas Instruments Incorporated | Accelerometer with central mass in support |
US5349844 * | Sep 11, 1992 | Sep 27, 1994 | Trc Companies, Inc. | System and method for resonant filter mass monitoring |
US5668303 * | Apr 21, 1993 | Sep 16, 1997 | Forschung E.V Fraunhofer-Gesellschaft Zur Foerderung Der Angewandten | Sensor having a membrane as part of an electromechanical resonance circuit forming receiver and transmitter converter with interdigital structures spaced apart from one another |
US6457361 * | Aug 31, 1999 | Oct 1, 2002 | Ngk Insulators, Ltd. | Mass sensor and mass sensing method |
US7089813 * | Feb 18, 2005 | Aug 15, 2006 | Ngk Insulators, Ltd. | Mass sensor and mass sensing method |
US7279131 * | Jul 1, 2004 | Oct 9, 2007 | Uop Llc | Method and apparatus for mass analysis of samples |
US7458265 * | Oct 27, 2004 | Dec 2, 2008 | Drexel University | Piezoelectric cantilever sensors |
US20050016276 * | Jun 4, 2004 | Jan 27, 2005 | Palo Alto Sensor Technology Innovation | Frequency encoding of resonant mass sensors |
US20070089519 * | Jun 9, 2006 | Apr 26, 2007 | Georgia Tech Research Corporation | High-Q longitudinal block resonators with annexed platforms for mass sensing applications |
Citing Patent | Filing date | Publication date | Applicant | Title |
---|---|---|---|---|
US8719959 * | Aug 27, 2009 | May 6, 2014 | Sii Nano Technology Inc. | Cantilever, cantilever system, and probe microscope and adsorption mass sensor including the cantilever system |
US8854149 * | Sep 14, 2012 | Oct 7, 2014 | National Tsing Hua University | MEMS resonator, manufacturing method thereof, and signal processing method using MEMS resonator |
US20100058499 * | Aug 27, 2009 | Mar 4, 2010 | Masatsugu Shigeno | Cantilever, cantilever system, and probe microscope and adsorption mass sensor including the cantilever system |
US20140002201 * | Sep 14, 2012 | Jan 2, 2014 | National Tsing Hua University | Mems resonator, manufactoring method thereof, and signal processing method using mems resonator |
U.S. Classification | 73/649, 73/579, 333/186 |
International Classification | G01H11/00 |
Cooperative Classification | G01H11/08 |
European Classification | G01H11/08 |
Date | Code | Event | Description |
---|---|---|---|
Jun 9, 2006 | AS | Assignment | Owner name: GEORGIA TECH RESEARCH CORPORATION, GEORGIA Free format text: ASSIGNMENT OF ASSIGNORS INTEREST;ASSIGNORS:HAO, ZHILI;AYAZI, FARROKH;REEL/FRAME:017990/0805 Effective date: 20060609 |
May 15, 2012 | CC | Certificate of correction | |
Apr 28, 2014 | FPAY | Fee payment | Year of fee payment: 4 |