|Publication number||US8161585 B2|
|Application number||US 12/872,288|
|Publication date||Apr 24, 2012|
|Filing date||Aug 31, 2010|
|Priority date||Sep 4, 2007|
|Also published as||CN101815470A, CN101815471A, CN101854865A, EP2185075A2, EP2185075A4, EP2192859A1, EP2192859A4, EP2203118A1, EP2203118A4, US7784127, US8122542, US20090056021, US20090056022, US20090070935, US20100319128, WO2009032927A1, WO2009032933A1, WO2009032935A2, WO2009032935A3|
|Publication number||12872288, 872288, US 8161585 B2, US 8161585B2, US-B2-8161585, US8161585 B2, US8161585B2|
|Inventors||Serge Nukhim Kuro, Graham T. Reitz, Bradley J. Brunker, Brent Harper|
|Original Assignee||Tomotherapy Incorporated|
|Export Citation||BiBTeX, EndNote, RefMan|
|Patent Citations (54), Non-Patent Citations (3), Referenced by (2), Classifications (18), Legal Events (3)|
|External Links: USPTO, USPTO Assignment, Espacenet|
This application is a divisional of and claims priority to U.S. patent application Ser. No. 12/204,628 filed on Sep. 4, 2008, which claims priority to U.S. Provisional Patent Application No. 60/969,904, filed on Sep. 4, 2007. The entire contents of each application are incorporated herein by reference.
This invention relates to a radiation therapy imaging and treatment system. More specifically, the invention relates to a patient support device for use with such a system.
Medical equipment for radiation therapy treats tumorous tissue with high energy radiation. The dose and the placement of the dose must be accurately controlled to ensure both that the tumor receives sufficient radiation to be destroyed, and that damage to the surrounding and adjacent non-tumorous tissue is minimized. Intensity modulated radiation therapy (“IMRT”) treats a patient with multiple rays of radiation each of which may be independently controlled in intensity and/or energy. The rays are directed from different angles about the patient and combine to provide a desired dose pattern. In external source radiation therapy, a radiation source external to the patient treats internal tumors. The external source is normally collimated to direct a beam only to the tumorous site. Typically, the radiation source includes either high-energy X-rays, electrons from certain linear accelerators, or gamma rays from highly focused radioisotopes, though other types of radiation sources are possible.
One way to control the position of the radiation delivery to the patient is through the use of a patient support device, such as a couch, that is adjustable in one or more directions. The use of a patient support device is well known in the medical field, with similar patient support devices being used in CT scanning devices and Magnetic Resonances Imagers (MRIs). The patient support device allows the patient to be moved into and out of the field of the radiation to be delivered and in some cases, allow for adjustments of patient position during a radiation treatment.
When a patient support device, such as a couch, is used in this manner, there are many variables that need to be accounted for. For example construction materials and configuration of suitable electronics necessary to operate the couch must be carefully selected to ensure smooth operation of the couch, and precise measurement of couch position (when the couch has multiple movable parts). When these features are thoughtfully considered in the environment of radiation delivery, the patient support device can be a key tool in improving patient outcomes.
The present invention provides a patient support device comprising a base, a table assembly supported by the base and configured to support a patient, a motor electrically coupled to and operable to control motion of the table assembly, a controller electrically coupled to the motor, the controller operable to generate a signal to brake the motor when power to the motor is interrupted, and a brake control module. The brake control module is electrically coupled to the motor and the controller and is operable upon reactivation of the motor. The brake control module includes a passive dynamic load module electrically coupled to the motor to increase speed of the motor, a rectification module electrically coupled to the motor and operable to convert AC voltage to DC voltage when the AC voltage reaches a predetermined value, a controlled dynamic load module electrically coupled to the passive dynamic load module, and a switch electrically coupled to the controlled dynamic load module and operable to connect and disconnect the controlled dynamic load module to the motor to control a braking operation of the motor.
In another aspect, the present invention provides a radiation therapy treatment system comprising a patient support device and a control system. The patient support device includes a table assembly configured to support a patient, and a motor electrically connected to the table assembly and operable to control movement of the table assembly. The control system is electrically connected to the motor and operable to control a speed of the motor and provide linear motion of the table assembly when power to the couch is interrupted.
Other aspects of the invention will become apparent by consideration of the detailed description and accompanying drawings.
Before any embodiments of the invention are explained in detail, it is to be understood that the invention is not limited in its application to the details of construction and the arrangement of components set forth in the following description or illustrated in the following drawings. The invention is capable of other embodiments and of being practiced or of being carried out in various ways. Also, it is to be understood that the phraseology and terminology used herein are for the purpose of description and should not be regarded as limiting. The use of “including,” “comprising,” or “having” and variations thereof herein is meant to encompass the items listed thereafter and equivalents thereof as well as additional items. Unless specified or limited otherwise, the terms “mounted,” “connected,” “supported,” and “coupled” and variations thereof are used broadly and encompass both direct and indirect mountings, connections, supports, and couplings.
Although directional references, such as upper, lower, downward, upward, rearward, bottom, front, rear, etc., may be made herein in describing the drawings, these references are made relative to the drawings (as normally viewed) for convenience. These directions are not intended to be taken literally or limit the present invention in any form. In addition, terms such as “first,” “second,” and “third” are used herein for purposes of description and are not intended to indicate or imply relative importance or significance.
In addition, it should be understood that embodiments of the invention include hardware, software, and electronic components or modules that, for purposes of discussion, may be illustrated and described as if the majority of the components were implemented solely in hardware. However, one of ordinary skill in the art, and based on a reading of this detailed description, would recognize that, in at least one embodiment, the electronic based aspects of the invention may be implemented in software. As such, it should be noted that a plurality of hardware and software based devices, as well as a plurality of different structural components may be utilized to implement the invention. Furthermore, and as described in subsequent paragraphs, the specific mechanical configurations illustrated in the drawings are intended to exemplify embodiments of the invention and that other alternative mechanical configurations are possible.
The radiation module 22 can also include a modulation device 34 operable to modify or modulate the radiation beam 30. The modulation device 34 provides the modulation of the radiation beam 30 and directs the radiation beam 30 toward the patient 14. Specifically, the radiation beam 30 is directed toward a portion 38 of the patient. Broadly speaking, the portion 38 may include the entire body, but is generally smaller than the entire body and can be defined by a two-dimensional area and/or a three-dimensional volume. A portion or area desired to receive the radiation, which may be referred to as a target or target region, is an example of a region of interest. Another type of region of interest is a region at risk. If a portion includes a region at risk, the radiation beam is preferably diverted from the region at risk. Such modulation is sometimes referred to as intensity modulated radiation therapy (“IMRT”).
The modulation device 34 can include a collimation device 42 as illustrated in
In one embodiment, and illustrated in
The radiation therapy treatment system 10 can also include a detector 78, e.g., a kilovoltage or a megavoltage detector, operable to receive the radiation beam 30, as illustrated in
The system 10 can also include a patient support device, shown as a couch 82, operable to support at least a portion of the patient 14 during treatment. While the illustrated couch 82 is designed to support the entire body of the patient 14, in other embodiments of the invention the patient support need not support the entire body, but rather can be designed to support only a portion of the patient 14 during treatment. The couch 82 moves into and out of the field of radiation along an axis 84 (i.e., Y axis). The couch 82 is also capable of moving along the X and Z axes as illustrated in
With reference to
The upper support 94 includes an upper surface 102 and a lower surface 106 that contacts an upper surface 110 of the lower support 98. As shown in the illustrated embodiment, the lower surface 106 includes a bearing layer 114 that is intended to reduce friction between the lower surface 106 and the upper surface 110 of the lower support 98 when the upper support 94 is moved with respect to the lower support 98. Specific details of the bearing layer 114 and its application are discussed in co-pending U.S. patent application Ser. No. 12/204,617, the entire contents of which are incorporated herein by reference.
The table assembly 92 is movable in the X, Y, and Z directions, as illustrated in
Once the user actuates the buttons 144 of the keypad 140, the table assembly 92 will move at the direction of the user. In conventional couch designs, a hydraulic lifting system is utilized to move the table assembly 92 in the Z direction. The hydraulic lifting system is a convenient way to achieve some control over the lowering of the table assembly 92, and has the benefit of allowing the table assembly 92 to be lowered when there is no power delivered to the system 10. When the power to the couch 82 is disrupted while a patient is in the treatment position, the table assembly 92 needs to be lowered to allow the patient to exit the couch, and such lowering must be done in a controlled manner. However, hydraulic systems are more expensive to implement, are less reliable, and are less accurate in their range of motion.
With reference to
The lowering mechanism 160 includes power braking resistors to dissipate energy from a motor 170 to control the downward motion of the table assembly 92. The braking resistors act as a damping or deceleration device, taking the energy output of the motor 170 and allowing for controlled lowering of the table assembly 92, even in the powered off situation. This allows for regulated control of a free running motor that provides linear motion of the mechanical system under nonlinear external loads, even if the power to the system is interrupted. The braking resistors are designed so that no matter the load on the table assembly 92, the downward speed of the table assembly 92 remains the same. By keeping the speed constant, even with a dynamic load, control of the motion is achieved.
Using the lowering mechanism 160, the motor 170 becomes a generator during the lowering process of the table assembly 92 (i.e., if the power is uncontrolled, the power and speed increase as the couch drops). To prevent this, there needs to be a change in the load resistance applied proportionately to the power generated. When the generator has too high of a load, it begins braking. The effective value of the resistance is changed by connecting and disconnecting a power resistor. If the power resistor were constantly applied to the lowering mechanism 160, the speed of the table assembly 92 would increase as the table assembly 92 is lowered (simulating a free fall) that could cause the table assembly 92 to crash at the bottom of the path of motion. By alternating the connection of the power resistor to the lowering mechanism 160, the table assembly 92 is protected from crashing. By applying a non-linear load to the lowering mechanism 160, the speed drop of the table assembly 92 is linear such that the resistance linearizes what was previously non-linear motion. The frequency with which the power resistor is connected to the lowering mechanism 160 changes the effective resistance within the braking circuitry.
The lowering mechanism 160 also includes support arms 164 that couple the table assembly 92 to a riser 168 of the base 93. As shown in the illustrated embodiment, the lowering mechanism 160 includes two pairs of support arms 164, with each arm 164 within a pair of arms being parallel to the other. As the table assembly 92 is raised and lowered, a longitudinal axis of each arm 164 within a pair remains parallel to the other arm, and a plane P1 formed by the longitudinal axis of one pair of arms does not intersect a plane P2 formed by the longitudinal axis of the other pair of arms.
Movement of the table assembly 92 in the Z axis, as described in some detail above, utilizes an electromechanical roller screw. The Z axis motion is controlled by a dual feedback mechanism. Incremental feedback is provided by the roller screw, and a direct drive encoder looks at angle and provides absolute feedback. All axes of the couch 82 have step-move capabilities due to their control mechanisms. In the Z direction, doing a step-move will correct for cobra motion in the Y axis direction.
The lowering mechanism 160 includes a motor control system 169 as illustrated in
The motor control system 169 includes a system enable interlock 186 where pin 1 is connected to SYS enable bus 190, pin 2 is connected to system VDC bus 182, pin 3 is connected through the enable bus 194 to pin 5 of the motor controller 198, and pin 4 is connected through bus 202 to a coil 206 d of a motor power switch 206 and to a coil 210 c of a motor brake release (MBR) interlock switch 210.
The motor controller 198 includes pin 1, pin 2, and pin 3 connected to the 3-Ph AC bus 178, pin 4 connected to system VDC bus 182, pin 6 connected through bus 214 to normally-open contact 210 a of the MBR interlock switch 210, and pin 7, pin 8, and pin 9 connected through 3-Phase Motor Controller Bus 218 to appropriate normally-open contacts 206 a, 206 b, 206 c of the motor power switch 206.
The motor 170 is connected through a shaft 222 to the mechanical non-linear variable external load PEXT 174. A three-phase power bus 226 connects the motor 170 to the appropriate common contacts 206 a, 206 b, 206 c of the motor power switch 206. The motor 170 includes a motor brake release (MBR) 230 connected through bus 234 to common contact 210 a of the MBR interlock switch 210.
The motor control system 169 also includes an emergency dynamic braking control unit 238 connected through 3-Phase Power bus 242 to appropriate normally-closed contacts 206 a, 206 b, and 206 c of the motor power switch 206. A delta connected 3-Phase resistive dynamic load 246 is connected to appropriate phases of the 3-Phase power bus 242. A star connected 3-Phase capacitive dynamic load 250 is connected to appropriate phases of the 3-Phase power bus 242. A 3-Phase rectifier 254, with common points of diodes D1-D4, D2-D5, and D3-D6, is connected to appropriate phases of the 3-Phase Power bus 242. The common points of the diodes D4, D5, D6 are connected to ground and the common points of the diodes D1, D2, D3 are connected to the power-control-sensor bus 258.
The power-control-sensor bus 258 is connected to a voltage sensor dynamic load 262 (RVSL), and the other side of the RVSL is connected to ground. The power-control-sensor bus 258 is also connected to a current sensor dynamic load 266 (CCSL), and the other side of the CCSL is connected to ground. The power-control-sensor bus 258 is also connected to a controlled dynamic load 270 (RCDL). The other side of the RCDL is connected to a normally-open contact of a power switch 274, while the common contact of the power switch 274 is connected to ground. The power-control-sensor bus 258 also is connected to a positive input of an operation amplifier 278.
The emergency dynamic braking control unit 238 includes an emergency power supply 282, which contains two rechargeable batteries BT1, BT2 connected in series. The negative lead of the battery BT1 is connected to ground and the positive lead of the battery BT2 is connected through bus 286 to common contact 210 b of the MBR interlock switch 210. The normally closed contact 210 b is connected through bus 290 to the normally open contact of an emergency motor brake release switch 294. The common contact of this switch 294 is connected through bus 298 to coil 302 c of a system VDC interlock switch 302 and to normally closed contact 210 a of the MBR interlock switch 210. The common point of the batteries BT1 and BT2 are connected through bus 306 to normally open contact 302 b of the system VDC interlock switch 302. The common contact 302 a is connected to system VDC through bus 182 and normally closed contact 302 a is connected to coil 210 c of the MBR interlock switch 210 through bus 362. Common contact 302 b is connected to bus 310, which supplies analog computer load 314 and is connected to pin 1 of a voltage reference 318, to pin 1 of the operation amplifier 278, and to pin 1 of a switched oscillator 322. The voltage reference 318 is connected by pin 2 to ground and is connected by pin 3 through bus 326 to the negative input of the operation amplifier 278. The operation amplifier 278 is connected by pin 2 to ground and is connected by pin 3 through bus 330 to pin 3 of the switched oscillator 322. The switched oscillator 322 is connected by pin 2 to ground and is connected by pin 4 through bus 334 to pin 1 of the power switch 274.
The motor 170 is initiated when it receives a signal from the 3-Phase AC power on the bus 178, system VDC on the bus 182, and SYS enable signal on the bus 190. The system enable interlock 186 initiates the motor controller 198, the motor power switch 206, the MBR interlock switch 210, and the system VDC interlock switch 302 through normally closed contact 302 a. The motor controller 198 communicates with the motor 170 through the bus 218, normally open contacts 206 a, 206 b, 206 c of the motor power switch 206, and bus 226. The motor 170 begins acting through shaft 222 on the mechanical non-linear variable external load PEXT 174.
The motor controller 198 also communicates with the motor brake release 230 through bus 214, normally open contact 210 a, and bus 234 to disengage the motor brake release 230. When main power 3-Phase AC on bus 178 and system VDC on bus 182 are interrupted, the switches 206, 210 are disengaged and the motor brake release 230 is engaged.
The motor 170 is connected to the emergency dynamic braking control unit 238 through bus 226, normally closed contacts 206 a, 206 b, 206 c, and through 3-Phase power bus 242. To activate the emergency mode when the mechanical non-linear variable external load PEXT 174 begins acting through shaft 222 on the motor 170, the user needs to push and hold the emergency motor brake release switch 294. The MBR release switch 294 is activated through bus 298 to engaged coil 302 c of the system VDC interlock switch 302, and to switch 302 by contact 302 a to disengaged coil 210 c of the MBR interlock switch 210. The motor is reactivated when the MBR 230 is disengaged. The MBR 230 is disengaged when the MBR release switch 294 is released through bus 298, normally closed contact 210 a, and bus 234. The motor 170 begins acting under external load PEXT 174, and the 3-Phase AC voltage from the motor 170 begins to interact with Passive Dynamic Loads 338. The motor 170 gets first two stages of dynamic braking action on the AC Dynamic Loads 342 of the Passive Dynamic Loads 338. The AC Dynamic Loads 342 includes two types of AC loads: (1) the 3-Phase Resistive Dynamic Load 246 which transfers AC energy from the motor into heat, and (2) the 3-Phase Capacitive Dynamic Load 250 which shifts AC phases from the motor 170. Both heat dissipation and phase shift increases current from the motor 170 and this current increases eddy currents in the motor 170 which affect braking action in the motor 170.
The motor 170 continuously increases speed under PEXT 174 until the AC voltage from the motor reaches a certain value, and then the AC voltage begins rectification by 3-Phase Rectifier 254. The rectified AC voltage begins a third stage in the DC passive dynamic braking on the Voltage Sensor Dynamic Load 262 (VVSL) of the Passive Dynamic Loads 338. The motor 170 proceeds to increase speed under PEXT 174 and the DC voltage increases too, until a certain value is reached, and then the Analog Computer Load 314 begins to control the dynamic braking action. The Analog Computer Load 314 is supplied by two sources: (1) the Emergency Power Supply 282 through BT1, through bus 306, normally open contact 302 b, and bus 310, and (2) motor 170 through bus 226, normally closed contacts 206 a, 206 b, 206 c, bus 242, 3-Phase Rectifier 254, and Power-Control-Sensor Bus 258.
There are two conditions of operation of the motor control system 169:
Only the Passive Dynamic Loads 338 are working. And time charge tCH of the capacitor CCSL 266 will have an infinite value.
The Passive Dynamic Loads 338 and the Controlled Dynamic Loads 346 begin working together, and the time charge tCH of the capacitor CCSL 266 operates up to a value of the voltage xVRF 318. This value of the time charge tCH will be inversely proportional to a value of the load PEXT 174.
The time charge tCH of the capacitor CCSL 266 up to value of the voltage xVRF is illustrated in
where RSE is the resistance of the sensor equivalent, which is calculated by formula (4) illustrated in
VVSL is the Thevenin voltage source equivalent of the Motor 170 which is calculated by formula (6) illustrated in
n is determined by formula (8) (provided below) illustrated in
where n is a mechanical system coefficient, n=0 . . . 1 (n is a positive number)
PINT is a system mechanical internal resistance 350
PEXT is a mechanical non-linear variable external load 174
xVRF is the value of the Voltage Reference 318
x is an adjusting coefficient which allows adjustment of the Emergency Dynamic Braking Control Unit 238 for different motors and different values of the braking speed control.
The operation amplifier 278 controls changes of the voltage on the Capacitor Sensor Dynamic Load (CCSL) 262 and compares it with the value of the voltage on the Voltage Reference (xVRF.) 318. When the voltage VCSL on capacitor CCSL 262 becomes equal to or higher than the value on the xVRF 318, the operation Amplifier 278 starts Switched Oscillator 322. The Switched Oscillator 322 is switched with a predetermined time tDDCH of the Power Switch 274. The Power Switch 274 connects the Controlled Dynamic Load 270 to ground, which begins the fourth stage of the dynamic braking action of the motor 170 on the Controlled Dynamic Load 270. The Controlled Dynamic Load 270 discharges Capacitor CCSL of the Current Sensor Dynamic Load 266. The predetermined time discharge tDDCH should be greater than the time constant of the discharge capacitor CCSL
During the predetermined time tDDCH that Operation Amplifier 278 controls the voltage on the Capacitor Sensor Dynamic Load 262, the operation amplifier 278 continuously compares the CCSL voltage with value of the voltage on the Voltage Reference 318 (xVRF) until the voltage xVRF becomes equal to or higher than the value CCSL 262. When the predetermined time tDDCH elapses, the Switched Oscillator 322 disconnects Power Switch 274, which disconnects RCDL 270 from ground, which begins the fifth stage of the dynamic braking action of the motor 170 on the Current Sensor Dynamic Load 266 (CCSL).
The instantaneous current IINST through the capacitor of the Current Sensor Dynamic Load 266 (CCSL) is determined by:
When the voltage on the Current Sensor Dynamic Load 266 (CCSL) again becomes equal to or greater than the value on the Voltage Reference 318 (xVRF), the Operation Amplifier 278 restarts the Switched Oscillator 322. The Switched Oscillator 322 is switched with predetermined time tDDCH of the Power Switch 274. The Power Switch 274 is connected to Controlled Dynamic Load 270 to ground, which again begins the fourth stage of the dynamic braking action of the motor 170 on the Controlled Dynamic Load 270. The Controlled Dynamic Load 270 discharges the capacitor of the Current Sensor Dynamic Load 266 (CCSL). That cycling will continue on the shaft 222 of the motor 170 and the mechanical non-linear variable external load PEXT 174 and during that cycle the Emergency Dynamic Braking Control Unit 238 will manage linear motion of the whole mechanical system.
Some of the unique features of the present invention include
(1) A motor-turned-generator 170 is a source for generating braking forces and a source of information about its conditions.
(2) The load power link, the sensors link, and the control link are on the same bus 258.
(3) The voltage sensor dynamic load 262 (VCSL) and the current sensor dynamic load 266 (CCSL) are multifunctional:
(4) The dynamic braking of the motor occurs in multiple stages:
(6) The capacitor of the current sensor dynamic load 266 is not a reactive load with respect to the motor 170 because the capacitor is charged from the motor 170 and is discharged through a resistor(s) of the controlled dynamic load 270, which means that the capacitor does not return charged energy back to the motor 170. The capacitor continues to be non-linear load.
(7) The controlled dynamic loads 346 include a two-cycle controlled dynamic load which provides a two-cycle controlled dynamic braking action on the motor 170. The first cycle is controlled by the controlled dynamic load 270, which provides the dynamic braking action on the motor 170 and discharges the capacitor of the current sensor dynamic load 266 (CCSL) during predetermined time tDDCH. The second cycle is controlled by the current sensor dynamic load 266, which provides dynamic braking action on the motor 170 during the charge time tCH of the capacitor of the current sensor dynamic load 266.
(8) When motor 170 operates as a motor-turned-generator it has the following properties:
(9) The analog computer load 314 includes combined properties of the analog computer and the controlled dynamic loads.
Additional features of this invention can be found in the following claims.
|Cited Patent||Filing date||Publication date||Applicant||Title|
|US3348114||May 11, 1964||Oct 17, 1967||Westinghouse Electric Corp||Motor drive apparatus with dynamic braking|
|US3497786||Aug 30, 1967||Feb 24, 1970||Picker Corp||Motor speed control circuit|
|US3675099 *||Jul 2, 1971||Jul 4, 1972||Gen Motors Corp||Induction motor regenerative braking system|
|US3755712 *||Sep 7, 1972||Aug 28, 1973||Square D Co||Fault protection for dc motor thyristor power supplies|
|US3851233 *||Jun 22, 1972||Nov 26, 1974||Sherman L||Electro-kinetic, parallel-series, dynamic brake for alternating current motors|
|US3974384 *||Oct 22, 1974||Aug 10, 1976||Siemens Aktiengesellschaft||X-ray diagnostic apparatus for radioscopy and X-ray film exposures including an X-ray tube having a rotating anode|
|US4354112 *||Nov 28, 1980||Oct 12, 1982||Tokyo Shibaura Denki Kabushiki Kaisha||X-ray cine radiography apparatus|
|US4987583 *||Apr 25, 1989||Jan 22, 1991||General Electric Company||Automatic backout control for a motorized positioning X-ray apparatus|
|US5044354||Jun 29, 1990||Sep 3, 1991||Siemens Aktiengesellschaft||Apparatus for treating a life form with focussed shockwaves|
|US5105141 *||Jul 12, 1990||Apr 14, 1992||General Electric Cgr Sa||Device to control the speed of two-phase or three-phase motors|
|US5113420||Dec 24, 1990||May 12, 1992||Texaco Inc.||Method and apparatus for positioning a sample with repeatable accuracy|
|US5210893||Sep 4, 1991||May 18, 1993||Kabushiki Kaisha Toshiba||Couch apparatus for medicine|
|US5317616||Mar 19, 1992||May 31, 1994||Wisconsin Alumni Research Foundation||Method and apparatus for radiation therapy|
|US5351280||Jun 9, 1993||Sep 27, 1994||Wisconsin Alumni Research Foundation||Multi-leaf radiation attenuator for radiation therapy|
|US5442675||Jun 9, 1993||Aug 15, 1995||Wisconsin Alumni Research Foundation||Dynamic collimator for radiation therapy|
|US5446548||Oct 8, 1993||Aug 29, 1995||Siemens Medical Systems, Inc.||Patient positioning and monitoring system|
|US5528650||Feb 2, 1994||Jun 18, 1996||Swerdloff; Stuart||Method and apparatus for radiation therapy|
|US5548627||Jun 9, 1993||Aug 20, 1996||Wisconsin Alumni Research Foundation||Radiation therapy system with constrained rotational freedom|
|US5625663||Aug 11, 1995||Apr 29, 1997||Wisconsin Alumni Research Foundation||Dynamic beam flattening apparatus for radiation therapy|
|US5647663||Jan 5, 1996||Jul 15, 1997||Wisconsin Alumni Research Foundation||Radiation treatment planning method and apparatus|
|US5651043||Mar 27, 1995||Jul 22, 1997||Kabushiki Kaisha Toshiba||Radiotherapy system|
|US5661773||Jun 7, 1995||Aug 26, 1997||Wisconsin Alumni Research Foundation||Interface for radiation therapy machine|
|US5751781||Oct 4, 1996||May 12, 1998||Elekta Ab||Apparatus for treating a patient|
|US5754622||May 3, 1996||May 19, 1998||Siemens Medical Systems, Inc.||System and method for verifying the amount of radiation delivered to an object|
|US5754623||May 31, 1996||May 19, 1998||Kabushiki Kaisha Toshiba||Radiotherapy system|
|US5835562||Jun 7, 1995||Nov 10, 1998||Hologic, Inc.||Medical radiological apparatus including optical crosshair device for patient positioning and forearm and spinal positioning aides|
|US6094760 *||Aug 4, 1998||Aug 1, 2000||Sumitomo Heavy Industries, Ltd.||Bed system for radiation therapy|
|US6442777||Jul 12, 2000||Sep 3, 2002||Siemens Aktiengesellschaft||Patient support for a diagnostic radiography system|
|US6637056||Jun 3, 2002||Oct 28, 2003||Analogic Corporation||Lifting apparatus and method for patient table|
|US6653806 *||Mar 13, 2000||Nov 25, 2003||Kabushiki Kaisha Yaskawa Denki||Dynamic brake circuit and semiconductor inverter using dynamic brake circuit|
|US6769145||Nov 16, 2000||Aug 3, 2004||Maquet Gmbh & Co. Kg.||Bearing surface for a medical examining-table|
|US6857147 *||Mar 4, 2003||Feb 22, 2005||Ge Medical Systems Global Technology Company, Llc||Synchronization drive for a longitudinal axis telescopic guidance mechanism|
|US6906493 *||Jan 20, 2004||Jun 14, 2005||Molon Motor And Coil Corporation||Electronic brake for motor|
|US6955464||Jun 3, 2002||Oct 18, 2005||Analogic Corporation||Horizontal drive apparatus and method for patient table|
|US7008105||May 9, 2003||Mar 7, 2006||Siemens Aktiengesellschaft||Patient support device for radiation therapy|
|US7028356||Mar 4, 2003||Apr 18, 2006||Ge Medical Systems Global Technology Company, Llc||Multiconfiguration braking system|
|US7552490 *||Jan 24, 2006||Jun 30, 2009||Accuray Incorporated||Method and apparatus for patient loading and unloading|
|US20030107349 *||Jan 16, 2001||Jun 12, 2003||Lawrence Haydock||Ac power generating system|
|US20030222617||May 23, 2003||Dec 4, 2003||Yasuhiro Nakai||Motor control apparatus|
|US20040172756||Mar 4, 2003||Sep 9, 2004||Baskar Somasundaram||Synchronization drive for a longitudinal axis telescopic guidance mechanism|
|US20050114996 *||Jan 10, 2005||Jun 2, 2005||Baskar Somasundaram||Multiconfiguration braking system|
|US20050283911 *||Jun 25, 2004||Dec 29, 2005||Roussy Richard B||Leveling system for a height adjustable patient bed|
|US20060042009||Aug 28, 2004||Mar 2, 2006||General Electric Company||Table drive system for medical imaging apparatus|
|US20070093359 *||Jun 12, 2006||Apr 26, 2007||Mitsubishi Denki Kabushiki Kaisha||Vehicular power control apparatus|
|US20070107128 *||Nov 15, 2005||May 17, 2007||General Electric Company||Braking system for a positioner in a medical imaging apparatus|
|US20080235873||Mar 26, 2007||Oct 2, 2008||General Electric Company||Table drive system|
|US20080279328||Nov 14, 2006||Nov 13, 2008||Koninklijke Philips Electronics N.V.||Systems and Methods Using X-Ray Tube Spectra For Computed Tomography Applications|
|US20090056022||Sep 4, 2008||Mar 5, 2009||Tomotherapy Incorporated||Patient support device|
|US20090070935||Sep 4, 2008||Mar 19, 2009||Tomotherapy Incorporated||Patient support device|
|US20090159677 *||Dec 20, 2007||Jun 25, 2009||General Electric Company||Contactless power and data transfer system and method|
|US20090268491 *||Oct 29, 2009||Igo, Inc.||Power converter with retractable cable system|
|CA2091275A1||Mar 9, 1993||Sep 20, 1993||Stuart Swerdloff||Method and apparatus for radiation therapy|
|JPH11313900A||Title not available|
|WO2005041835A2||Oct 29, 2004||May 12, 2005||Tomotherapy Incorporated||System and method for calibrating and positioning a radiation therapy treatment table|
|1||Extended Search Report from European Patent Office for Application No. 08799171.7 dated Apr. 21, 2011.|
|2||Office Action from Chinese Patent Office for Application No. 200880105521.6 dated Feb. 25, 2011.|
|3||PCT/US2008/075267 International Search Report and Written Opinion dated Jan. 21, 2009.|
|Citing Patent||Filing date||Publication date||Applicant||Title|
|US9038219 *||Jul 29, 2013||May 26, 2015||Siemens Aktiengesellschaft||Patient support apparatus and also a positioning method for positioning a patient's head within a surgical head restraint unit|
|US20140033437 *||Jul 29, 2013||Feb 6, 2014||Patrick Gross||Patient support apparatus and also a positioning method for positioning a patient's head within a surgical head restraint unit|
|U.S. Classification||5/601, 5/611, 318/375, 378/209, 5/616|
|Cooperative Classification||A61B6/467, A61N5/1049, A61B6/0457, A61B6/465, A61B6/547, A61B6/469, H02P31/00|
|European Classification||A61B6/54H, H02P31/00, A61B6/04C, A61B6/46B8, A61B6/46D|
|Oct 24, 2011||AS||Assignment|
Owner name: TOMOTHERAPY INCORPORATED, WISCONSIN
Free format text: ASSIGNMENT OF ASSIGNORS INTEREST;ASSIGNORS:KURO, SERGE NUKHIM;REITZ, GRAHAM;BRUNKER, BRAD;AND OTHERS;SIGNING DATES FROM 20081017 TO 20081022;REEL/FRAME:027108/0012
|Oct 7, 2015||FPAY||Fee payment|
Year of fee payment: 4
|Jan 13, 2016||AS||Assignment|
Owner name: CERBERUS BUSINESS FINANCE, LLC, AS COLLATERAL AGEN
Free format text: ASSIGNMENT FOR SECURITY - PATENTS;ASSIGNORS:ACCURAY INCORPORATED;TOMOTHERAPY INCORPORATED;REEL/FRAME:037513/0170
Effective date: 20160111