US 8117921 B2 Abstract A flowmeter is disclosed. The flowmeter includes a vibratable flowtube, and a driver connected to the flowtube that is operable to impart motion to the flowtube. A sensor is connected to the flowtube and is operable to sense the motion of the flowtube and generate a sensor signal. A controller is connected to receive the sensor signal. The controller is operable to determine an individual flow rate of each phase within a multi-phase flow through the flowtube.
Claims(20) 1. A system comprising:
a liquid fraction probe configured to measure a liquid fraction of a flow of fluid that includes a first liquid, a second liquid, and a gas;
a vibratable flowtube configured to receive the flow of fluid;
a driver connected to the flowtube and operable to impart motion to the flowtube;
a sensor connected to the flowtube and operable to sense the motion of the flowtube and generate a sensor signal; and
one or more processing devices connected to access the sensor signal and to access the measured liquid fraction, the one or more processing devices being configured to determine a corrected liquid fraction based on the sensor signal and the measured liquid fraction.
2. The system of
3. The system of
4. The system of
5. The system of
wherein the one or more processing devices are configured to determine the corrected liquid fraction based on the gas void fraction.
6. The system of
wherein the one or more processing devices are configured to determine a corrected water cut.
7. The system of
8. The system of
9. The system of
10. The system of
wherein the one or more processing devices are configured to determine the corrected liquid fraction based on the measured temperature or the measured pressure.
11. A method comprising:
measuring a liquid fraction of a flow of fluid that includes a first liquid, a second liquid, and a gas;
passing the flow of fluid through a vibratable flowtube;
imparting motion to the flowtube;
generating a signal based on motion of the flowtube; and
determining a corrected liquid fraction of the flow of fluid based on the measured liquid fraction and the signal.
12. The method of
13. The method of
14. The method of
wherein the method further comprises determining a corrected mass flow rate and a corrected density based on the corrected liquid fraction, the apparent density, and the apparent mass flow rate.
15. The method of
wherein determining a corrected liquid fraction of the flow of fluid comprises determining a corrected liquid fraction of the flow of fluid based on the gas void fraction.
16. The method of
17. The method of
18. The method of
19. The method of
20. The method of
wherein determining a corrected liquid fraction of the flow of fluid comprises determining the corrected liquid fraction based on the measured temperature or the measured pressure.
Description Under 35 U.S.C. §120, this application claims priority to, and is a continuation of U.S. application Ser. No. 11/681,992, filed Mar. 5, 2007, and titled MULTIPHASE CORIOLIS FLOWMETER, which is a continuation of U.S. application Ser. No. 11/069,931, filed Mar. 2, 2005, now U.S. Pat. No. 7,188,534, issued Mar. 13, 2007, which claims priority to U.S. application Ser. No. 60/549,161, filed on Mar. 3, 2004, and titled MULTIPHASE CORIOLIS FLOWMETER. U.S. application Ser. No. 11/069,931 is also a continuation-in-part of U.S. application Ser. No. 10/773,459, filed Feb. 9, 2004, and titled MULTI-PHASE CORIOLIS FLOWMETER, now U.S. Pat. No. 7,059,199, issued Jun. 13, 2006, which itself claims priority under 35 U.S.C. §119(e) to both of (i) U.S. application Ser. No. 60/445,795, filed on Feb. 10, 2003, and titled MULTIPHASE CORIOLIS FLOWMETER, abandoned, and (ii) U.S. application Ser. No. 60/452,934, filed on Mar. 10, 2003, and titled MULTIPHASE CORIOLIS FLOWMETER. All of the above-listed applications are hereby incorporated by reference. This description relates to flowmeters. Flowmeters provide information about materials being transferred through a conduit, or flowtube. For example, mass flowmeters provide an indication of the mass of material being transferred through a conduit. Similarly, density flowmeters, or densitometers, provide an indication of the density of material flowing through a conduit. Mass flowmeters also may provide an indication of the density of the material. For example, Coriolis-type mass flowmeters are based on the Coriolis effect, in which material flowing through a conduit is affected by a Coriolis force and therefore experiences an acceleration. Many Coriolis-type mass flowmeters induce a Coriolis force by sinusoidally oscillating a conduit about a pivot axis orthogonal to the length of the conduit. In such mass flowmeters, the Coriolis reaction force experienced by the traveling fluid mass is transferred to the conduit itself and is manifested as a deflection or offset of the conduit in the direction of the Coriolis force vector in the plane of rotation. According to one general aspect, a system includes a controller that is operable to receive a sensor signal from a first sensor connected to a vibratable flowtube containing a three-phase fluid flow that includes a first liquid, a second liquid, and a gas, the controller being further operable to analyze the sensor signal to determine an apparent flow parameter of the fluid flow, a second sensor that is operable to determine an apparent flow condition of the fluid flow, and a corrections module that is operable to input the apparent flow parameter and the apparent flow condition and determine a corrected flow parameter therefrom. Implementations may include one or more of the following features. For example, the corrections module may be further operable to input the apparent flow parameter and the apparent flow condition and determine a corrected flow condition therefrom. The apparent flow parameter may include an apparent bulk density of the fluid flow, or an apparent bulk mass flow rate of the fluid flow. The second sensor may include a liquid fraction probe that is operable to determine a liquid fraction measurement identifying a volume fraction of the first liquid with respect to the second liquid, or a void fraction determination system that is operable to determine a void fraction of the gas within the fluid flow. A component flow rate determination system may be included that is operable to determine a flow rate of the first liquid within the fluid flow. The component flow rate determination system may be implemented at the controller, the corrections module, the second sensor, or a host computer in communications with the controller, the corrections module, or the second sensor. A component flow rate determination system may be included that is operable to determine a flow rate of the gas within the fluid flow. Implementation of the corrections module may be associated with a processor of the controller, or with a processor of the second sensor. A host computer may be in communication with the controller or the second sensor and operable to implement the corrections module. In the system, the second sensor may be operable to output a first apparent flow condition value to the controller for use in determination of a first corrected flow parameter value, and the controller may be operable to output the first corrected flow parameter value to the second sensor for determination of a first corrected flow condition value, and the second sensor may be operable to output a second corrected flow condition value to the controller for use in determination of the corrected flow parameter value. The correction module may include a neural network that is operable to input the apparent flow parameter and the apparent flow condition and output the corrected flow parameter and a corrected flow condition. The neural network may include a first correction model that is particular to a type of the second sensor and flow condition and that is operable to output a corrected flow condition, and a second correction model that is particular to a type of the apparent flow parameter and that is operable to output the corrected flow parameter, wherein the first correction model may be operable to correct the apparent flow condition based on the apparent flow condition and the corrected flow parameter, and the second correction model may be operable to correct the apparent flow parameter based on the apparent flow parameter and the corrected flow condition. The controller may be operable to correct the apparent flow parameter based on a theoretical relationship between the apparent flow parameter and the corrected flow parameter. The controller may be operable to correct the apparent flow parameter based on an empirical relationship between the apparent flow parameter and the corrected flow parameter. The system may include a conduit connecting the second sensor and the vibratable flowtube, such that the fluid flow flows through the second sensor, the pipe, and the vibratable flowtube. The first liquid, the second liquid, and the gas may be co-mingled with one another within the fluid flow during determination of the flow condition by the second sensor. According to another general aspect, an apparent bulk density of a multi-phase flow through a flowtube is determined, the multi-phase flow including a first liquid, a second liquid, and a gas. An apparent bulk mass flow rate of the multi-phase flow is determined, and a first mass flow rate of the first liquid is determined, based on the apparent bulk density and the apparent bulk mass flow rate. Implementations may include one or more of the following features. For example, an apparent flow condition of the multi-phase flow other than the apparent bulk density and the apparent bulk mass flow rate may be determined, wherein determining the first mass flow rate of the first liquid comprises determining the first mass flow rate based on the apparent flow condition. In determining the first mass flow rate of the first liquid, a corrected flow condition may be determined, based on the apparent flow condition. In determining the corrected flow condition, a corrected bulk density and a corrected bulk mass flow rate may be determined. Determining the apparent flow condition may include determining an apparent liquid fraction measurement of a volume fraction of the first liquid within the multi-phase flow, and/or determining an apparent gas void fraction of the gas within the multi-phase flow. Determining the first mass flow rate of the first liquid may include determining a corrected bulk density, based on the apparent bulk density, and determining a corrected bulk mass flow rate, based on the apparent mass flow rate. Determining the corrected bulk density and determining the bulk mass flow rate may include determining a corrected flow condition, based on the apparent flow condition. According to another general aspect, a flowmeter includes a vibratable flowtube containing a three-phase flow including a first liquid, a second liquid, and a gas, a driver connected to the flowtube and operable to impart motion to the flowtube, a sensor connected to the flowtube and operable to sense the motion of the flowtube and generate a sensor signal, and a controller connected to receive the sensor signal and determine a first flow rate of a first phase within a three-phase flow through the flowtube, based on the sensor signal. According to another general aspect, a method of improving an output of a flowmeter includes determining an apparent bulk density of a multi-phase flow through a flowtube, the multi-phase flow including a first liquid, a second liquid, and a gas, determining an apparent bulk mass flow rate of the multi-phase flow, determining an apparent flow condition of the multi-phase flow, and correcting the apparent bulk density or the apparent mass flow rate, based on the apparent bulk density, the apparent mass flow rate, and the apparent flow condition. According to another general aspect, a method of improving an output of a liquid fraction probe includes determining an apparent bulk density of a multi-phase flow through a flowtube, the multi-phase flow including a first liquid, a second liquid, and a gas, determining an apparent bulk mass flow rate of the multi-phase flow, determining an apparent liquid fraction the first liquid within the multi-phase flow, and correcting the apparent liquid fraction to obtain a corrected liquid fraction, based on the apparent bulk density, the apparent mass flow rate, and the apparent liquid fraction. Implementations may include one or more of the following features. For example, a gas void fraction of the gas within the multi-phase flow may be determined based on the apparent bulk density, the apparent mass flow rate, and the corrected liquid fraction. According to another general aspect, a method of obtaining a gas void fraction measurement includes determining an apparent bulk density of a multi-phase flow through a flowtube, the multi-phase flow including a first liquid, a second liquid, and a gas, determining an apparent bulk mass flow rate of the multi-phase flow, determining an apparent gas void fraction of the gas within the multi-phase flow, and correcting the apparent gas void fraction to obtain a corrected gas void fraction, based on the apparent bulk density, the apparent mass flow rate, and the apparent gas void fraction. Implementations may include one or more of the following features. For example, a liquid fraction of the first liquid within the multi-phase flow may be determined based on the apparent bulk density, the apparent mass flow rate, and the corrected gas void fraction According to another general aspect, a system includes a conduit having a fluid flow therethrough, the fluid flow including at least a first liquid component, a second liquid component, and a gas component, a vibratable flowtube in series with the conduit and having the fluid flow therethrough, a first sensor operable to determine a first apparent property of the fluid flow through the conduit, a second sensor connected to the flowtube and operable to sense information about a motion of the flowtube, a driver connected to the flowtube and operable to impart energy to the flowtube, a control and measurement system operable to measure a second apparent property and a third apparent property of the fluid flow, and a corrections system operable to determine a corrected first property, a corrected second property, and a corrected third property, based on the first apparent property, the second apparent property, and the third apparent property. According to another general aspect, a system includes a controller that is operable to determine a first apparent property of a fluid flow in which a first liquid, a second liquid, and a gas are co-mingled, a meter that is operable to measure a second apparent property of the fluid flow, and a corrections module that is operable to input the first apparent property and output a first corrected property, wherein the meter is operable to input the first corrected property and the second apparent property and output a second corrected property. The details of one or more implementations are set forth in the accompanying drawings and the description below. Other features will be apparent from the description and drawings, and from the claims. Types of flowmeters include digital flowmeters. For example, U.S. Pat. No. 6,311,136, which is hereby incorporated by reference, discloses the use of a digital flowmeter and related technology including signal processing and measurement techniques. Such digital flowmeters may be very precise in their measurements, with little or negligible noise, and may be capable of enabling a wide range of positive and negative gains at the driver circuitry for driving the conduit. Such digital flowmeters are thus advantageous in a variety of settings. For example, commonly-assigned U.S. Pat. No. 6,505,519, which is incorporated by reference, discloses the use of a wide gain range, and/or the use of negative gain, to prevent stalling and to more accurately exercise control of the flowtube, even during difficult conditions such as two-phase flow (e.g., a flow containing a mixture of liquid and gas). Although digital flowmeters are specifically discussed below with respect to, for example, Referring to The digital transmitter Density measurements of the material flowing through the flowtube are related to, for example, the frequency of the motion of the flowtube Similarly, mass flow through the flowtube The temperature in the flowtube It should be understood that both the pressure of the fluid entering the flowtube A liquid fraction probe A void fraction sensor Various techniques, represented generally in In this description, an amount of gas in a flowing fluid, measured by the void fraction sensor or otherwise determined, is referred to as void fraction or α, and is defined as α=volume of gas/total volume=volume of gas/(volume of liquid+volume of gas). Accordingly, a quantity referred to herein as the liquid fraction is defined as 1−α. In many applications where mass flow measurements are required, the void fraction of the flow can be as high as 20, 30, 40% or more. However, even at very small void fractions of 0.5%, the fundamental theory behind the Coriolis flowmeter becomes less applicable. Moreover, a presence of gas in the fluid flow also may affect both an actual and a measured value of a density of the fluid flow, generally causing the density measurement to be, and to read, lower than if the fluid flow contained only the liquid component. That is, it should be understood that a density ρ Beyond this physical phenomenon, a Coriolis meter measuring a two-phase fluid flow containing gas may output a density reading ρ U.S. Pat. No. 6,505,519, which is incorporated by reference above, discloses that such a variation of ρ Somewhat similarly, an indicated bulk mass flow rate MF Examples of detailed techniques for correcting ρ Further in Once the factors of ρ Once the component flow rates MF Superficial velocities are referred to herein as those velocities that would exist if the same mass flow rate of a given phase was traveling as a single phase through the flowtube Slip velocities refer to a condition in which gas and liquid phases in a two-phase flow have different average velocities. That is, an average velocity of a gas AV A flow regime is a term that refers to a characterization of the manner in which the two phases flow through the flowtube The existence of a particular flow regime is known to be influenced by a variety of factors, including, for example, a gas void fraction in the fluid flow, an orientation of the flowtube Information about phase slip may be determined at least in part from flow regime knowledge. For example, in the bubble flow regime, assuming the bubbles are uniformly distributed, there may be little relative motion between the phases. Where the bubbles congregate and combine to form a less uniform distribution of the gas phase, some slippage may occur between the phases, with the gas tending to cut through the liquid phase. In In In Once the existence of two-phase flow is established, a corrected bulk density ρ Once ρ In Once corrected density ρ Once determined, the liquid/gas component flow rates may be output or displayed ( In some instances, this determination may be sufficient ( In particular, superficial velocities of the gas and liquid, SV As shown in Eqs. 1 and 2, determination of superficial velocities in this context relies on the earlier determination of MF Using the superficial velocities and other known or calculated factors, some of which may be stored in the flow regime maps database Flow regime(s) in two phase liquid/gas flow may be described by contours on a graph plotting the liquid superficial velocity versus the gas superficial velocity. As just described, an improvement to determinations of ρ In Eqs. 3 and 4 assume that there is no slip velocity (i.e., phase slip) between the liquid and gas phases (i.e., average velocity of the gas phase, AV In the second instance and thereafter ( If phase slip does exist ( As a result, the liquid and gas flow rates are determined ( As described above, gas entrained in liquid forms a two-phase flow. Measurements of such a two-phase flow with a Coriolis flowmeter result in indicated parameters ρ In Other graphing techniques may be used; for example, true void fraction may be plotted against indicated void fraction. For example, Next, a gas-free density of the liquid, ρ An indicated mass flow rate MF An algorithmic density correction ( This can be used to define a quantity “density drop,” or Δρ, as shown in Eq. 11:
Note that Eq. 11 shows the quantity Δρ as being positive; however, this quantity could be shown as a negative drop simply by multiplying the right-hand side of the equation by −1, resulting in Eq. 12:
The quantity ρ As discussed extensively above, density measurements by a Coriolis meter, or any vibrating densitometer, generally are under-reported by the meter, and require correction. Accordingly, under two-phase flow Eqs. 12 or 13 may thus be used to define the following two quantities: a corrected or true density drop, Δρ There can be derived or empirically determined a relationship between Δρ As a result, the relationship may generally be derived, or at least proven, for each flowtube in each setting. For one model flowtube, known and referred to herein as the Foxboro/Invensys CFS10 model flowtube, it has been empirically determined that for some conditions the above functional relationship can be simplified to be only a function Δρ
To force the condition for both sides of Eq. 16 to be zero when there is no apparent density drop relationship results in Eq. 17: Once the true density drop is determined, then working back through the above equations it is straightforward to derive the true mixture density ρ A tabular correction for density ( The table It should be understood that either or both of the algorithmic and tabular forms may be extended to include multiple dimensions, such as, for example, gain, temperature, balance, or flow regime. The algorithmic or tabular correction also may be extended to include other surface fitting techniques, such as, for example, neural net, radical basis functions, wavelet analyses, or principle component analysis. As a result, it should be understood that such extensions may be implemented in the context of A density of the gas, ρ Further, the gas may be known to have specific characteristics with respect to factors including pressure, temperature, or compressibility. These characteristics may be entered along with an identification of the gas, and used in determining the current gas density ρ Once the factors ρ Although the above discussion presents techniques for determining void fraction α The mass flow rate correction system In applying the tabular correction for mass flow rate ( The table Normalized values MF In an algorithmic approach, as with density, the correction for mass flow may be implemented by way of a theoretical or an empirical functional relationship that is generally understood to be of the form ΔMF=f(MF For some cases the function can simplify to a polynomial, such as, for example, the polynomial shown in Eq. 19:
For some set of conditions, the functional relationship can be a combination of a polynomial and exponential, as shown in Eq. 20:
In one implementation, m in Eq. 20 may be replaced by apparent superficial liquid velocity SV It should be understood that, as with the density corrections discussed above, either or both of the algorithmic and tabular forms may be extended to include multiple dimensions, such as, for example, gain, temperature, balance, or flow regime. The algorithmic or tabular correction also may be extended to include other surface fitting techniques, such as, for example, neural net, radical basis functions, wavelet analyses, or principle component analysis. As a result, it should be understood that such extensions may be implemented in the context of All of the above functional relationships for mass flow rate may be restated using gas fraction (α) or liquid fraction (100−α) instead of density drop, as reflected in the table Having described density, void fraction, and mass flow rate corrections above in general terms, for the purpose of, for example, simultaneously calculating individual flow component (phases) flow rates in a two-phase flow, the below discussion and corresponding figures provide specific examples of implementations of these techniques. More specifically, the ½″ data was taken with a 0.15 kg/s flow rate and a 0.30 kg/s flow rate; the ¾″ data was taken with a 0.50 kg/s flow rate and a 1.00 kg/s flow rate; and the 1″ data was taken with a 0.50 kg/s flow rate, a 0.90 kg/s flow rate, and a 1.20 kg/s flow rate. In
In
The model formula, based on Eq. 17, provides Eq. 25:
To summarize, the drop in density correction is performed in the transmitter
The model formula for this situation is shown as Eq. 27:
In this case, the coefficient are: a To summarize, mass flow correction in the transmitter
As will be appreciated, the above description has a wide range of applications to improve the measurement and correction accuracy of a Coriolis meter during two phase flow conditions. In particular, the techniques described above are particularly useful in measurement applications where the mass flow of the liquid phase and the mass flow of the gas phase must be measured and/or corrected to a high level of accuracy. One exemplary application is the measurement of the mass flow of the liquid phase and the measurement of the gas phase in oil and gas production environments. The above discussion is provided in the context of the digital flowmeter of Many of the above-given equations and calculations are described in terms of density, mass flow rate, and/or void fraction. However, it should be understood that the same or similar results may be reached using variations of these parameters. For example, instead of mass flow, a volumetric flow may be used. Additionally, instead of void fraction, liquid fraction may be used. The above discussion provides examples of measuring component mass flow rates in a two-phase flow. Flowmeters also may be used to measure further mixed flows. For example, a “three-phase” flow or “mixed two-phase flow” refers to a situation in which two types of liquid are mixed with a gas. For example, a flowing mixture of oil and water may contain air (or another gas), thus forming a “three-phase flow,” where the terminology refers to the three components of the flow, and does not generally imply that a solid material is included in the flow. The flowmeter system In For example, an apparent mass flow rate of a three-phase fluid flow within the flowtube Once obtained, corrected values such as mass flow rate, density, water cut, or void fraction (or some combination thereof) may be output to a host computer More generally, an example of the system For example, in some implementations, the digital transmitter Further, although the external sensors In In some implementations, the corrections system For example, the mass flow rate and density corrections may be incorporated into one module, while the water cut correction module Along the same lines, it should be understood that the component flow rate determination system Various examples of the above and other implementations, as well as examples of specific techniques for obtaining corrected flow measurements and individual component flow rates, are described in more detail below. In general, however, it should be understood that the system As a result, determinations of individual component flow rates do not require separation of the three-phase fluid flow into separate flows containing one or more of the constituent components. For example, when the three-phase flow contains oil, water, and gas, it is not necessary to separate the gas from the oil/water liquid combination in order to perform measurements (e.g., mass flow rate) on the oil portion of the resulting oil/liquid flow. Accordingly, reliable measurements of an amount of oil produced, for example, at an oil production facility, may be made easily, quickly, inexpensively, and reliably. Also in Somewhat similarly, a pressure transmitter Further in Meanwhile, the transmitter Also, as referenced above, The corrections system For example, as discussed in more detail below, the models In the example of Therefore, as illustrated, by feeding the values of a first determination of a corrected water cut value from the water cut error model In In other implementations, and, for example, where a specific water cut probe, Coriolis meter, and configuration thereof with respect to one another are known and assumed to be unchanging, then it may be possible to construct a single error model that inputs all three measurements of water cut, mass flow rate, and density, and outputs corrected values of all three (along with, possibly, a corrected gas void fraction). In such implementations, it may not be necessary to feed sequential results back into the error model in order to obtain all three (or four, or more) corrected values. In In this way, a corrected water cut ( For example, the corrected mass flow rate may be determined based only on apparent measurements, such as apparent mass flow rate, or may be determined based on these factors along with an already-corrected density and/or gas void fraction measurement. Similar comments apply, for example, to techniques for obtaining corrected density and/or gas void fraction measurements. Also, it should be apparent that other factors and parameters may be used in calculating corrected values that are not necessarily shown in Further, as referenced above, a given correction may be obtained multiple times, with later corrections being based on intervening corrections of other parameters. For example, a first-corrected water cut measurement may be obtained, and may then be revised based on a following void fraction determination, to obtain a second-corrected water cut measurement. Once some or all of the corrected parameters are obtained, individual component flow rates for one or more of the first liquid component, second liquid component, and gas component may be obtained ( Thus, in Based on this knowledge, an apparent gas void fraction α Once these values are known, a correction for gas void fraction α Then, the improved water cut measurement may be used to determine and improve the liquid density measurement ( The process Specific equations and discussion for implementing the example processes For example, water cut in a two-phase flow is defined as the volume fraction of water in the two-phase (e.g., oil-water) mixture, when devoid of gas. Under this condition, water cut is given by Eq. 29: Generally, in the case of just a two-phase oil-water flow, where no gas is present, the Coriolis flowmeter may measure the mixture (bulk) density, ρ The volumetric flow rate of the liquid (oil-water) mixture may be derived using Eq. 30: Eqs. 29 and 30, however, cannot be directly applied when three distinct phases (i.e., oil, water and gas) are in a co-mingled stream, i.e., a three-phase flow, as discussed above with respect to In the implementations described above, the independent information is provided by another device installed in-line with the Coriolis flowmeter, which encounters the same three-phase mixture, i.e., the water cut probe The transmitter The gas void fraction, α, as referenced above, is defined as the volume fraction occupied by the gas phase in the three-phase mixture. A definition of α in terms of apparent or non-corrected values, is provided above and repeated here as Eq. 32:
The density of the gas phase in Eq. 32 above may be calculated based on an independent measurement of process pressure and temperature. For example, pressure may be measured with the pressure transmitter In Eq. 32, and as already described with respect to Specifically, following application of mass flow and bulk density corrections, an updated gas void fraction is obtained ( For each specific water cut device, the relationship between water cut and the effect of gas void fraction may be known as shown in Eq. 33:
With the water cut reading updated, the process is repeated, starting with Eq. 31, until suitable convergence criteria has been satisfied. Then, the corrected three-phase mixture (bulk) mass flow rate, density, and gas void fraction may be reported at process temperature. The individual volumetric flow rate of each phase/component is then calculated and corrected to standard temperature using, for example, the American Petroleum Institute (API) equations for crude oil and produced water, and the AGA algorithms for produced gas. These functionalities also may be incorporated into the transmitter For example, in one implementation, the water cut meter Then, the water cut measurement may be used to determine a density of the total liquid component (e.g., a density of a combined oil and water portion of the three-phase flow), perhaps using Eq. 31 ( At this point, first values for corrected and bulk density and corrected bulk mass flow rate may be determined ( With the revised water cut measurement and other parameters, a revised gas void fraction measurement may be obtained. Then, as shown, further corrections to the bulk mass flow rates and bulk density may be performed, and this process may be repeated until a suitable level of correction is reached. And, as described above with respect to Based on this information, corrected values for bulk density and bulk mass flow rate may be determined ( As with The water cut probe Other techniques for use with the processes of In A mass flow rate of a first liquid component (e.g., water) is then determined ( Specifically, an apparent gas void fraction is determined ( Then, an apparent gas flow rate is determined (
Then, an apparent superficial gas velocity is determined (
An apparent liquid flow rate may then be determined (
Apparent superficial liquid velocity can then be determined (
Then, an error rate for liquid mass flow measurement is determined (
This fractional liquid mass flow error as a function of both apparent superficial liquid and apparent superficial gas flows (normalized) may be estimated using a polynomial expression shown in Eq. 39, where the error term is shown as e
In Eq. 39, due to the size of the expression(s), the following notation is used: v A corrected liquid mass flow rate measurement may be determined (
Then, the water cut and component densities may be determined ( For example, the water and oil mass flowrates may be calculated, using Eqs. 41 and 42:
Then, using the corrected mixture density (or corrected gas void fraction), the gas mass flowrate may be determined using Eqs. 43 and 44: In some cases, there may be uncertainty in the polynomial fit to the error curves, where the effect of this uncertainty on the corrected mass flowrate is given by Eq. 45:
Eq. 45 helps to explain why some data may exhibit large errors when the correction algorithms are use outside a tested region. For example, if the calculated error at a given flowrate is −70%, but the true error is −75% then the model error is only 5%, but the error in the corrected mass flow is
Such a calculation also may be used in 2-phase flow modeling results as described above, to consider resulting residual error in the modeling. In one implementation, the model least square fit may be modified to minimize the resultant mass flow error rather than the model error. Also, generally speaking, a flowtube may be expected to exhibit small mass flow errors, so that if a flowmeter is expected to correct for large errors, then the error modeling (and hence experimental data) becomes relatively more important. Thus, as described above with respect to It should be noted that the actual determination of the gas mass flow may be affected by uncertainty in the mixture density and a relative difference in density between the liquid and gas phase. Also, it should be understood that the density correction polynomial discussed above may be more or less applicable depending on, for example, flowtube orientation. As a result, for example, horizontal flow may result in a lower error than vertical flow, or vice-versa. In the above-described approach, the use of superficial liquid and gas velocities may enable the correction algorithms to include knowledge of the multi-phase flow regimes encountered, which may lead to better correction algorithms. The results on practical data indicate that the correction polynomial curves may benefit from being designed on data spanning the expected flow-ranges, and from being ‘jacketted’ to prevent spurious results when exposed to data outside the known range. Although the implementations discussed above make use of an external water cut probe or similar technique, other implementations could be used that rely on the external void fraction sensor/meter Additionally, as referenced above, other devices, such as those designed to determine an “oil-cut,” rather than a water cut, may be used. Further, if oil and water in a mixture have well-separated densities, a sampling system may be used that takes a representative sample of the mixture, de-gasses it and uses a Coriolis meter to determine the water cut. As described, in the case of single liquid two-phase flow, knowledge of the liquid and gas densities at the operation temperature and pressure may be used with the corrected density and massflow measurements to calculate each of the liquid and gas mass flow rates, and, thereby, the liquid and gas volumetric flow. In the case of three-phase flow, extra, external measurements may be used to enable the estimation of gas mass flow and the mass flow of each of the two liquids. In the case of water and oil liquid mixture, the water cut of the mixture may be measured up-stream of the Coriolis meter, as explained and illustrated above. In one implementation, it may be assumed that the two liquids do not interact in such a way as to invalidate the assumption that the mixture of the two liquid behave as a single liquid as far as the interaction with the gas phase is concerned. This assumption makes the three-phase flow an extension of the single-liquid two-phase flow, the extra measurements being used to determine the mixed liquid density and to decouple the separate liquid massflows, after two-phase flow calculations are applied. As further discussed above, a Coriolis meter will generally under-read both the mixture density and the mixture massflow of a liquid/gas mixture. To compensate for the errors in these raw measurements and estimate the true measurement values, a model for the error surfaces may be used so as to find a mapping between the raw density and massflow measurements, and the value of the raw measurement errors, for both massflow and density measurements, i.e., to perform a data fitting. As already pointed out, both the density and massflow error curve may depend on many factors, such as, for example, meter size, meter orientation (e.g., horizontal vs. vertical), and a nominal liquid mass flow. Accordingly, corrections may be developed for each individual meter size and orientation. In other implementations, the compensations may be scaled according to meter size and/or adjusted according to meter alignment. General data fitting techniques include, for example, table lookup, polynomial/rational function interpolation, non-linear methods, and neural networks, among others. For example, For the design of a flexible compensation technique for two- or three-phase flow errors in a Coriolis meter, neural network models present at least the following advantageous features. For example, such models provide the ability to derive a non-linear functional mapping from a sufficiently large and representative database of relevant measurements, without prior knowledge of the underlying physical model of the process. Such a feature may be particularly advantageous in the example of the two/three-phase flow compensation problem, where actual physical processes inside the tube may be difficult to obtain. Further, development time for a viable solution for a particular problem may be significantly reduced compared with other data fitting techniques, which may rely on domain expertise. For example, in the particular case of two-phase flow compensation, changing meter size/orientation/type might completely change the shape of the raw measurements surface, and for a conventional data technique, this may imply a process of finding another form for the functional mapping that is not guaranteed to find a solution in a reasonable time. By contrast, using the same neural network architecture, the neural network training may find the “best” (in the sense of the cost function chosen to control the network training) solution for the data available by adjusting its internal parameters during the training process. The following discussion provides explanation of one example of a neural network, i.e., an MLP model. Specifically, To model raw measurements error surfaces for density and mass flow, as discussed above, a functional mapping may be given by MeasError=F(dd, {dot over (m)}), with dd the apparent drop in mixture density and {dot over (m)} the apparent massflow of the liquid. It should be noted that this notation is slightly different from the above notation for the same parameters, i.e., Δρ and MF, respectively. An output y
In general terms, an MLP is a feed-forward neural network architecture with several layers of units. Being feed-forward means that the data flows monotonically from inputs to outputs, without inner loops; this ensures that the outputs function is deterministic. In order to ensure the universality property, the MLP used for two-phase flow measurement error compensation may be a two-layer architecture with sigmoidal activation functions for hidden units In this case, the sigmoidal activation function may be given by Then, an output of the MLP used as a function of the inputs can be written as in Eq. 47:
That is, Eq. 47 represents a non-linear function in apparent drop in mixture density and massflow, with nh the number of hidden units The network parameters w There are various methods for choosing the number of hidden units The general outline of one implementation of a training process is described below. The data available for the training is divided into three independent sets: the training set (used to iteratively change the values of the MLP weights to minimize the cost function); the validation set (used to stop the training early to avoid over-fitting the training data); and the test set (used to choose the number of hidden units). In one implementation, the network training starts with an initial set of network weights w The cost function also may be evaluated over the validation set, and the training stopped when this starts increasing, so that a suitable compromise between the fit of the training data and the generalization capabilities may be achieved. That is, over-fitting from training to convergence over the training set may be avoided. If enough data is available, a test set also may be used to assess the performance of several MLP trained, as described, but with different numbers of hidden units to choose the architecture that gives the minimum cost function over the test set. In the case of massflow compensation, for a low GVF region, the compensation accuracy may be increased if this area is considered separate from the rest and modeled accordingly. Such approach(es) suggest the use of a “committee of models,” also referred to as a “mixture of experts,” with separate but overlapping areas of expertise to enable soft switching between the models. An example of such a committee, one used to compensate the raw massflow errors for a 1″ flowtube in vertical alignment, is: Model 1: 0-1.5.kg/s, 0-15% GVF Model 2: 0-1.5 kg/s, 10% GVF upwards Model 3: 1.2 kg/s upwards, 0-15% GVF Model 4: 1.2 kg/s upwards, 10% GVF upwards A different model, referred to as a “blanket model,” also may be trained using the whole range of flows and GVFs. The blanket network may be used to provide a rough idea about the true liquid massflow. Using this estimation together with the estimated true GVF (based on the density compensation model), a specific expert model (or a combination of two expert models if the data falls in the overlap region) may be selected, and its compensation applied. Based on this data, compensation solutions for density and liquid massflow errors as described above may be derived and validated online, using independent test data, as shown in The test data in this example included thirteen flowlines, with nominal flows ranging from 0.6 kg/s to 3 kg/s, in steps of 0.25 kg/s, with GVF steps of 2%, giving a total of 266 experimental points, an average of 20 points per flowline. In the following description and figures, reference is made to the following test conditions: Test00wc—4000 bpd Test00wc—6000 bpd Test06wc—3000 bpd Test06wc—4000 bpd Test06wc—6000 bpd Test06wc—8000 bpd Test13wc—3000 bpd Test13wc—6000 bpd Test25wc—3000 bpd Test25wc—7000 bpd Test35wc—3000 bpd Test35wc—7000 bpd Test50wc—3000 bpd Test50wc—5000 bpd Test50wc—7000 bpd Test50wc—8000 bpd In this context, In The Coriolis principle and relate techniques, as described above, provide estimates for an overall mass flow and density of the three-phase, mixed fluid. Knowledge of the true fluid densities and (perhaps estimated or corrected) water cut, together with models of two-phase flow errors, gives estimates of the fluid-only mass flow rate, and the gas void fraction (GVF). Thus, in Accordingly, The gas volumetric flow may be sensitive to errors in the density calculation at low GVFs. For example, with 2% GVF, a 1% absolute error in the estimate of GVF may lead to 50% error in the estimated volumetric gas flow. Such large relative errors may generally be associated with relatively low gas flows, and, therefore, may be unlikely to be significant in oil and gas applications, such as the examples described herein. More specifically, Finally, Thus, gas flow errors may be seen to be sensitive to water cut errors at low GVF, where this influence may decrease with higher GVFs. In this context, the water-cut meter has an error even at 0% GVF, due to the presence of residual amounts of gas ‘carry-under’ from the process, as follows (in absolute water-cut units) with respect to the specified test results referred to above: Test00wc—4000 bpd: −0.52 Test00wc—6000 bpd: −1.91 Test06wc—3000 bpd: −0.89 Test06wc—4000 bpd: −0.74 Test06wc—6000 bpd: −1.53 Test06wc—8000 bpd: −2.78 Test13wc—3000 bpd: 1.17 Test13wc—6000 bpd: 0.87 Test25wc—3000 bpd: 0.91 Test25wc—7000 bpd: −0.56 Test35wc—3000 bpd: 0.74 Test35wc—7000 bpd: −0.35 Test50wc—3000 bpd: 3.89 Test50wc—5000 bpd: 2.64 Test50wc—7000 bpd: 2.90 Test50wc—8000 bpd: 2.31 For the purpose of correcting the gas-induced errors, the water-cut meter was considered without error at 0% GVF (as in In With the data as described, the water-cut meter reading may be corrected from errors as large as −40% to mainly within 2 percent absolute error, as shown in Raw water-cut errors are described and shown above with respect to Thus, in order to consider such variations in mass flow and density errors that are caused by variations in water cut measurement(s), The data is based on the oil data discussed above, but with nominal water-cut values of 0, 5.5, 13.1, 24.8, 35.6 and 49%. The distribution of flowlines per nominal water-cut is as follows: 0%: 4000 and 6000 bpd 5.5%: 3000, 4000, 6000 and 8000 bpd 13.1%: 3000 and 6000 bpd 24.8%: 3000 and 7000 bpd 35.6%: 3000 and 7000 bpd 49%: 3000 5000, 7000 and 8000 bpd Improved models for mixture density and massflow errors were trained using the true water-cut value as an extra input. The accuracy of the resulting corrections on the training data is given in The described errors may be reduced by having more data points. For example, for most of the water cuts there were only two flow lines. The number of data points in the set may be insufficient to be able to identify outliers. With more and better data quality, perhaps smaller mass flow and density errors may be possible, even allowing for a range of water cut values. Thus, These illustrated calculations represent a “complete” set, suitable for oil continuous applications. The data is based on the oil data as described above, with nominal water-cut values of 0, 5.5, 13.1, 24.8, 35.6 and 49%. The water-cut, massflow and density corrections used are based on the whole data set for the range of water-cut from 0 to 50%. The water cut correction model used is the same as is discussed above with respect to The water, oil and gas correction accuracies achieved after successive calculations are shown in A set of analysis tools and correction algorithms have been illustrated that, given appropriate data for the oil, water, and gas used in a specific application, may compensate for gas-induced errors in Coriolis and water cut readings, thereby to deliver volumetric gas, water and oil flow rates. As described above, a massflow meter may be capable of maintaining operation in the presence of a high percentage of gas in a measured flow, both with a single or a mixed liquid (i.e., in two-phase or three-phase flow). Estimates and/or apparent measurements of the liquid-gas mixture density and massflow may thus be obtained. However, these estimates have errors that depend on various factors, including, for example, the gas void fraction and/or the true liquid massflow, which may be so large as to render the raw measurements useless. By using an additional measurement parameter, such as, for example, a water cut or gas void fraction measurement, along with the apparent mass flow rate and density measurements, corrected values for all of these parameters, and others, may be obtained. Moreover, by cycling through these measurements and calculations with ever-improved corrections, successively improved values may be obtained, as, for example, the corrections converge to specific values. As described above, techniques for performing these corrections may be based on data-fitting techniques that seek to determine, for example, existing error rates in a particular setting or configuration, so that these errors may be accounted for in future measurements and corrections. As such, these techniques may be dependent on an extent of a correlation between the settings/configurations in which the data was obtained, and the settings/configurations in which they are ultimately applied. Related or other correction techniques may be used that seek to characterize fluid flow(s) in a more general sense, i.e., using fluid flow equations that seek to describe a behavior of the flow as a physical matter. For example, the well-known Navier-Stokes equations may be used in this sense. Specifically, the three-dimensional unsteady form of the Navier-Stokes equations describe how the velocity, pressure, temperature, and density of a moving fluid are related. The equations are a set of coupled differential equations and may, in theory, be solved for a given flow problem by using methods from calculus, or may be solved analytically, perhaps using certain simplifications or adjustments that may be determined to be helpful and applicable in a given circumstance. These or related equations may consider, for example, convection (a physical process that occurs in a flow of gas in which some property is transported by the ordered motion of the flow), and/or diffusion (a physical process that occurs in a flow of gas in which some property is transported by the random motion of the molecules of the gas, and which may be related to the viscosity of the gas). Turbulence, and the generation of boundary layers, are the result of diffusion in the flow. By using such fluid flow equations and characteristics, correction techniques may be obtained for many or all of the parameters and techniques discussed above. For example, such fluid flow equations may be used in defining a general correction model, which may be supplemented by data-fitting techniques such as those described above, or vice-versa. A number of implementations have been described. Nevertheless, it will be understood that various modifications may be made. Accordingly, other implementations are within the scope of the following claims. Patent Citations
Non-Patent Citations
Classifications
Legal Events
Rotate |